We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byCarissa Blackham
Modified about 1 year ago
A Perspective on Measurement Uncertainty Analysis for Commercial Aircraft Test Engines High-accuracy data is critical to vetting engine performance indices such as compressor efficiency and thrust specific fuel consumption. Environmental and engine operating conditions can significantly vary during a flight cycle consisting of takeoff, climb, cruise, descent, and landing – influencing the elemental systematic uncertainties of measurements. (Ref.  provides general definitions of error sources.) Full knowledge of the measurands and measuring systems is a prerequisite to estimating and mitigating the uncertainties of the measurements. The objectives are: 1) to identify and discuss subtle sources of errors for flight-test measurements and 2) demonstrate importance of implementing applicable engineering principles for realistic assessment of such sources of uncertainties. INTRODUCTION SELECT SOURCES OF ERRORS & EXAMPLES Thermal effects can influence zero-shift and operating output of a sensor such as pressure transducer. Acurate determination of sensor operating temperature is imperative for devices exhibiting a high sensitivity to temperature perturbation. Inadequate accounting of thermal interactions between the measurement system and its surroundings can compromise the data validity. Thermal Effects Physical Change Any unaccounted physical mutations that may occur within an instrument during a flight cycle can impact the calibration uncertainties. For instance, the calibration curve of a volumetric turbine flowmeter (Figure 1) may shift due to a variation in the blade tip clearance resulting from a radial displacement of the casing caused by changes in the fuel and ambient pressure / temperature. This is analogous to studies examining the efficiency losses associated with the blade tip clearances in gas turbines [2, 3]. (References on turbine flow meters: [4, 5, 6].) Example: casing of a turbine flowmeter is made of stainless steel – modulus of elasticity (E) ≈195 GPa and Poisson ratio (ν) ≈ 0.3; inside radius (r i ) = 20 mm; outside radius (r o ) = 30 mm; outside pressure (P o ) = 100 kPa, and fuel pressure (P i ) = 5000 kPa. Using the equation below, the inner radial displacement (∆r i ) is estimated to be 0.0012 ± 0.0001 mm. Although the estimated radial displacement may not have any discernible impact on the flow rate uncertainty, the underlying analysis exemplifies the importance of examining the theoretical hypotheses for realistic uncertainty assessment. The inner radial displacement is determined from : 2014 NCSL International Workshop and Symposium © 2014 United Technologies Corporation riri roro Blade Tip Clearance Fuel Figure 1. Schematic of a turbine flowmeter (not to scale). Certain elemental errors, such as hysteresis, are typically prescribed as a percentage of the sensor full span. Hence, congruity between the sensor capacity and the range of interest is important for error control. Optimizing the sensor range, however, may not be feasible in cases where the instrument (e.g., strain-gauge rotary torque sensor) is adhered to the test specimen (e.g., fan shaft). Example: For a torque sensor capacity of 80 kN.m with a systematic uncertainty of ± 0.10% F.S. (dominated by linearity / hysteresis), the torque uncertainty becomes: ± 0.08 kN.m at a 95% C.L. Figure 2 depicts an asymptotically decreasing trend of the relative uncertainty (in percentage of reading) with the increasing part load. For the arbitrary range of interest, 50% – 60% F.S., shown in Figure 2, the relative uncertainty is equal or greater than ± 0.17% of the reading, compared to ± 0.10% of the reading in the ideal scenario (red line). (Dependency of linearity / hysteresis errors on calibration range is also discussed in Ref. .) A pressure transducer, on the other hand, can be isolated (via valves) from the test object, allowing use of multiple devices with various sub- spans – leading to more accurate measurements during a flight cycle. Instrument Span Mismatch Figure 2. Results of an example for rotary torque measurements. (Note: Torque measurement errors due to variations in temperature- dependent material properties of test specimen are not included.)
