Download presentation

Presentation is loading. Please wait.

Published byAngelina Quinn Modified over 4 years ago

2
Irradiation of 300 m phi detectors from MICRON and HAMAMATSU cont. Second independent estimate: 6h 2 cycles, 12h 1 cycle, 14h no beam, 5.5h 1cycle estimated fluence by program assuming constant flux = real fluence / 1.37 Best estimate: real fluence = (1.5 +/- 10%) measured fluence Best estimate: real fluence = (1.5 +/- 10%) measured fluence

3
Irradiation of 300 m r detector from MICRON 12 July 18:30, Calibration of beam w/o detector SEC=2.71 10 6, fluence = 0.16 10 14 p/cm 2 F- factor = 5.9 10 6 Estimate SEC=50.8 10 6 for 3 10 14 p/cm 2 20:45, Start irradiation with detector 13 July 17:15, stop SEC=49.2 10 6 Highest fluence measured 2.93 10 14 p/cm 2 F- factor = 6.0 10 6 No complication with 1 and 2 cycles, only 30min stop of PS measured fluence is OK

4
Irradiation of 300 m phi detectors from MICRON and HAMAMATSU cont. Al for profile measurement and final calibration total time: 39h, first 6 h: 2 cycles, 14 h with no beam numbers given by program are completely unreliable Try to get correct estimate: Need to assume that the beam detector didnt changed, ( 2 cycles/ 1 cycle, shuttle problem, PS camera for beam steering ) 1. Add different calibrations: 1+3+4 = 4.5 10 14 p/cm 2 4.5 / 2.8 = 1.6 ? (assumes Al foil for calibration was in the center of the beam) Correction for 6h 2cycles and 12h 1cycle on calibration 3: 3.35 10 14 p/cm 2 3.68 10 14 p/cm 2

5
Irradiation of 300 m phi detectors from MICRON and HAMAMATSU 26 July 16:55, Detectors + Al foils in beam First calibration 16:55 - 18:15 SEC=2.769 10 6, fluence = 0.14 10 14 p/cm 2 F-factor = 5.1 10 6 Second calibration 18:38 - 27 July 10:38 SEC=71.98 10 6, fluence = 2.14 10 14 p/cm 2 F-factor = 3.1 10 6 !!!! (this calibration was used to define final SEC counts for 3 10 14 p/cm 2 to be 98.87 10 6 ) Third calibration 18:38 - 27 July 12:48 SEC=76.15 10 6, fluence = 3.35 10 14 p/cm 2 F-factor = 4.6 10 6 (nobody noticed fluence > 3 10 14 p/cm 2 ) Fourth calibration 28 July 2:33 - 8:12 SEC=89.69 10 6, fluence = 0.71 10 14 p/cm 2 F-factor = 5.2 10 6 additional complications: First 6 h with 2 cycles, irradiation stopped 27 July 12:48 - 28 July 2:33

6
Irradiation of 200 m phi detector from MICRON 28 July 12:50, Detector + Al foils in beam Calibration starts at 14:26, SEC=2.013 10 6 Calibration stops at 16:05, SEC=5.265 10 6 measured fluence = 0.213 10 14 p/cm 2 SEC = 3.252 10 6 F-factor = 6.5 10 6 (normal) wanted 5 10 14 p/cm 2 SEC = 81.6 10 6 instead goal set to: SEC = 150 10 6 ! Explains factor 2 higher dose 29 July 13:32, irradiation stops, SEC = 146.3 10 6 Minor correction: 5 hours 1 cycle, 19 hours 3 cycles real fluence 9% less than estimated by program divide all numbers by 1.09

Similar presentations

Presentation is loading. Please wait....

OK

When you see… Find the zeros You think….

When you see… Find the zeros You think….

© 2018 SlidePlayer.com Inc.

All rights reserved.

To make this website work, we log user data and share it with processors. To use this website, you must agree to our Privacy Policy, including cookie policy.

Ads by Google

Ppt on operating system memory management Ppt on porter's five forces article Ppt on levels of organization Ppt on csr as per companies act 2013 Ppt on abraham lincoln biography Ppt on web designing Ppt on switching networks Ppt on grease lubrication pdf Ppt on art and craft movement style Download ppt on fundamental rights and duties of citizens