We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byKiley Hearing
Modified over 2 years ago
Copyright © Yokogawa Electric Corporation MW100 Quick Start Guide The following guide is designed to help the user take a MW100 system and quickly get an input channel and math channel configured for measurement and recording and then display the results in real-time. Yokogawa Network Solutions Presents:
Copyright © Yokogawa Electric Corporation Getting Connected Part 1 Many computers are assigned addresses on a network via DHCP. When not connected to a DHCP network, the computer fails to get a proper IP address assigned to it and defaults to a 169 series address. When initially configuring a MW100 – you may need to use an Ethernet cross-over cable and a static IP address for your PC.
Copyright © Yokogawa Electric Corporation Getting Connected Part 2 Install the MW100 software that comes with each Head unit and open the MW100 IPConfig software. If for some reason the software cannot find your MW100 – ensure that security software is not blocking the search effort. Alternately the MW100 can be temporarily set to address by setting dip-switch 8 on the front of the unit to the down position and restarting the unit. Connection can then be made through a web browser by browsing to this address.
Copyright © Yokogawa Electric Corporation Getting Connected Part 3 If Parts 1 and 2 worked, then open a web browser and type the IP address of the MW100 into the address bar and connect to the MW100. If necessary configure communication settings locally on the MW100 through Communication Setting link.
Copyright © Yokogawa Electric Corporation Reconstructing MW100 System Whenever an MW100 system is first assembled or a module is changed – it needs to be reconstructed so the modules can be individually configured. An MW100 system should automatically detect what modules are attached. If attaching a new module to an older MW100 system – ensure the MW100 head unit has the latest firmware so that it can recognize the module.
Copyright © Yokogawa Electric Corporation Setting Measurement Rates The MW100 can scan input modules at up to three different rates – with a single rate assigned to each module. All channels within a module will be scanned every interval. Select Apply to execute changes.
Copyright © Yokogawa Electric Corporation Setting Math Rate MW100 has powerful math channels that can be used to give input channels offsets, calculate flow totals, perform logical operations, and many other useful features. Math channels will calculate at the specified interval. Select Apply to execute changes.
Copyright © Yokogawa Electric Corporation Setting Recording Options The MW100 supports interval based recording and event (trigger based) recording. Select Apply to execute changes.
Copyright © Yokogawa Electric Corporation Set Date and Time All data recorded by the MW100 is time and date stamped – ensure the correct information is set within the MW100 as well as on PCs that will be viewing the data files. Confirm that time zone and daylight savings time are appropriately set as well or data may be incorrectly offset.
Copyright © Yokogawa Electric Corporation Setting Temperature Unit Yokogawa data acquisition systems treat the temperature unit as a fundamental setup parameter and it is important to select this early on in the configuration process as changing it once a unit is fully configured will reset items like channel configurations and other settings. Select Apply to execute changes.
Copyright © Yokogawa Electric Corporation Configuring Inputs Channels can be individually configured with universal input modules supporting TCs, RTDs, VDC, DI (level or contact) and ADC via a shunt. Select Apply to execute changes. Be careful to select the appropriate Apply as the second Apply – applies global settings (bulk changes).
Copyright © Yokogawa Electric Corporation Configuring Math Channels Math channels can be recorded and displayed like regular input channels and allow for powerful capabilities like offsets, totals, logical conditions, and other more advanced features. Select Apply when to execute changes.
Copyright © Yokogawa Electric Corporation Configuring an Input Channel Alarm The MW100 has a powerful alarm system that supports up to 4 alarms per input channel as well as 4 alarms per math channel. Select Apply to Execute changes. Note that if the underlying channel is changed then the associated alarm setting may be reset. Yokogawa implemented this as a safety procedure to ensure that alarms are forced to be reconfigured when an associated channel is changed.
Copyright © Yokogawa Electric Corporation Configuring Tag Names Optionally each input channel and math channel can be given a name. Select Apply to execute changes.