Ali Jalalzadeh-Azar, Ph.D., PE Depending on the instrument, the response characteristics of a measuring device can vary with altitude as illustrated in the examples below. Pressure sensor: The time constant of a pressure sensor for the engine intake air is in part influenced by the velocity of the compression / expansion wave propagating through the pressure tubing. Treating air as a perfect gas, the wave velocity is approximated by that of sound (α) as a function of the specific heat ratio (γ), gas constant (R ), and absolute temperature (T ): The delay in the instrument response time due to the connecting line ≈ L / α, where L is the length of the conduit. When the ambient temperature decreases from 25 o C (298 K) at the takeoff to -20 o C (253 K) at a cruise altitude, the delay in the instrument response time will increase by about 8%, whose uncertainty can be estimated based on the uncertainties of the conduit length and the confined air temperature. In addition, any variation within the transducer during a flight cycle can potentially impact the sensor time constant, which is a function of the damping ratio and natural frequency . (The effective time constant of the entire measurement system can be analytically estimated based on historical test data .) Thermocouple : The response time is negatively corrolated with the convection heat transfer coefficient. Altitude-induced change in ambient pressure and temperature alters inlet air density, influencing the Reynolds number and, hence, the heat transfer coefficient. Similarly, the response behavior of gas-path thermocouples varies with the changing operating conditions in a flight cycle. Instrument Response Variation This study aimed to stress 1) the need for addressing the elemental errors that are not necessarily captured in the calibration data for the measurement devices used in flight tests and 2) the importance of applying engineering principles to help quantify such uncertainties. The error sources examined in this paper are : Thermal Effects, Instrument Span Mismatch, Physical Change, and Instrument Response Variation, of which the last two are more likely to be overlooked. Quantifying thermal effects can be challenging in cases where the effective operating temperature of the instrument is subject to complex installation and / or conceptual errors. Unavoidable Instrument Span Mismatch refers to a limitation of certain devices, such as rotary torque sensors, that are inseparable from the test specimen and, hence, do not allow a span specification commensurate with the operating range of interest. In contrast, pressure transducers are not subject to this constraint. Physical change within an instrument (e.g., volumetric turbine flowmeter) can potentially alter the effective calibration curve. This additional source of error can be a concern for both steady-state and transient modes in test flights. Instrument Response Variation is of particular importance for transient modes (e.g, climb, ascent, and descent). Such a variation was elaborated in conjunction with the temperature-dependent compression / expansion wave velocity affecting the response time of pressure transducers. CONCLUSIONS Acknowledgments : The author wishes to thank Douglas S. Breindel, John C. Breslin, James W. Dunn, and Al Krejmas of Pratt & Whitney, a United Technologies Company, for their support. References 1.H.W. Coleman and W.G. Steele, Jr., Experimentation and Uncertainty Analysis for Engineers, 3 rd ed. John Wiley & Sons, Inc, New York, NY, 2009. 2.P.J. Newton, G.D. Lock, S.K. Krishnababu, H.P. Hodson, W.N. Dawes, J. Hannis, and C. Whitney, Heat Transfer and Aerodynamics of Turbine Blade Tips in a Linear Cascade, Transaction of the ASME, Vol. 128, pp. 300-309, 2006. 3.T.C. Booth, P.R. Dodge, and H.K. Hepworth, Rotor Tip Leakage: Part 1: Basic Methodology, ASME J. Eng. Power, 104, pp. 154-161, 1982. 4.R. Kalivoda, Fundamentals of Liquid Turbine Meters, TP02001/Technical Paper 103B, FMC Technologies, 1998. 5.P.W. Stoltenkamp, Dynamics of turbine flow meters, Technische Universiteit Eindhoven. 2007. 6.Warshawsky, H.F. Hobart, and H.L. Minkin, Small Turbine-Type Flowmeters for Liquid Hydrogen, NASA Technical Memorandum, NASA TM X-52984, 1971. 7.A.P. Boresi, R.J. Schmidt, Advanced Mechanics of Materials, 6th ed, John Wiley & Sons, 2003. 8.G. Wegener, W. Nold, J. Andrae, and K. Molitor, K. Measurement Uncertainty of Rotating Torque Transducers When Used in Partial Load Ranges, XVIII IMEKO World Congress, Metrology for a Sustainable Development, September 17 – 22, Rio de Janeiro, Brazil. 2006. 9.E.O. Doebelin, Measurement Systems Application and Design, McGraw-Hill, New York, NY, 2004. 10.A.A. Jalalzadeh-Azar, Analytical Evaluation of Response Characteristics of Temperature and Pressure Measurement Systems for Gas Turbine Engines, Proceedings of NCSL International Workshop and Symposium. Nashville, TN, July 14-18, 2013, 2013. 2014 NCSL International Workshop and Symposium © 2014 United Technologies Corporation
It is very difficult to measure the small change in volume of the mercury. If the mercury had the shape of a sphere, the change in diameter would be very.
