We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byGerald Hamersley
Modified about 1 year ago
© 2010, TSI Incorporated MicroPIV Particle Image Velocimetry System for Micron Resolution Flow Measurements
© 2010, TSI Incorporated Key Aspects Non-invasive global flow measurement technique Systems provide accurate measurement of fluid flow in micro fluidic environments Utilizes special concepts in –illumination, scattered light collection, seed particle concentration, optical design and data analysis to offer the most powerful and versatile system Worked with pioneers in MicroPIV to incorporate unique ideas into the system ( Meinhart, Adrian, Wereley, Santiago )
© 2010, TSI Incorporated Bio-MEMS applications Chemical and biological analysis – “lab-on-a-chip” for medical and defense applications Microfabricated needles for drug delivery Flow sorter (Caltech) Lab-on-a-chip, Caliper Caltech
© 2010, TSI Incorporated Aero-MEMS applications Micron scale supersonic nozzles (MIT) Remote surveillance aircraft (UCLA, Caltech) Micro Air Vehicle (MAV): –flapping-wing vehicle < 15 cm in all directions (UCLA) Micro-jet engine (Epstein et al., MIT) Microfabricated supersonic nozzle (MIT) MEMS-controlled remote aircraft (Caltech) 500 m
© 2010, TSI Incorporated Micro PIV Issues Reduce flare from walls by using fluorescent particles and –filter the illuminating wavelength Particles must be very small to fit in small measurement volume –Small particles have significant Brownian motion Volume illumination instead of light sheet. –Creates unwanted images from out-of-focus particles Reduce background glow – Implies shallow channels or low concentrations of particles Short times between pulses and short pulses, even in slow flows –Need pulsed lasers
© 2010, TSI Incorporated Details on Microflow Measurement Flow In Flow Out Flood Illumination Microscope Objective GlassPlate Flow Model Focal Plane Laser beam
© 2010, TSI Incorporated System for Microflow Measurements System for microflow measurements utilizes many of the same components as the standard PIV system Main components: Dual Nd:YAG laser at energy level of 15 mJ to 50 mJ P OWERVIEW Plus and PIVCAM series of cameras I NSIGHT 3G software for Data capture and analysis, with enhanced capabilities for microflow measurements Microscope optics
© 2010, TSI Incorporated Laser illumination system for Microflow Measurements Nd:YAG laser –15 mJ to 50 mJ energy level Laser light attenuator –used with the laser to provide more precise control of the intensity of illumination Diffusion optics –employed in the Model 660000 microscope to ensure uniform light intensity to the flow model Option to use laser light guide –to deliver the light to the microscope
© 2010, TSI Incorporated Cameras for MicroFlow Measurements Powerview Plus and PIVCAM family of cameras are applicable for microflows measurements Powerview Plus 2MP, 4MP, 11MP, HS-200 –with high pixel resolution and 12-bit dynamic range PIVCAM 14-10 –with 1360 by 1024 pixel resolution. 12-bit dynamics range
© 2010, TSI Incorporated INSIGHT 3G software for MicroFlow Measurements Image capture –With full pixel resolution or Area of Interest capture Image analysis schemes –including FFT, Hart correlation and Direct correlation as Plug-ins Image enhancement algorithms for microflows measurements including –the averaging of correlation functions, image averaging and image subtraction Different filter schemes –to improve signal-to noise level of correlation function
© 2010, TSI Incorporated Removing the background Image with backgroundBackground
© 2010, TSI Incorporated Image with background removed Flow around a bubble
© 2010, TSI Incorporated Average Correlation Algorithm Image A (t = t 0 ) Image B (t = t 0 + t ) Correlation R AB Image Sequence 1 2 3 N A1A1 B1B1 A2A2 B2B2 B3B3 BNBN ANAN A3A3 RA1B1RA1B1 R A N B N RA2B2RA2B2 RA3B3RA3B3 (Ensemble Ave.) Peak Search Courtesy: C. Meinhart
© 2010, TSI Incorporated Arithmetic Average Correlation RA1B1RA1B1 RA2B2RA2B2 RA3B3RA3B3 RANBNRANBN Courtesy: C. Meinhart
© 2010, TSI Incorporated Key Aspects Configuration based on unique ideas developed by –Carl Meinhart, Juan Santiago, Steve Wereley and Ron Adrian Special concepts in illumination, image capture, seeding and data analysis (Patented) PowerView cameras – cameras with mask designed for PIV I NSIGHT – unique tools designed in for PIV –Background removal, averaging, correlation averaging –Plug-ins (make the package always the most advanced) –Plug-in library
© 2010, TSI Incorporated Optical Systems for Microflow Measurements Two types of Optical systems are available to make measurements in microchannels of different dimensions System with the Model 660000 Microscope –Offers a complete microscope and is used to measure flows in channels with width down to even less than 100 µm! System using the Model 610048 special optics assembly –provides flexibility in system arrangement and is suitable for measuring flows in channels with width greater than 1000 µm