Copyright © Yokogawa Electric Corporation Configuring Display Groups The MW100 allows for up to 18 groups of channels to be configured for display purposes. In monitoring mode – up to two groups can be seen at a time with the ability to change groups on the fly. Select Apply to execute changes.
Copyright © Yokogawa Electric Corporation Display by Channel Number or Tag Name When groups of channels are displayed in the real-time monitor they can be viewed by name or channel number. If no name is configured – the display will revert to channel number. Select Apply to execute changes.
Copyright © Yokogawa Electric Corporation Display by Channel Number or Tag Name Now that the MW100 has been configured, it is time to monitor it in real-time. Ensure that the MW100 software has be installed on the PC or that the PC has the latest copy of Sun’s JAVA virtual machine installed (www.java.com) so that real-time display modules (JAVA applets) can execute. Select Apply when switching operation modes.www.java.com
Copyright © Yokogawa Electric Corporation Real-time Monitoring of Two Groups The MW100 web interface can be used to display two groups of channels simultaneously in different formats. It can also be used to manually set analog outputs, insert messages into trends, start/stop math, start/stop recording, acknowledge alarms, and manually set digital outputs. If the interface fails to populate properly there could be a problem with the JAVA virtual machine (conflicting versions or it may not be installed).
Copyright © Yokogawa Electric Corporation Alarm Summary The MW100 has a powerful alarming system that can log and display alarm status. The Alarm Summary can be reached from the main web menu or from the real-time monitoring page and will include time and date stamps for all logged alarms. The alarm summary page does not update automatically – so you will need to select the update button or refresh the page to see new alarms.
1 Fluke DAQ Software Visual Demo Guide 2680A-APSW.
DEMONSTRATION FOR SIGMA DATA ACQUISITION MODULES Tempatron Ltd Data Measurements Division Darwin Close Reading RG2 0TB UK T : +44 (0) F :
DSL-2870B How to Change ADSL Username and Password in your modem router How to Change Wireless Channel in your modem router How to Open Ports in.
11 ©Copyright By D-Link HQ TSD James Chu Smart III Switch Smart Console Utility Ver 2.
Self Guided Tour of TagMate™QC Display Tag DATA Tag Data: PRINT TagMate™QC SETUP Display STATS Tags: START and READ SEND All Data to PC Additional Capabilities.
Advantages of using R.PC.R Software Now that you know some of the advantage let’s take a look at the software … Reliable and Comprehensive Reporting Recording.
Riso Digital Education Center Appendix for SRC Training Basic Course 1.
FireClass FC501. What’s FC501 ? An entry level Triple Circuit Single Loop addressable system featuring Intelli-Zone mapping An “out of the box” panel.
Inputs: Digital and 4 configurable for Digital or 10 bit Analog Outputs:4 - 8 amp Relays or 700ma Transistors (1 PWM) 4 Line by 18 Character fully.
DSL-2730B, DSL-2740B, DSL-2750B. Agenda DHCP Reservation Port Forwarding Guide MAC Filtering Web Filtering How to Enable Security on Wireless.
An Overview When Connecting to Yaskawa Drives Date: 8/14/06, Rev: PP.AFD.26.
EMerge Browser Managed Security Platform Module 3: Startup eMerge Certification Course Physical connection TCP/IP Characteristics of PC Initial connection.
Home Media Network Hard Drive Training for Update to 2.0 By Erik Collett Revised for Firmware Update.
TAX-AIDE Network Router Setup Network Printer Setups July SMT/TCS Training - Dallas1.
CENTURION™ (C4-SERIES) Erin Cox, Market Research Analyst, Natural Gas Production Controls - Presenter Sanjay Kumar, Market Research Specialist & Product.
Yokogawa Electric Corporation ISD-MASP-S05010 Copyright © Yokogawa Electric Corporation February 16, 2005 Solution Marketing Group Industrial Automation.
Yokogawa Corporation of America Network Solutions Business Unit DAQMASTER Features and Benefits Yokogawa Corporation of America Network Solutions Business.
Application Guide For Mesh AP – MAP-3120 How to Quick Configure MAP-3120 for the first time? FW:
Module 6 – Switch Configuration CCNA 3 Cabrillo College.
