Download presentation

Presentation is loading. Please wait.

Published byApril Liddle Modified over 3 years ago

1
Derivation of initial electron beam energy spectrum Janusz Harasimowicz Establishment for Nuclear Equipment http://www.zdaj.com

2
Electron beams Range of electrons depends on the energy. Percentage depth dose curve shape depends on contribution of particular energies of particles directed onto the water phantom surface.

3
Electron beam energy Electron beam mean energy at phantom surface can be derived, ex. IAEA TRS 381: E 0 [MeV] = 0.818 + 1.935 R J 50 + 0.040 (R J 50 ) 2 [cm] However, direct measurement of full energy distribution of the beam is difficult and accelerator time consuming. Cannot be performed in oncology department. Is it possible to get information about initial electron beam energy spectrum from the water phantom measurements?

4
Energy spectrum derivation Proposed method: PDD measurement. Simulation of dose distribution for mono- energetic beams (Monte Carlo method). Derivation of each simulated beam contribution to measured PDD curve shape (error backpropagation algorithm). Calculated weights of particular simulated PDD curves should give information about real energy spectrum distribution.

5
Measurements Measurements were performed for Coline 15 accelerator (Establishment for Nuclear Equipment ZdAJ, Poland) for ”12 MeV” (nominal) electron beam. To minimize energy losses, scattering foils and applicator were removed and jaws were set for 40 cm x 40 cm field. Water phantom RFA-300 (Scanditronix) and plane-parallel ionization chamber NACP-02 (Scanditronix) were used.

6
Monte Carlo BEAMnrc Monte Carlo code was used: Modified Coline 15 treatment head model. Radiation source: parallel circular electron beam with 2D Gaussian XY distribution (2 mm FWHM) directed onto the exit vacuum window. Water phantom at SSD=100 cm. Depth dose curves calculated for monoenergetic beams in energy range from 1 MeV to 15 MeV with 250 keV interval. Weighted sum of simulated PDDs was fitted to measured PDD curve (inverse Monte Carlo). Rogers DWO, Faddegon BA, Ding GX, Ma C-M, Wei J, Mackie TR. BEAM: A Monte Carlo code to simulate radiotherapy treatment units. Medical Physics 1995 22:503-524

7
Error backpropagation algorithm Used error backpropagation algorithm is based on minimization of difference between measured and simulated values as stated by equation: Q = 0.5 Σ i [m i – Σ j (w j c i j )] 2 m i → depth dose measured at the i-th point c i j → depth dose calculated at the i-th point for the j-th energy bin w j → weight of the j-th energy bin

8
Error backpropagation algorithm Q = 0.5 Σ i [m i – Σ j (w j c i j )] 2 δQ/δw j = Σ i [–c i j (m i – Σ j (w j c i j ))] Δw j = –η δQ/δw j Δw j = η Σ i [c i j (m i – Σ j (w j c i j ))]

9
Error backpropagation algorithm To speed up the fitting procedure, a „momentum term” was added: Δw j k+1 = –η δQ/δw j k + α Δw j k Without the „momentum term” With the „momentum term”

10
Fitting procedure Initialization Mean square error Q calc. Derivative δQ/δw calc.Q < tolerance ? Weights change Results YES NO

11
Results Difference <1% ~1 MeV Depth dose [%] Depth [mm] Relative contribution to PDD Energy [MeV]Depth [mm] Relative difference [%]

12
Results NOTICE NOTICE: Difference in PDDs <1% is not equal to method uncertainty! Derivation of fitted energy uncertainty for any complicated case is rather difficult. However, one can try to estimate uncertainty by analysing matched simulated beams to the known results (ex. for monoenergetic spectrum).

13
Uncertainty analysis Known spectrum Fitted spectra

14
Uncertainty analysis Energy [MeV] Relative contribution to PDD

15
Other results Source: Source: Deng J, Jiang SB, Pawlicki T, Li J, Ma C-M. Derivation of electron and photon energy spectra from electron beam central axis depth dose curves. Phys. Med. Biol. 46 (2001) Random creep algorithm used but more advanced method adapted: Four-source model for beam phase-space reconstruction. Separation of photons and electrons contribution to dose distribution. Would be interesting to check the method with error backpropagation algorithm.

16
Conclusions It seems to be possible to derive energy from depth dose measurements in water phantom. But with what accuracy? Derived energy spectrum of Coline 15 is rather wide. Possible explanations: Large energy intervals (250 keV). Energy slit filter (located in the deflection system) has not been optimized yet (and perhaps too wide energy spectrum is getting to the exit vacuum window). Wrong source definition. Set up and measurement errors. Further studies are needed.

17
Conclusions Scheme of Coline 15 energy slit filter (located in the deflection system)

18
Appendix: Build-up region Measured dose higher than it arises form MC simulations! Difference in the build-up region: 2-4%

19
Appendix: Build-up region Bruce Faddegon (UCSF Comprehensive Cancer Center, San Francisco) found similar differences of 2-4% in parallel- plate and simulated surface dose for 6-21 MeV electron beams delivered with a Siemens Primus accelerator with the jaws wide open and no applicator. Agreement was much better with diode measurements.

20
Appendix: Build-up region Source: Source: Faddegon B, Schreiber E, Ding X. Monte Carlo simulation of large electron fields. Phys. Med. Biol. 50 (2005)

Similar presentations

OK

1 Neutron Effective Dose calculation behind Concrete Shielding of Charge Particle Accelerators with Energy up to 100 MeV V. E Aleinikov, L. G. Beskrovnaja,

1 Neutron Effective Dose calculation behind Concrete Shielding of Charge Particle Accelerators with Energy up to 100 MeV V. E Aleinikov, L. G. Beskrovnaja,

© 2018 SlidePlayer.com Inc.

All rights reserved.

To ensure the functioning of the site, we use **cookies**. We share information about your activities on the site with our partners and Google partners: social networks and companies engaged in advertising and web analytics. For more information, see the Privacy Policy and Google Privacy & Terms.
Your consent to our cookies if you continue to use this website.

Ads by Google

Ppt on fiscal policy in india 2010 Ppt on effect of global warming on weathering Ppt on conservation of momentum lab Just in time ppt on production Ppt on endangered species of birds Ppt on speed control of dc motor using pwm Ppt on non agricultural activities in nigeria Ppt on lines in fashion design Ppt on information technology industry in india Ppt on polynomials for class 8