Presentation on theme: "Validation & Sample Size Selection"— Presentation transcript:
1 Validation & Sample Size Selection Quality Day Orange Empire ASQ – November 1, 2013Larry Bartkus
2 Verification and Validation Verification: Confirmation by examination and provision of objective evidence that the specified requirements have been fulfilled.Validation: Establishing objective evidence that something consistently produces a result or product meeting its predetermined requirements.
3 What Does This Mean?Where the results of a process cannot be fully verified by subsequent inspection and test, the process shall be validated with a high degree of assurance and approved according to established procedures.Validation can use tools such as: Verification uses:Design of Experiments ExaminationRegression Analysis TestWorst case testing InspectionThis entire process requires a careful analysis of materials, processes, requirements, and test capabilities to determine what exactly can be verified and what cannot be verified.
4 Other DefinitionsProcess validation – establishing by objective evidence that a process consistently produces a result or product meeting its predetermined specifications.Design validation – establishing by objective evidence that device specifications conform with user needs and intended use(s).Installation Qualification – Establishing confidence that process equipment and ancillary systems are capable of consistently operating within established limits and tolerances.Process Performance Qualification – Establishing confidence that the process is effective and reproducible.Product Performance Qualification – Establishing confidence through appropriate testing that the finished product produced by a specified process(es) meets all release requirements for functionality and safety.
5 QualificationsQualifications are really a subset of Validation. These address the specific equipment, process, and product.Webster refers to qualification as “a restriction in meaning or application” and “a limiting modification” and also “a condition or standard that must be complied with”.We are then talking about very specific equipment, processes, and product characteristics. The specificity is demonstrated in the Regulations as follows:“Installation qualification studies establish confidence that the process equipment and ancillary systems are capable of consistently operating within established limits and tolerances.”Each process shall be defined and described with sufficient specificity so that employees understand what is required.‘it is necessary to demonstrate that the specified process has not adversely affected the finished product.”
6 Product Development Phases Phases in the Product Development ProcessConcept and Feasibility1Design and Development2Final Validation & Ramp-up3Market Launch4
7 DRDs, CTQsIt is essential to quantify all important quality characteristics and test for them. Design Requirement Documents or Product Matrices are a must. The CTQ (Critical to Quality) Tree can be a good tool for this effort!
8 Laws and Regulations FDA 21 CFR §820.75 § Process ValidationWhere the results of a process cannot be fully verified by subsequent inspection and test, the process shall be validated with a high degree of assurance and approved according to established procedures. The validation activities and results, including the date and signature of the individual(s) approving the validation and where appropriate the major equipment validated, shall be documented.Each manufacturer shall establish and maintain procedures for monitoring and control of process parameters for validated processes to ensure that the specified requirements continue to be met.Each manufacturer shall ensure that validated processes are performed by qualified individual(s).For validated processes, the monitoring and control methods and data, the date performed , and where appropriate, the individual(s) performing the process or the major equipment used shall be documented.(c) When changes or process deviations occur, the manufacturer shall review and evaluate the process and perform revalidation where appropriate. These activities shall be documented.
9 Standards ISO Clause 7.3.6Design and development validation Clause of ISO Design and development validation shall be performed in accordance with planned arrangements (see 7.3.1) to ensure that the resulting product is capable of meeting the requirements for the specified application or intended use. Validation shall be completed prior to the delivery or implementation of the product (see Note 1).Records of the results of validation and any necessary actions shall be maintained (see 4.2.4).As part of design and development validation, the organization shall perform clinical evaluations and/or evaluation of performance of the medical device, as required by national or regional regulations (see Note 2).NOTE 1 If a medical device can only be validated following assembly and installation at point of use, delivery is not considered to be complete until the product has been formally transferred to the customer.NOTE 2 Provision of the medical device for purposes of clinical evaluations and/or evaluation of performance is not considered to be delivery.
10 Standards ISO Clause 7.5.2Validation of Processes for Production and Service Provision of ISO Clause General RequirementsThe organization shall validate any processes for production and service provision where the resulting output cannot be verified by subsequent monitoring or measurement. This includes any processes where deficiencies become apparent only after the product is in use or the service has been delivered.
11 GHTF Overview of N99 The Global Harmonization Task Force Quality Management Systems - Process Validation GuidanceThis process validation guidance is intended to assist manufacturers in understanding quality management system requirements concerning process validation and has general applicability to manufacturing (including servicing and installation) processes for medical devices. The guidance provides general suggestions on ways manufacturers may prepare for and carry out process validations.This guidance provides general suggestions on ways manufacturers may prepare for and carry out process validations. Other ways may be equally acceptable; some regulatory requirements place the responsibility on the manufacturer to specify those processes which require validation and the qualification of personnel who operate validated processes. Regardless of the method used to validate the process, records of all validations activities should be kept and the final outcome documented.
12 Validation in Software This guidance applies to: · Software used as a component, part, or accessory of a medical device; · Software that is itself a medical device (e.g., blood establishment software); · Software used in the production of a device (e.g., programmable logic controllers in manufacturing equipment); and · Software used in implementation of the device manufacturer's quality system (e.g., software that records and maintains the device history record). This document is based on generally recognized software validation principles and, therefore, can be applied to any software. For FDA purposes, this guidance applies to any software related to a regulated medical device, as defined by Section 201(h) of the Federal Food, Drug, and Cosmetic Act (the Act) and by current FDA software and regulatory policy. This document does not specifically identify which software is or is not regulated.
13 Process Validation in Pharmaceuticals The CGMP regulations for validating pharmaceutical (drug) manufacturing require that drug products be produced with a high degree of assurance of meeting all the attributes they are intended to possess (21 CFR (a) and (a)). Process Validation and Drug Quality Effective process validation contributes significantly to assuring drug quality. The basic principle of quality assurance is that a drug should be produced that is fit for its intended use. This principle incorporates the understanding that the following conditions exist: Quality, safety, and efficacy are designed or built into the product. Quality cannot be adequately assured merely by in-process and finished-product inspection or testing.
14 Why Process Validation is important? Process Validation ContinuumLotsProcessParametersProduct/Process DevelopmentProcess ControlProcess Validation (IQ/OQ/PQ/PPQ)Process Validation sets the optimal range and determines the process control requirements to ensure long-term success.
17 Discussion: Validation or Verification? Sterilization processesManual cutting processesAseptic filling processesTesting for color, turbidity, total pH for solutionsHeat treating processesVisual inspection of printed circuit boardsNumerical control cutting processesValidation Verification Verification, but validation may be preferred
21 RisksAlpha Risk is the probability of rejecting a good lot. It can be expressed as: Alpha=1-Probacc (at the AQL level). It can also mean in hypothesis testing rejecting the null hypothesis when it is true. Alpha is also known as a Type I Error or the “Producers Risk” Beta Risk is the probability of accepting a bad lot. It can be expressed as: Beta=Probacc(at the LTPD level) It can also mean accepting a hypothesis that is false. Beta is also known as Type II Error or the “Consumers Risk”
30 Calculations using Software (Variables) From Minitab
31 Power AnalysisThe power of a statistical test is the probability, given that Ho is false, of obtaining sample results that will lead to its rejection. “Power of a test” is clearly an important characteristic. To put it in other words, a powerful test is one that has a high probability of claiming that a difference exists when it really does. -The larger the effect size, d, the greater the power. -For any given effect size (other than zero) the larger the sample size, the greater the power of the test.
Your consent to our cookies if you continue to use this website.