Ch. 6 – Switch Configuration CCNA 3 version Switch LED indicators.
DNR-322L & DNR-326. Agenda: How to Access DNR-322L using mydlink account How to Access the Web Interface of DNR-322L_DNR-326 via NVR Search Utility How.
Alarm Mode An in-depth guide. Overview Main module for loading configuration and monitoring system – Reader Open/Close – Arm/disarm zones – Control relays.
6 February LAN-50/500 Ethernet Communication.
Initial Set Up KX-NS1000 For step by step : 16 May, 2012.
Proctor Caching Overview. 2 Proctor Caching Diagram.
Simultaneous Dual Band Wireless N Router CY-SWR CY-SWR1100 Dual Band Wireless N Router Reporting Date :
MCDST : Supporting Users and Troubleshooting a Microsoft Windows XP Operating System Chapter 5: User Environment and Multiple Languages.
System Startup Configuring APACS & AAN Controllers.
Quick guide to ASIMON configuration For version 3.0 or greater SAFETY AT WORK Date: 3/18/2009.
AQS Web Quick Reference Guide Changing Raw Data Values Using Maintenance 1. From Main Menu, click Maintenance, Sample Values, Raw Data 2. Enter monitor.
Iomega iConnect Wireless Datastation Training Presentation By Erik Collett
System Administration and Basic Functionality Version 4.0 – September 2007 Q-Advisor Quick Start.
1 Basic Installation and GUI Tech Basic Installation and GUI : Objectives Installing the Quadro Configuring the Quadro Installing IP phones.
ActEyes IP Cam Training Part 1: Camera Setup. Network Setup Overview Each IP Cam on the network will need its own IP address to be assigned. This address.
Copyright © 2014 EMC Corporation. All Rights Reserved. Linux Host Installation and Integration for Block Upon completion of this module, you should be.
Ch. 6 – Switch Configuration CCNA 3 version Overview Identify the major components of a Catalyst switch Monitor switch activity and status using.
Monitoring systems COMET types MS55 & MS6 MS55/MS6 Configuration.
NAS-7101 / NAS-7201 TorrentFlux Utility Installation SATA NAS Server Copyright © PLANET Technology Corporation. All rights reserved.
Getting to Know the Digital Hub Can you sketch/draw the digital hub system – subsystems? Can you Identify Point of failures?
7200 Samsung Confidential & Proprietary Information Copyright 2006, All Rights Reserved. 1/16 OfficeServ 7200 Enterprise IP Solutions Data Server S/W Upgrade.
DWR-113 FAQ’s 3G WiFi Router. Agenda: How do I set up and install my router How do I change the Administrator password on my router I have forgotten.
Date: File:PRO1_12E.1 SIMATIC S7 Siemens AG All rights reserved. Information and Training Center Knowledge for Automation Troubleshooting.
DAHUA TECHNOLOGY… …MAKE YOUR LIFE SAFER 浙江大华技术股份有限公司 ZheJiang Dahua Technology CO., LTD. Associate Certified Class Dahua Security Certification Smart PSS.
Tapit Call Accounting software for the 21st Century Copyright Trisys, Inc Trisys, Inc 215 Ridgedale Avenue Florham Park, NJ (973)
MCDST : Supporting Users and Troubleshooting a Microsoft Windows XP Operating System Chapter 9: Troubleshooting Power Management and I/O Devices.
NETGEAR CONFIDENTIAL Smart Switches FS526T / FS750T / GS748T / GS724T.
CACI Proprietary Information | Date 1 PD² SR13 Client Upgrade Name: Semarria Rosemond Title: Systems Analyst, Lead Date: December 8, 2011.
ETM Series Digital Video Recorders Now Available in 120fps with 4, 9, or 16 Channel Systems The Fully Featured, Affordable Solution for Entry Level Markets.
Specview 32 Release 2.5 Enhancements February 2008.
© 2017 SlidePlayer.com Inc. All rights reserved.