© 2010, TSI Incorporated Components of the assembly include –a relay lens, microscope traverse body, a long working distance microscopic objective, fluorescent filter and fluorescent particles The assembly offers –high flexibility in flow model arrangements. –It can be positioned either vertically or horizontally to accommodate the model configuration. This is especially attractive if the model is part of a larger assembly from which the model could not be separated Designed to make flow measurements for –Channel width larger than 1000 µm Model 610048 Special Optics Assembly
© 2010, TSI Incorporated Model 610048 Special Optics Assembly
© 2010, TSI Incorporated POWERVIEW 4MP Camera with Model 610048 Optics Assembly Model 610048 Special Optics Assembly
© 2010, TSI Incorporated Stereoscopic MicroPIV System
© 2010, TSI Incorporated System using the Model 660000 Microscope Model 660000 microscope is –an inverted microscope with all the accessories to be used as a stand-alone unit. –epi-fluorescent attachment, a double lamp housing with halogen lamp and a 10X microscope objective are part of the microscope Components to be used with the Microscope for PIV –Model 660004 laser light attenuator assembly –Model 660006 laser light guide assembly –Camera relay lens
© 2010, TSI Incorporated System using Model 660000 Microscope
© 2010, TSI Incorporated Microscope with epi- fluorescent optics Computer (Post processing) Laser Microscope Objective Filter Cube BeamExpander 12 bit CCD Camera (2048 x 2048 pixels) Focal Plane Exciter 532 nm Emitter Sample Cover Vector Plot Fluorescent Particles Pulsed green laser (532 nm) illuminates flow volume and excites particles. Particles fluoresce, emitting light of a different wavelength Emitted light image is captured by digital camera. Images are processed to produce vector flow plots.
© 2010, TSI Incorporated Microscope-based System System Capabilities Microscope with all the accessories to be operated as a stand-alone unit Inverted microscope configuration and side-mounted camera arrangement –offer the utmost stability for microflows measurements Upright Microscope configuration can also be used Epi-Fluorsecent attachment with filter cube –uses the same optical train for the illumination and particle emission collection Collection of fluorescent emission from particles –so that only the particle images are captured and not reflected light from the model
© 2010, TSI Incorporated MicroPIV System using Upright Microscope
© 2010, TSI Incorporated Microscope-based System System Capabilities Diffusion optics provide –sufficient and uniform laser intensity to the flow model Various microscope objective lenses –(from 4X to 40X) to allow imaging of flow in different model sizes Combination of the camera relay lenses and microscope objective lenses –provides the best illumination quality for various imaging areas Attachment for laser light guide –provides flexibility of laser light delivery from the laser
© 2010, TSI Incorporated Microscope-based System System Capabilities Can Switch between laser illumination and white light illumination –Facilitates alignment and focusing of the model Laser safety optics included in the system –Prevents viewing laser light by the operator Flow model holder –Accommodate models of varies sizes and shapes System is capable of measuring microflows in a wide range of channels (or channel widths) –Even within channels with width less than 100 µm
© 2010, TSI Incorporated Micro PIV example Santiago, et al. 1998 Hele-Shaw flow around a red blood cell. (Santiago, et al. 1998)
© 2010, TSI Incorporated Application of micro-PIV Inkjet meniscus visualization and micro PIV results (courtesy of Meinhart et al., 1999)
© 2010, TSI Incorporated Examples of MicroFlow Measurements Flow in a diverging micro-channel Image field Vector field
© 2010, TSI Incorporated Examples of Microflow
© 2010, TSI Incorporated Microchannel flow 150 m 500 m
© 2010, TSI Incorporated Kobayashi, Taniguchi Oshima Lab University of Tokyo Microchannel measurements
© 2010, TSI Incorporated Microflow Channels
© 2010, TSI Incorporated Flow around an obstacle
© 2010, TSI Incorporated Time Resolved PIV Systems.
Micro PIV An optical diagnostic technique for microfluidics (e.g. MEMS, biological tissues, inkjet printer head) Requirements: Measure instantaneously.
1 Imaging Techniques for Flow and Motion Measurement Lecture 3 Lichuan Gui University of Mississippi 2011 Particle Image Velocimetry (PIV) Introduction.
Measurements in Fluid Mechanics 058:180:001 (ME:5180:0001) Time & Location: 2:30P - 3:20P MWF 218 MLH Office Hours: 4:00P – 5:00P MWF 223B-5 HL Instructor:
TSI Incorporated Copyright© 2002 TSI Incorporated A Sample of PIV Applications.
Measurements in Fluid Mechanics 058:180 (ME:5180) Time & Location: 2:30P - 3:20P MWF 3315 SC Office Hours: 4:00P – 5:00P MWF 223B-5 HL Instructor: Lichuan.
© 2010, TSI Incorporated Global Sizing Velocimetry (GSV)
© 2009, TSI Incorporated Stereoscopic Particle Image Velocimetry.
Elementary Mechanics of Fluids Lab # 3 FLOW VISUALIZATION.
1 Imaging Techniques for Flow and Motion Measurement Lecture 15 Lichuan Gui University of Mississippi 2011 Multi-phase Flow PIV Techniques.