Chapter 4 ENERGY ANALYSIS OF CLOSED SYSTEMS Dr. Kagan ERYURUK Copyright © The McGraw-Hill Companies, Inc. Permission required for reproduction or display.
Lecture #9 Ehsan Roohi Sharif University of Technology Aerospace Engineering Department 1.
Chapter 12 Compressible Flow 12-1: Stagnation properties 12-2: 1-D isentropic flow 12-3: Isentropic flow through nozzles.
1 Temperature Measurement. 2 How to measure temperature Temperature can be measured by detecting changes in various temperature-dependent properties
Instrumentation Classification of Instruments : 1. Active / Passive Instruments Active : Here, transducer itself is the source of power. Eg : Displacement.
Convection in Flat Plate Boundary Layers P M V Subbarao Associate Professor Mechanical Engineering Department IIT Delhi A Universal Similarity Law ……
Chalmers University of Technology Elementary axial turbine theory –Velocity triangles –Degree of reaction –Blade loading coefficient, flow coefficient.
DESIGN OF CASCADE for AXIAL FLOW COMPRESSORS Arrangement of aerofoils for best Group Performance… P M V Subbarao Professor Mechanical Engineering Department.
Theory of Turbine Cascades P M V Subbarao Professor Mechanical Engineering Department Its Group Performance, What Matters.……
ME 322: Instrumentation Lecture 23 March 13, 2015 Professor Miles Greiner Transient TC response, Modeling, Expected and observed behaviors, Lab 9, Plot.
Chalmers University of Technology Lecture 5 Shaft power cycles –Cycle selection –Technology trends Aircraft engine performance –Thrust and propulsion efficiency.
LECTURER PROF.Dr. DEMIR BAYKA AUTOMOTIVE ENGINEERING LABORATORY I.
TURBO – SUPER CHARGERS Presented by Thomson george.
Chalmers University of Technology Discussion on design task 1 Elementary axial turbine theory –Velocity triangles –Degree of reaction –Blade loading coefficient,
1 Seventh Lecture Error Analysis Instrumentation and Product Testing.
Turbomachinery Design Considerations EGR 4347 Analysis and Design of Propulsion Systems.
AROMA 2.0 Structural Damping – Influence of Variable Normal Load on Friction Damping Mohammad Afzal, KTH Sound and Vibration MWL TURBO POWER.
CHAPTER 7 ENTROPY Lecture slides by Mehmet Kanoglu Copyright © The McGraw-Hill Education. Permission required for reproduction or display. Thermodynamics:
Experimental and numerical studies on the bonfire test of high- pressure hydrogen storage vessels Prof. Jinyang Zheng Institute of Process Equipment, Zhejiang.
L.J.INSTITUTE OF ENGINEERING AND TECHNOLOGY compressors Shah Aadishkumar Aileshbhai (enrollment no ) (div A roll no. 55) Guided by Mr.hemangsir.
MAE 4262: ROCKETS AND MISSION ANALYSIS Basics of Chemical Rocket Performance Mechanical and Aerospace Engineering Department Florida Institute of Technology.
CHAPTER 18 Transient Pressure Measurement. The difference between the pressure at the mouth of the Pitot tube and that in the manometer decreases exponentially.
Physical Fluid Dynamics by D. J. Tritton What is Fluid Dynamics? Fluid dynamics is the study of the aforementioned phenomenon. The purpose.
