2 Multi-Parameter Resting 12-Lead PC Based Resting 12-Lead ECGNotebook Based Resting 12-LeadPDA / Smart Phone Resting 12-LeadAmbulance to ER Resting 12-LeadMulti-Hour 12-Lead MonitoringBlue Tooth 100 MetersIncludes Simple Stress Test Program4.0 Standard or Premier ProgramsOptional 5.0 Stress Test Program
3 Multi-Parameter Resting 12-Lead Interpretation / SuggestionSerial ComparisonSAECG Late PotentialsVectorcardiographyHi-Frequency 12-Lead AnalysisQT, QTc, and QTd AnalysisAxis DeviationFrequencyCardioGramPage-Scan Full-Disclosure Review
4 Operational Options Standard ECG Quick ECG Multi-Hour ECG Monitoring Simple Stress Test
5 Quick ECG ECG Acquisition for a few seconds. ECG, Time, and Heart Rate are displayed.Upon display of clean ECG, perform a mouse click.ECG is immediately analyzed for interval measurements.Upon verification of correct intervals, perform a mouse click12-Lead test is completed.
7 HM-CARDIO MD 12-Lead ECGThe 4.0 Standard program includes the 12-Lead ECG program, plus the Interpretation program.The Interpretation program includes the Axis Deviation analysis.The quick visual edit of the vertical marker locations for:* Beginning of P-wave* End of P-wave* Beginning of Q* End of S* Beginning of T* End of Tprovides the correct foundation for the Interpretation program.HL7 and Dicom outputs available.
8 HM-CARDIO MD 12-Lead ECG The 4.0 Premier program includes: * Interpretation* Simple Stress Test* SAECG Late Potentials* QTc & QTd* FrequencyCardioGram (FCG)* Multi-Hour ECG with HRV* Hi-Frequency ECG* VectorCardioGraphy (VCG)ECG Acquisition is with:* Blue Tooth Transmitter (100 meters)* USB Hi-Res Acquisition Box* HM-Holter MINI-RecorderA low-cost and simple Holter CS-6A program is an optional add-on to the Winnervision program.
9 PDA / Mobile Phone 12-Lead ECG Blue Tooth sends 12-Lead ECG to PDA or Smart Phone (PSP), with the Resting 12- Lead software.PSP displays, analyzes, stores, and has printer outputs for the 12-Lead.ECGs can be collected for a large number of patients, and then transferred to the 12-Lead PC station in the hospital’s EKG Lab.12-Lead ECG strips can be ed to PCs or other physicians. This makes for a pocket ECG system that the physician can take into any out-of-the-office environment.Ideal for sports medicine and Ambulance monitoring environments.
10 Ambulance Resting 12-Lead ECG Ambulance to hospital ECG for immediate treatment of cardiac patient emergencies.Blue Tooth ECG to Smart Phone.Smart Phone transmits 12-Lead ECG on demand or at pre-set intervals.Ambulance personnel receive verification for life-saving interventions.Upon arrival at hospital, 100% of Ambulance 12-Lead ECG is transferred to CS-12 for review in real-time mode and Holter mode.ECG continues in Emergency Room.
11 Multi-Hour ECG Monitoring 12-Lead Hospital Patient Monitoring with Blue Tooth.Single patient ECG station.Analysis of:ArrhythmiasSTQTc and QTdHeart Rate VariabilityPatient can be mobile with Blue Tooth Transmitter.Sample rate of 4,096 with12-Bit resolution.
12 Simple Stress Test Included in 4.0 Premier. Displays: Heart Rate Target Heart Rate% of Target Heart RateDelta Heart Rate ChangeElapsed TimeThe reference beginning ECG is shown at the far left side for each ECG Lead.ECG prints available at all times during Stress Test.Stress Test reviewed in Page Scan mode.
