Presentation on theme: "STEMI Recognition Class"— Presentation transcript:
1 STEMI Recognition Class 12 and 15 Lead AcquisitionSTEMI Recognition Class
2 12 Lead Rapid Acquisition Module 1 Electrode LocationModule 2 Electrode PlacementModule 3 How to do a 15 LeadModule 4 DemonstrationsModule 5 Reducing ArtifactModule 6 Tips and Techniques
3 Module 1 Electrode Location Electrodes must be placed in the proper position to obtain an accurate 12 Lead ECG
4 Module 1 Electrode Location 12 leads obtained from 10 electrodes4 on the limbs6 on the chest
5 Module 1 Electrode Location Limb lead positioning is simple.The electrodes are placed off the torso, on the limbs. The most correct positioning is near the wrist and ankles.However, EMS generally places the electrodes on/near the torso to limit artifact while transporting.
6 Module 1 Electrode Location Chest leads have specific anatomic locationsV1, V2, V4 the rest are placed in relationship to these leads
18 Module 2 Electrode Placement Here’s another approach to locating the 4th intercostal space
19 Module 2 Electrode Placement Best view laterally
20 Module 2 Electrode Placement Locate the supersternal notch (1) at the top of the manubriumPalpate down appox 2” until you find the sternal angle, slide your finger laterally to the right, you’re finger is now on the 2nd ribPalpate down into the 2nd 3rd and 4th intercostal space1
21 Module 2 Electrode Placement After placing V2, palpated down to the 5th intercostal space midclavicular line and place V4
22 Module 2 Electrode Placement 15 Leads are simpleRemove V4 and move it 5th intercostal space, midclivicular on the Right side of Pt’s chestV1V2V3V4RV4
23 Module 2 Electrode Placement 15 Leads are simpleRemove V4 and move it 5th intercostal space, midclavicular on the Right side of Pt’s chest
55 Module 4 Reducing Artifact Unless you have a clear ECG to analyze, all you interruptive skills are of little use
56 Module 4 Reducing Artifact Stress labs obtain clear ECG’s while the Patient is running on a treadmillWe should be able to obtain a 12 lead while the Patient is laying still
57 Module 4 Reducing Artifact As the heart depolarizes, an electrode on the Pt’s skin picks up the electrical activity
58 Module 4 Reducing Artifact It can also pick up other electrical signals
59 Module 4 Reducing Artifact To reduce artifact, we have to increase the heart’s signal and reduce the other electrical activity
60 Module 4 Reducing Artifact Artifact Reduction Strategy:Helping the electrode gel to better penetrate the skin will increase the signal strength from the heart and reduce the signal strength from other sources.
67 Module 5 Tips and Techniques Supine is the proper position, if the Pt will tollerate
68 Module 5 Tips and Techniques When the Pt changes position, the heart moves within the chest. This can cause ECG changes similar to a misplaced electrode.
69 Module 5 Tips and Techniques Do your best to maintain the modesty of a female Pt
70 Module 5 Tips and Techniques You could try wide medical tape
71 Module 5 Tips and Techniques A folded blanket or towel may help hold the electrodes in place
72 Module 5 Tips and Techniques Strand each lead out individuallyWhen ECG cables are looped around IV lines, O2 tubing, BP cuff tubing, or dangling between squad bench and stretcher you will have more artifactMake sure the Pt isn’t twiddling the ECG cablesIf unable to lay supine for ECG, place them semi-fowlers and breathing normallyDo not allow Pt to prop themselves up by the arms or you will have muscle tremor artifact
73 Module 5 Tips and Techniques If the Pt is cold/shivering, cover with blanket or sheet prior to capturing the 12 Lead
74 Module 5 Tips and Techniques Same Pt as before, covered with a towel
75 Module 5 Tips and Techniques For some Pt’s obtaining a clear ECG will be difficult (e.g. respiratory distress Pt, sitting up)However, in most cases it is possible to a 12 Lead ECG with excellent, or at least acceptable data qualityIt just takes effort. A desire to obtain a clean 12 Lead and the knowledge to trouble shoot problems
76 Module 5 Tips and Techniques Myocardial Infarctions are not like broken bones, and therefore, ECG’s are not like X-rays.If you’re treating a Pt with a broken hip. That x-ray could be taken now, 10 min’s from now, an hour from now and what would you see? A broken hip.With MI the events in the coronary artery can be changing moment by moment. The ECG can be very dynamic as well.There is a value to obtaining repeat ECG’s when you suspect MI. Making a habit of doing early and repeat ECG’s will help you identify a STEMI that could easily be missed.
83 Module 5 Tips and Techniques This short course provides you with what you need to know in order to rapidly obtain a 12 Lead ECG that’s both clear and accurateJust as the case with ECG interpretation, acquisition also requires practice. After you’ve done this 20 or 30 times, you’ll become comfortable and confidentFIN