We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byJohnny Crooke
Modified about 1 year ago
Simdis Trouble Shooting AC Analytical Controls February 2001
Problem areas Contamination Leaks and column septum bleed Electronic / mechanical failure
Baseline symptoms Ghost peaks : Contamination by septum Solvent / wash bottles in auto sampler is contaminated Wrong syringe in auto sampler
Baseline drift Leak Carrier gas line Improper column installation Saturated filter Carrier gas purity Column contaminated with HB components
Poor peak shape Column overload Bad column – liner connection Broken column Old column
No peaks Auto sampler / syringe problems Incorrect signal assignment Flame out No carrier gas
Poor Separation Bad column – liner connection Check method instrument settings – Flow too high Contaminated calibration mixture Old column
Reference Off spec. Contaminated solvents causes a bad blank Detector nut leak Incorrect calibration table settings Septum leak Column overload
Low Calibration Peaks Preparation of calibrant mixture Hydrogen / air ration on FID Detector leak Restricted FID Wrong FID jet
ENVE 202 Dr. Aslıhan Kerç GAS CHROMATOGRAPHY. Gas Chromatography (GC) *Gas chromatography is a chromatographic technique that can be used to separate.
Conversion of Agilent EI GC/MSD Systems From Helium To Hydrogen Carrier Gas Bruce D. Quimby, Ph.D. GPD Solutions Group and Harry F. Prest, Ph.D. GC/MS.
Service Section Technical Training. Air Conditioning Troubleshooting.
Advanced Topics in STR DNA Analysis AAFS 2006 Workshop #6 Seattle, WA February 20, 2006 Dr. John M. Butler Dr. Bruce R. McCord CE Troubleshooting
Steering Systems Chapter 44. Steering system problem diagnosis Steering system maintenance Steering column service Manual steering gearbox service (recirculating-ball.
Guilin, January, 9 – 13, 2006 Dr. Birgit Schmauser, BfArM, Bonn Verification of applicability of the validated/compendial API analytical method for the.
GC and GC-MS. Gas Chromatography Function Components Common uses Chromatographic resolution Sensitivity.
Steering Solutions Services Corp. 11/17/ Legged 5 Why Analysis 3 Legged 5 Why – Effective Root Cause Analysis A Focused Approach to Solving Chronic.
2400 Series 2 CHNS/O Analyser Paul Gabbott PETA Page 2.
GAS CHROMATOGRAPHY. Classification based on Mobile Phase Gas Chromatography Gas - solid Gas - liquid Stationary Phase Sample MUST be volatile at temperatures.
INDUSTRIAL HYGIENE - SAMPLING OF GASES AND VAPORS UNIVERSITY OF HOUSTON - CLEAR LAKE.
Trouble Shooting (Mechanical Ventilation) Arthur Sadhanandham Medical ICU, CMC.
Binks Spray Tips July 2010 Surface Preparation - General Anything you dont want to see in the final finish must be removed from the surface Surface.
DETERMINING THE CORRECT CONCENTRATIONS OF VOC IN COOLING TOWER WATER John W. (Jack) Small President Star Instruments, Inc. 100 Park Avenue League City,
CAUSE & EFFECT DIAGRAM (Fishbone or Ishikawa Diagram) Dr. Ömer Yağız Department of Business Administration Eastern Mediterranean University TRNC.
Instrumental Analysis The course is designed to introduce the student to modern methods of instrumental analysis In modern analytical chemistry. The focus.
MERCEM300Z Intelligent mercury monitoring Florian Greiter, Produkt Management.
INDUSTRIAL HYGIENE - PRINCIPLES AND INSTRUMENTATION FOR CALIBRATING AIR SAMPLING EQUIPMENT UNIVERSITY OF HOUSTON - CLEAR LAKE.
1. Four LLs were published in February Transmission Relaying – Undesired Blocking 2.Lack of Separation for Critical Control Power Supply Leads.
INDUSTRIAL HYGIENE DIRECT-READING INSTRUMENTS FOR GASES, VAPORS, AND PARTICULATES UNIVERSITY OF HOUSTON - DOWNTOWN.
Leisure Vac Auto pool Cleaner. Trouble shooting Guide: Top Head Assembly Problem: Debris in the top head, causing cleaner to not work due to poor water.
R410a Training R410 Refrigerant… Why and When? Basic Refrigeration Cycle Importance of Airflow Superheat and Subcooling Installation Troubleshooting.
1 Gas Chromatography There an be many parts to a gas chromatography system but the basic components include: An injection system. A column (controllable.
INDUSTRIAL HYGIENE - AIR SAMPLING UNIVERSITY OF HOUSTON - DOWNTOWN.
Measurements in Fluid Mechanics 058:180:001 (ME:5180:0001) Time & Location: 2:30P - 3:20P MWF 218 MLH Office Hours: 4:00P – 5:00P MWF 223B-5 HL Instructor:
Chemical Sensors for Autonomous and Lagrangian Platforms Ken Johnson, Monterey Bay Aquarium Research Institute.
Wall Mounted J series Troubleshooting. Troubleshooting area Electrical circuit / hardware Refrigerant circuit / component.
HVAC | Slide 1 of 29 May 2006 Heating, Ventilation and Air- Conditioning (HVAC) Part 2: HVAC systems and components Supplementary Training Modules on Good.
RESNET STANDARD 2004 – 03 BLOWER DOOR TEST PROCEDURES RESNET NEW REQUIREMENTS January 1, 2007.
© 2012 Delmar, Cengage Learning Chapter 30 Oxyfuel Welding and Cutting Equipment, Setup, and Operation.
© 2016 SlidePlayer.com Inc. All rights reserved.