13 Resting 12-Lead with Simple Holter Resting 12-Lead security key is also programmed with the simple CS-6a Holter program.For the non-cardiologist market.CS-6a consists of a 24-hour mini-Holter recorder (300-4) and a Holter software program.The can be a 3-Lead or 12-Lead recording for 24-hours.CS-6a is a Page-Scan and Full Disclosure program. There is no Template editing, but the program includes the Category Event displays of ECG abnormalities that can be edited,
14 HM-CARDIO MD program includes: HM-CARDIO MD SAECG Late PotentialsVectorcardiographyHi-Frequency ECGQTcSAECG program allows for manual edit.Vectorcardiography also includes the Timed VCG which allows the VCG Loops to be directly compared with the XYZ ECG.HM-CARDIO MDSAECG Late PotentialsSAECGVectorCardioGraphyVCG
16 FCG CADgram FrequencyCArdioGram The FCG CADgram detects and locates blockages in the coronary arteries.It does not require any exercise protocol. The Resting 12-Lead is the preferred environment for the FCG test.A positive FCG CADgram is a strong indication for an exercise stress test.
17 Hi-Frequency ECG Mini ECG Pocket Card High Resolution QRS ECGHi-Frequency ECGThe Hi-Frequency ECG is to closely examine the notches and slurs of the expanded QRS.Mini ECG Pocket CardThe program produces a ECG patient information card that can be carried in the patient’s wallet or purse.Mini ECG Card for Patient Wallet or Purse
18 Holter Add-On to HM-CARDIO MD Holter Full Disclosure
19 Holter Add-On toHM-CARDIO MD Sixteen (16) Ventricular Ectopic ECG’sUses device.For the non-cardiologist market.This is a Page-Scan and Full Disclosure Holter ECG system with ECG print-outs.The Category ECG Events of Pause, ST, VE, V-Runs, SVE, Min. HR, Max. HR, etc. are displayed for ECG acceptance or rejection.PDF ing of ECG data is included.Database storage and retrieval is included.Instant of Holter Report Pages
20 Specifications Blue-tooth ECG transmitter and receiver HM-CARDIO MD BLUE TOOTHMethod of transmitting and receiving: Digital;Transmitting and receiving frequency: 2.4G Hz;Working distance: Model: 15m visible;Transmitter voltage: DC 6.0V;Receiver voltage: DC 5.0V.Blue-tooth ECG transmitter and receiverSample rate: <=4096Hz;Sample resolution: 12bit;Lead: 12-lead standard ECG;Input circuit current: ≤0.1μA;Input impedance: ≥5 MΩ;Noise electric level: ≤15μVp-p;Standard sensitivity: 10mm/mV ±5%;Anti-polarization voltage: if add ±300mV DC polarization voltage, the change of sensitivity is not more than ±5%;CMRR: ≥60 dB;Frequency response: Hz（+0.4 dB, -3.0 dB）;Lower frequency feature: time constant is not less than 3.2s;Stability: when the power supply transient change is 10%, the baseline wondering is not more than ±5%. The sensitivity error is not more than ±5%.50Hz filter: ≥20dB;Equipment safety type: the wired ECG acquisition box is class II, type-BF; the blue-tooth ECG transmitter is type-BF with internal power; the blue-tooth ECG receiver is class II, type-B;Working environment: Temperature: ℃, Relative humidity: ≤70%, Atmospheric pressure: hPa;Storage environment: Temperature: ℃, Relative humidity: ≤93%, Atmospheric pressure: hPa.
21 Equipment Overview Bluetooth Transmitter (front) Bluetooth Transmitter (back)Bluetooth ReceiverPatient Cables
22 Operating Instructions 1. Connect the Bluetooth receiver to the USB port of the computer.
23 Operating Instructions 2. Connect the patient cable to the Bluetooth transmitter. 3. Insert the batteries into the Bluetooth transmitter.
24 Operating Instructions 4. Hold the power switch at the top of the Bluetooth transmitter for 3 seconds. The LED lights on the front will light up to indicate it is on and active.