Week 10: Imaging Flow Around a Radio Controlled Race Car.
Laser Anemometry P M V Subbarao Professor Mechanical Engineering Department Creation of A Picture of Complex Turbulent Flows…..
Lyes KADEM, Ph.D; Eng Particle Image Velocimetry for Fluid Dynamics Measurements Laboratory for Cardiovascular Fluid Dynamics MIE.
1. Resolution (R) : separation of close objects, light wavelength NA, numerical aperture 2. Contrast : distinction of objects from background “light field”
Ben Falconer. Background A bit about me Ben Falconer Came to Warwick 2006 Computer and Information engineering MEng project Project based around PIV Current.
ABRF meeting 09 Light Microscopy Research Group. Why are there no standards? Imaging was largely an ultrastructure tool Digital imaging only common in.
Dr Gihan Gawish Hydrodynamic focusing is a technique used to provide more accurate results from flow cytometers or Coulter counters for determining the.
Particle Image Velocimetry Demo Outline (For reference) Topic NumberTopic NamePage Type 1Flow of PIVAnimated page.
Digital two photon microscopy for multiple fast signals acquisition Laboratory of Advanced Biological Spectroscopy (L.A.B.S.) University of Milan - Bicocca.
1 Imaging Techniques for Flow and Motion Measurement Lecture 5 Lichuan Gui University of Mississippi 2011 Imaging & Recording Techniques.
May Chuck DiMarzio, Northeastern University ECE-1466 Modern Optics Course Notes Part 7 Prof. Charles A. DiMarzio Northeastern University.
1 1 Introduction to Machine Vision Systems Professor Nicola Ferrier Room 3128, ECB
© 2010, TSI Incorporated Particle Image Velocimetry Fundamental Theory and Operation.
Practical aspects of fluorescence microscopy. Colour separation Sensitivity Resolution I want to improve …. What can I do?
FLOW CYTOMETRY Dr. MOHAMMED H SAIEMA LDAHR KAAU FACULTY OF APPLIED MEDICAL SCIENCES MEDICAL TECHNOLOGY DEPT. 2 ND YEAR MT INSTROMINTATION EXT
Microscopy Group 2 Cabatit, Mendoza, Ramos, Rodriguez, Tan.
© 2010, TSI Incorporated Volumetric 3-Component Velocimetry (V3V)
Ben Falconer Supervisors: Peter Bryanston-Cross, Brenda Timmerman.
QA for the Fibre Tracker 28 October 2004Brunel QA for fibre bundling Dr Paul Kyberd and Dr Peter Hobson School of Engineering & Design Brunel University,
Engineering Engineer -> μηχανικος Engineering ?-> μηχανικη ?? (College of Engineering -> ???) Engineers create: -design and build machines, structures.
3D Measurements by PIV PIV is 2D measurement 2 velocity components: out-of-plane velocity is lost; 2D plane: unable to get velocity in a 3D volume.
AFM-Raman and Tip Enhanced Raman studies of modern nanostructures Pavel Dorozhkin, Alexey Shchekin, Victor Bykov NT-MDT Co., Build. 167, Zelenograd Moscow,
1.Stable radiation source 2.Wavelength selector 3.Transparent sample holder: cells/curvettes made of suitable material (Table 7- 2) 4.Radiation detector.
NANO 230 Micro/Nano Characterization Scanning Electron Microscopes 1.
Aerodynamic measurement of high-speed base flows Background The separated wake at the base of cylindrical bodies presents a number of challenges to aerodynamic.
MICROMACHINING TECHNOLOGY. Introduction Micromachining is used to create parts of size in the order of meters. Materials on a micrometer-scale possess.
Biomedical Applications of Plasma Spectroscopy: A Preliminary Study Dr. Unnikrishnan V. K. Associate Professor Department of Atomic and Molecular Physics.
Advanced Biology Visualizing Cells. The Human Eye Resolution – The minimum distance two points can be apart and still be distinguished as two separate.
Science 10 – Unit C BIOLOGY Chapter 1 – The Microscope.
Forensic Instrumentation PART ONE Microscopes. A microscope is an optical instrument that uses a lense or combination of lenses to magnify and resolve.
1 ME 381R Fall Lecture 24: Micro-Nano Scale Thermal-Fluid Measurement Techniques Dr. Li Shi Department of Mechanical Engineering The University of Texas.
July 2008 Jonathan Sommer. Has two modes to view samples Optical mode (white light) and Fluorescence mode Depending on your sample either mode maybe.
Bioanalytik detect and identify Over 25 years experience in Imaging Benefits of NightOWL.
Presented to: International Aircraft Materials Fire Test Working Group By: Robert Ochs Date: Wednesday, October 21, 2009 Federal Aviation Administration.
Intelligent Vision Systems Image Geometry and Acquisition ENT 496 Ms. HEMA C.R. Lecture 2.
SCM 330 Ocean Discovery through Technology Area F GE.
Colloid Transport Project Project Advisors: Timothy R. Ginn, Professor, Department of Civil and Environmental Engineering, University of California Davis,
© 2017 SlidePlayer.com Inc. All rights reserved.