Download presentation

Presentation is loading. Please wait.

Published byXimena Messinger Modified over 2 years ago

1
ME 322: Instrumentation Lecture 33 April 14, 2014 Professor Miles Greiner

2
Announcements/Reminders This week: Lab 10 Vibrating Beam Sign up for 1.5-hour Lab 11 periods with your partner in lab Extra-Credit LabVIEW Workshop – Friday, April 18, 2014, 2-4 PM, Jot Travis Room 125D – Sign-up on WebCampus – If enough interest then we may offer a second session Noon-2 Help wanted (see me greiner@unr.edu)greiner@unr.edu – Summer: Help construct a convection heat transfer augmentation experiment – Spring 2015: ME 322r Lab Assistant

3
Lab 11 Unsteady Speed in a Karman Vortex Street Nomenclature – U = air speed (instead of V) – V CTA = Constant temperature anemometer voltage Two steps – Statically calibrate hot film CTA using a Pitot probe – Find frequency, f P with largest U RMS downstream from a cylinder of diameter D for a range of air speeds U Compare to expectations (St D = Df P /U = 0.2-0.21)

4
Setup D Tube P Static Total + - IPIP Variable Speed Blower Plexiglas Tube Pitot-Static Probe V C 3 in WC Barometer P ATM T ATM CTA myDAQ Cylinder V CTA

5
Before Experiment Construct VI (formula block) Measure P ATM, T ATM, and cylinder D Find and for air Air Viscosity from A.J. Wheeler and A. R. Ganji, Introduction to Engineering Experimentation, 2 nd Edition, Pearson Prentice Hall, 2004, p. 430.

6
Fig. 2 VI Block Diagram

7
Fig. 1 VI Front Panel

8
Calibrate CTA using Pitot Probe Cylinder Hot Film Probe Pitot Probe

9
Calibration Calculations

10
Hot Film System Calibration The fit equation V CTA 2 = aU 1/2 +b appears to be appropriate for these data. Using least squares the best values for the dimensional parameters are – a = 2.643 volts 2 s 1/2 /m 1/2 – b = 4.5742 volts 2

11
Standard Error of the Estimate x x x x x x x x V CTA 2

13
Unsteady Speed Downstream from a Cylinder Enter values of a and b in VI For each measurement use f S ~ 48,000 Hz, sampling time t T ~ 1 sec For each blower speed – Remove cylinder to measure average speed approaching cylinder U A – Return cylinder and measure unsteady speed Determine frequency f P with highest U RMS – Eyeball – Uncertainty in f P is larger of » Frequency resolution: ½(1/t T ) ~ 1/2 Hz, or » Eyeball range Repeat for ~5 different blower speeds UAUA Hot Film

14
Fig. 2 VI Block Diagram

15
Fig. 4 Spectral Content in Wake for Highest and Lowest Wind Speed (a) Lowest Speed (b) Highest Speed f p = 2600 Hz f p = 751 Hz

16
Dimensionless Frequency and Uncertainty

17
Comparison with Expectations Are the values you get for St within the expected range?

18
Demo Construct VI – Formula Block – Convert to Dynamic Data Perform calculations

Similar presentations

OK

ME 322: Instrumentation Lecture 36 April 20, 2015 Professor Miles Greiner Proportional Control.

ME 322: Instrumentation Lecture 36 April 20, 2015 Professor Miles Greiner Proportional Control.

© 2017 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Ppt on sales closing techniques Ppt on my dream project Ppt on paper tablet pc Ppt on seven wonders of the modern world Ppt on cross-site scripting vulnerabilities A ppt on loch ness monster video Ppt on introduction to object-oriented programming examples Ppt on food adulteration in india Ppt on life study of mathematicians Ppt on retail industry in india