Download presentation

Presentation is loading. Please wait.

Published byClaire Sparks Modified over 2 years ago

1
Hybrid Extensive Air Shower Detector Array at the University of Puebla to Study Cosmic Rays O. MARTINEZ, H. SALAZAR, L. VILLASEÑOR * + Grupo de Estudiantes Facultad de Físico-Matemáticas, Benemérita Universidad Autónoma de Puebla, Apartado Postal 1364, Puebla, Pue., 72000, México *On leave of absence from Institute of Physics and Mathematics, University of Michoacan, Morelia, Mich., 58040, México EMA 05 Morelia July 18-22, 2005

10
EAS Array Area: 4000 m^2 10 Liquid Ssintillator Detectors (Bicron BC-517H) 4 Water Cherenkov Detectors PMT Electron tubes 9353 K PMT EMI 9030 A

11
2200m a.s.l., 800 g/cm2. Located at Campus Universidad Autonoma de Puebla Hybrid: Liquid Scintillator Detectors and water Cherenkov Detectors Energy range 10^14- 10^16 eV

12
DAQ System Trigger: Coincidence of 4 central detectors (40mx40m) NIM y CAMAC. Uso digital Osciloscopes as ADCs. Rate: 80 eventos/h

14
Monitoring Use CAMAC scalers to measure rates of single partícles on each detector. Day-night variations <10% /mean around 3%

15
Calibration

16
~74 pe

17
LabView based DAS

18
MPV of EM peak = 0.12 VEM i.e. around 29 MeV, i.e., dominated By knock-on + decay electrons

19
Stopping muon at 0.1 VEM Decay electron at 0.17 VEM = 41 MeV Crossing muon at 1 VEM Alarcón M. et al., NIM A 420 [1-2], 39-47 (1999).

20
Data Analysis Arrival direction sin sin = d/c(t 2 -t 1 )

21
Angular distribution inferred directly from the relative arrival times of shower front in good agreement with the literature: cos p sen

22
Data Analysis Lateral Distribution Functions Energy Determination EAS-TOP, Astrop. Phys, 10(1999)1-9 The shower core is located as the center of gravity.

23
N e, obtained for vertical showers. The fitted curve is I k (N e /N ek ) - , gives =2.44±0.13 which corresponds to a spectral index of the enerfy distributions of =2.6

24
Cherenkov Liquid Scint Muons deposit 240 MeV in 1.20m high water and only 26 MeV in 13 cm high liquid, while electrons deposit all of their energy i.e., around 10 MeV. Therefore for 10 Mev electrons we expect: Mu/EM=24 for Cherenkov Mu/EM=2.6 for Liq. Scint. Muon/EM Separation

25
Mass Composition Hybrid Array Solution:

26
Iterations Start with Ne=82,300 Nmu = 32700 E0 = 233 TeV Iterations End with Ne=68000 Nmu = 18200 E0 = 196 TeV

27
Mass Composition Non-Hybrid Array Do a three parameter fit to :

28
Mass Composition Non-Hybrid but Composite Array Two Identical types of Cherenkov Detectors one filled with 1.20 m of water and the other with 0.60 m, i.e., VEM C’ =0.5VEM C i.e., do independent fits of EM and muon to NKG and Greissen LDF, respectively, where:

29
Conclusions We have checked the stability and performed the calibration of the detectors. We have measured and analyzed the arrival direction of showers. We determine the energy of the primary by measuring the total number of charged particles obtaining by integration of the fitted LDF. Study of Muon/Electromagnetic ratio is underway:

Similar presentations

© 2017 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Pituitary gland anatomy and physiology ppt on cells Ppt on touch screen controlled robot Ppt on power electronics in ieee format Ppt on solar energy technology Ppt on csr as per companies act 2013 Ppt on odisha culture Ppt on different types of dresses in india Ppt on formal letter writing Ppt on job rotation example Ppt on health tourism in india