Presentation on theme: "Status of the beam diagnostics for the SPES project Conforto NicolaSPES Control System GroupMarzo 2010."— Presentation transcript:
Status of the beam diagnostics for the SPES project Conforto NicolaSPES Control System GroupMarzo 2010
Conforto NicolaSPES Control System GroupMarzo 2010 Final goal: - Beam diagnostic system based on grid profiler and faraday cup read-out - Realisation of emittance-meter Work done: - Setup of an EPICS database to get data from faraday cup and beam profiler, and to set their position too. - Configuration of ADC boards, Digital IO boards, Digital Output boards. - Configuration of Stepper-Motor boards. - Use of CSS interfaces to set and use the diagnostic devices. - Final Setup of the hardware
Conforto NicolaSPES Control System GroupMarzo 2010 Digital Output Board: -Allows to set output bits -Actual usage: sets the position of the Faraday Cup. Farday Cup interface: - FC IN FC OUT switch button - Actual value (1nA = 1mV) - 10s, 30s, 60s averages (compress record, updated every 10s) - Plot of the actual value and the 60s average
Conforto NicolaSPES Control System GroupMarzo 2010 Beam Profile: - SET BP IN SET BP OUT buttons - Motor status info (BP position) - Actual value, both grids
Conforto NicolaSPES Control System GroupMarzo 2010 Problems: - Noises when using F.C. with suppressor - B.P. multiplexers are slightly different from each other - B.P. hardware problems on the horizontal grid Next steps: - Set final PV name for the new Pvs - Emittance meter
Your consent to our cookies if you continue to use this website.