Chapter 9 Gas Power Systems. Learning Outcomes ►Perform air-standard analyses of internal combustion engines based on the Otto, Diesel, and dual cycles,
Propulsion: Axial Flow Compressor & Fan Aerospace Engineering, International School of Engineering (ISE) Academic year : (August – December,
Chapter 5 ENERGY ANALYSIS OF CLOSED SYSTEMS Copyright © 2012 The McGraw-Hill Companies, Inc. Permission required for reproduction or display. Fundamentals.
Introduction Hydrogen has been successfully used in industry for decades, but current safety codes and standards must be updated for the situations encountered.
Lecture 3 Mechanical Measurement and Instrumentation MECN 4600 Department of Mechanical Engineering Inter American University of Puerto Rico Bayamon Campus.
The Effect of Working Fluid Inventory on the Performance of Revolving Helically-Grooved Heat Pipes Presenter: Dr. Scott K. Thomas, Wright State University.
06-November-2013 Thermo-Mechanical Tests BE-RF-PM Review of the CLIC Two-Beam Module Program Thermo-Mechanical Tests L. Kortelainen, I. Kossyvakis, R.
BC L TC l VCVC s a B The cylinder volume at any crank angle is: Compression ratio: Cylinder volume when piston at TC (s=l+a) defined as the clearance.
Material Measurement Laboratory Cryogenic Engineering Conference :45 AM Single-phase ambient and cryogenic temperature heat transfer.
Lecture 7: Measurement Systems 1. Objectives Review the basic elements of measurements- instrumentation systems. List and recognize the static and dynamic.
1 Second Law of Thermodynamics - Entropy. 2 Introduction The second low often leads to expressions that involve inequalities.
Energy and the Environment Spring 2014 Instructor: Xiaodong Chu ： Office Tel.: Mobile:
1 Design of a Multi-Stage Compressor Motivation: Market research has shown the need for a low-cost turbojet with a take-off thrust of 12,000N. Preliminary.
WELCOME \. TURBOCHARGER AND SUPERCHARGER INTRODUCTION The power out put of an engine depends upon the amount of air inducted per unit time and the degree.
Fuel-Air Modeling of Brayton Cycle P M V Subbarao Professor Mechanical Engineering Department Exact Modeling of Cycle is a first step for Energy Conservation…..
General Formulation - A Turbojet Engine Relevant physical quantities: mechanical forces such as thrust, drag, lift; thermodynamic properties such as energy,
Chapter One Characteristics of Instrumentation بسم الله الرحمن الرحيم.
Date of download: 6/25/2016 Copyright © ASME. All rights reserved. From: The Effect of Gas Models on Compressor Efficiency Including Uncertainty J. Eng.
Mechanical Properties Lec Stress and strain: What are they and why are they used instead of load and deformation? 2. Elastic behavior: when loads.
© From J. G. Webster (ed.), Medical instrumentation: application and design. 3 rd ed. New York: John Wiley & Sons, Figure 1.1 Generalized instrumentation.
Advanced Thermodynamics Note 6 Applications of Thermodynamics to Flow Processes Lecturer: 郭修伯.
Shaft Power Cycles Ideal cycles Assumptions: Both compressor and turbine are isentropic (reversible adiabatic, η= 100 %) Pressure losses in both of combustion.
Calibration and Static Response Measurement systems and each of their individual components ‘respond’ to inputs by producing a unique output for a given.
Chapter 4 ENERGY ANALYSIS OF CLOSED SYSTEMS Mehmet Kanoglu Copyright © The McGraw-Hill Companies, Inc. Permission required for reproduction or display.
Union College Mechanical Engineering MER301 -Measurement System and Uncertainty Analysis LECTURE 16-17: Measurement System Analysis and Uncertainty Analysis-Homework.
Characteristics of Instruments P M V Subbarao Professor Mechanical Engineering Department A Step Towards Design of Instruments….
© 2017 SlidePlayer.com Inc. All rights reserved.