Download presentation

Presentation is loading. Please wait.

1
**Bioimpedance Measurement**

Dept. of Biomedical Engineering YOUNHO HONG

2
**Impedance measure + Z - constant current source i=0 measured known**

resistance reactance

3
**Impedance measure (Example) You are given with i(t) = I sin(2πft)**

1 1+j Re Im i V t 1 You are given with i(t) = I sin(2πft) You measure v(t) = V sin(2πft+θ)

4
Impedance measure (Example) Z

5
Impedance measure

6
**Impedance measure Z X ? phase-sensitive Demodulation sin(2πft)**

cos(2πft) phase-sensitive Demodulation

7
Impedance measure In-phase Component

8
Impedance measure Quadrature Component

9
**Impedance measure V=VI+jVQ Vsin θ V θ Vcosθ (Quadrature Component)**

(In-phase Component) (Quadrature Component) V=VI+jVQ

10
**Two-electrode Method Inject current and then measure voltage V I skin**

equivalent circuit V Inject current and then measure voltage ZC1 ZC2 Zb

11
**Two-electrode Method Im Re Zbcosθb Zc1cosθ1 Zc2cosθ2 Zc1sinθ1 Zc2sinθ2**

Zbsinθb

12
**Four-electrode Method**

V I skin E1 E3 E4 E2 ZC1 ZC3 Zb1 Zb2 ZC4 ZC2 Zb

13
**Three-electrode Method**

V I skin E1 E3 E2 ZC1 ZC3 ZC2 Zb1 Zb We can measure skin impedance using Three and Four electrode Method

14
Noise Effect

15
**Spectrum Analizer “ DFT “ f1 f2 f3 fN … X V(t) … A1 A3 A2 AN**

sinw1t / cosw1t V(t) … sinw2t / cosw2t sinwNt / coswNt “ DFT “

16
**Bio Impedance Spectroscophy**

Z + - Repeat measurement for many frequencies θ f Zb f

17
Thank you.

Similar presentations

© 2017 SlidePlayer.com Inc.

All rights reserved.

Ads by Google