Download presentation

Published byShannon Newman Modified over 2 years ago

1
**Pencil-Beam Redefinition Algorithm Robert Boyd, Ph.D.**

2
**Pencil Beam Algorithms**

central axis of broad beam (Z) X pixel bounding pencil beams (2x2 mm2 at isocenter) Y Z X-Y plane normal to beam axis (Z)

3
**Pencil Beam Redefinition**

DX Z Z+DZ X-Y planes are spaced 5 mm apart on Z axis

4
**PBRA Physics Primary electron transport only**

delta-rays not modeled Multiple Coulomb scattering approximated with a Gaussian distribution large-angle scattering not modeled Mean collisional energy loss only catastrophic energy losses not modeled

5
**PBRA Physics Approximations**

PBRA requires measured central-axis depth dose curve PBRA uses an energy-dependent correction factor C(E) to match calculated and measured central axis depth dose curve

6
**Polyenergetic Spectrum**

7
**PBRA Correction Factor C(E)**

Solid Line: Monoenergetic PBRA C(E) Dashed Line: Polyenergetic PBRA C(E)

8
Polyenergetic PBRA

9
**20-MeV Horizontal Bone Slab**

Varian Clinac 2100, 15x15-cm2 open applicator, 100 cm SSD

10
**20-MeV Horizontal Bone Slab**

Varian Clinac 2100, 15x15-cm2 open applicator, 100 cm SSD

11
**20-MeV Horizontal Air Slab**

Varian Clinac 2100, 15x15-cm2 open applicator, 100 cm SSD

12
20-MeV Vertical Air Slab Varian Clinac 2100, 15x15-cm2 open applicator, 100 cm SSD

13
**20-MeV Vertical Air Slab Off-axis profile at 4.5 cm depth**

Varian Clinac 2100, 15x15-cm2 open applicator, 102 cm SSD

14
20-MeV Nose Surface Varian Clinac 2100, 15x15-cm2 open applicator, 100 cm SSD

15
**9-MeV Nose Surface Off-axis profile at 1 cm depth**

Varian Clinac 2100, 15x15-cm2 open applicator, 100 cm SSD

16
**PBRA Evaluation with Measured Data Set - Results**

PBRA was not able to achieve 4% or 2 mm dose calculation accuracy for all data points

17
**Beam Modeling electron source SADvir custom beam collimation L0**

isocenter patient

18
**Dual-Source Beam Modeling**

primary electron source secondary electron source custom beam collimation isocenter patient

19
**Dual-Source Model - 100 cm SSD**

Varian Clinac 1800, 9 MeV, 6x6-cm2 open applicator

20
**Dual-Source Model - 110 cm SSD**

Varian Clinac 1800, 9 MeV, 6x6-cm2 open applicator

21
IMC - Transverse Plane Varian 2100, 16 MeV, 15x15-cm2 applicator, 105 cm SSD

22
IMC - Transverse Plane Varian 2100, 16 MeV, 15x15-cm2 applicator, 105 cm SSD

23
IMC - Sagittal Plane Varian 2100, 16 MeV, 15x15-cm2 applicator, 105 cm SSD

24
IMC - Sagittal Plane Varian 2100, 16 MeV, 15x15-cm2 applicator, 105 cm SSD

25
**Parotid Gland - Transverse View**

Varian 2100, 16 MeV, 15x15-cm2 applicator, 100 cm SSD

26
**Parotid Gland - Transverse View**

Varian 2100, 16 MeV, 15x15-cm2 applicator, 100 cm SSD

27
**Ethmoid Sinuses - Transverse Plane**

Varian 2100, 16 MeV, 10x10-cm2 applicator, 100 cm SSD

28
**Ethmoid Sinus - Transverse Plane**

Varian 2100, 16 MeV, 10x10-cm2 applicator, 100 cm SSD

29
**Ethmoid Sinus - Profile at Y = 13.0 cm**

Varian 2100, 16 MeV, 10x10-cm2 applicator, 100 cm SSD

30
**Clinical Evaluation - Results**

Accuracy criteria was not achieved for entire irradiated volume, albeit only a small volume (< 3.5%) had dose differences greater than 4% and greater than 2 mm DTA. PBRA showed good agreement with Monte Carlo in matching isodose lines. Better modeling of physics will improve the accuracy of PBRA-calculated dose.

31
**Custom Bolus / Skin Collimation**

32
**Custom Bolus / Skin Collimation**

33
**Custom Bolus / Skin Collimation**

34
**Custom Bolus / Skin Collimation**

35
Electron Arc Therapy

36
Skin Collimation

37
**Arc Therapy with Skin Collimation**

38
**Pencil-Beam Divergence Current PBRA**

virtual source distance is equal to distance to broad beam virtual source mathematics assume “parallel”point beams integration performed over projected area svir Dx Dz

39
**Pencil-Beam Divergence divPBRA**

virtual source distance is a pencil beam-specific parameter mathematics assume divergent point beams integration performed over normal pixel width svir Dx Dz

40
**Local Pencil-Beam Divergence**

41
**20-MeV Horizontal Air Slab**

Varian Clinac 2100, 15x15-cm2 open applicator, 100 cm SSD

42
**Pencil-Beam Divergence Results**

divPBRA was more accurate than PBRA for most data points divPBRA was not able to achieve 4% or 2 mm accuracy for all data points Calculation times were approximately 30% longer

43
Arc Beam Modeling

44
**Future Work Dosimetry studies using PBRA**

Tomotherapy vs. conventional electron therapy Field matching for chest wall treatments Electron arc therapy planning using divPBRA Realistic dose deposition kernels using Monte Carlo Automated custom bolus/skin collimation planning using PBRA Translating PBRA to commercial system

45
**Acknowledgements Kenneth Hogstrom, Ph.D. Almon Shiu, Ph.D.**

Dennis Leavitt, Ph.D. Mitch Price, M.S. Melinda Chi, M.S Paul Alderson, B.S.

Similar presentations

OK

MONTE CARLO BASED ADAPTIVE EPID DOSE KERNEL ACCOUNTING FOR DIFFERENT FIELD SIZE RESPONSES OF IMAGERS S. Wang, J. Gardner, J. Gordon W. Li, L. Clews, P.

MONTE CARLO BASED ADAPTIVE EPID DOSE KERNEL ACCOUNTING FOR DIFFERENT FIELD SIZE RESPONSES OF IMAGERS S. Wang, J. Gardner, J. Gordon W. Li, L. Clews, P.

© 2018 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Download ppt on indus valley civilization pictures Ppt on spiritual leadership inc Ppt on traffic light controller project Ppt on credit default swaps pricing Ppt on book review of fish tales Elements of one act play ppt on ipad Ppt on power sharing in india download movies Ppt on landing gears Free ppt on history of cricket Ppt on indian culture and heritage