Download presentation

Presentation is loading. Please wait.

Published byDwayne Trindle Modified over 2 years ago

1
Dynamometry D. Gordon E. Robertson, PhD, FCSB Biomechanics Laboratory, School of Human Kinetics, University of Ottawa, Ottawa, Canada 1Biomechanics Laboratory, uOttawa

2
Dynamometry measurement of force, moment of force (torque) or power torque is a moment of force that acts through the longitudinal axis of an object (e.g., torque wrench, screw driver, engine) but is also used as another name for moment of force power is force times velocity (F.v) or moment of force times angular velocity (M Examples of power dynamometers are the KinCom, Cybex, home electrical meter 2Biomechanics Laboratory, uOttawa

3
Force Transducers devices for changing force into analog or digital signals suitable for recording or monitoring typically require power supply and output device types: –spring driven (tensiometry, bathroom scale) –strain gauge (most common) –linear variable differential transformer (LVDT) –Hall-effect (in some AMTI force platforms) –piezoelectric (usually in force platforms) Examples: cable tensiometer, KinCom, Cybex, Biodex, fish scale, force platform 3Biomechanics Laboratory, uOttawa

4
Tensiometer measures tension (non-directional force) in a cable, wire, tendon, etc. Biomechanics Laboratory, uOttawa4

5
Strain Gauge Force Transducers uses the linear relationship between strain (deformation, compression, tension) in materials to the applied force (stress) materials are selected that have relatively large elastic regions if material reaches plastic region it is permanently deformed and needs replacement 5Biomechanics Laboratory, uOttawa

6
Stress-Strain Measurements Instron 5567 (Neurotrauma Impact Science Laboratory, uOttawa) accurately measures stress and strain for a wide variety of materials Biomechanics Laboratory, uOttawa6

7
Strain Gauges can be uniaxial, biaxial, multiaxial require DC power supply (battery) can be wired singly, in pairs, or quartets can measure force, torque, or bending moment Biomechanics Laboratory, uOttawa7

8
Strain Link 8Biomechanics Laboratory, uOttawa

9
Strain Gauge Transducers 9Biomechanics Laboratory, uOttawa

10
Power Dynamometers potentiometer strain link lever arm 10Biomechanics Laboratory, uOttawa

11
Strain Gauge Lever CybexKinCom Biomechanics Laboratory, uOttawa11 use strain gauges to measure normal force moment is computed by multiplying by lever length

12
Bending Moment for Moment of Force 12Biomechanics Laboratory, uOttawa this knee brace was wired to measure bending moment it could therefore directly measure varus/valgus forces at the knee

13
Strain Gauge Force Transducers Advantages: –can measure static loads –inexpensive –can be built into wide variety of devices (pedals, oars, paddles, skates, seats, prostheses …) –portable Disadvantages: –need calibration –range is limited –easily damaged –temperature and pressure sensitive –crosstalk can affect signal (bending vs. tension, etc.) 13Biomechanics Laboratory, uOttawa

14
Force Platforms devices usually embedded in a laboratory walkway for measuring ground reaction forces Examples: Kistler, AMTI, Bertek Types: –strain gauge (AMTI, Bertek) –piezoelectric (Kistler) –Hall-effect (AMTI) Typically measure at least three components of ground reaction force (F x, F y, F z ) and can include centre of pressure (a x, a y ) and vertical (free) moment of force (M z ) 14Biomechanics Laboratory, uOttawa

15
Kistler Force Platforms standard in treadmill clear top portable 15Biomechanics Laboratory, uOttawa

16
Piezoelectric Force Platforms Advantages: –higher frequency response –more accurate –wide sensitivity range (1 N/V to 10 kN/V) Disadvantages: –electronics must be used to measure static forces, drift occurs during static measurements –expensive, cannot be custom-built –require 8 A/D channels 16Biomechanics Laboratory, uOttawa

17
AMTI Force Platforms small model standard model glass-top model 17Biomechanics Laboratory, uOttawa

18
Strain Gauge Force Platforms Advantages: –ability to measure static loads suitable for postural studies –inexpensive, can be custom-built –fewer A/D channels required (typically 6 vs. 8) Disadvantages: –typically fewer sensitivity settings –poorer frequency response –less accurate 18Biomechanics Laboratory, uOttawa

19
Equations for Computing Centres of Pressure centre of pressure locations are not measured directly Kistler: x = – (a[F x23 –F x14 ] – F x z) /F z y = (b[F y12 –F y34 ] – F y z) /F z AMTI:x = – (M y + F x z) /F z y = (M x – F x z) /F z Notice division by vertical force (F z ). This means centre of pressures can only be calculated when there is non-zero vertical force. Typically F z must be > 25 N. 19Biomechanics Laboratory, uOttawa

20
Impulse Force platforms can measure impulse during takeoffs and landings When the subject performs a jump from a static position, the takeoff velocity and displacement of the centre of gravity can be quantified Impulse = ≈ ( F ) t 20Biomechanics Laboratory, uOttawa

21
Takeoff Velocity To compute takeoff velocity divide the impulse by body mass For the vertical velocity, body weight must be subtracted v horizontal = Impulse horizontal / m v vertical = (Impulse vertical – W t ) / m where m is mass, W is body weight, and t is the duration of the impulse 21Biomechanics Laboratory, uOttawa

22
Centre of Gravity Displacement Displacement of the centre of gravity can also be quantified by double integrating the ground reaction forces. First divide the forces by mass then double integrate assuming the initial velocity is zero and the initial position is zero. Be sure to subtract body weight from vertical forces. Care must be taken to remove any “drift” from the force signals. 22Biomechanics Laboratory, uOttawa

23
Centre of Gravity Displacement s horizontal = s vertical = To compensate for drift (especially with Kistler force platforms) high-pass filtering is necessary. 23Biomechanics Laboratory, uOttawa

24
Squat Jump (BioProc2) Example of a vertical squat jump (starts in full squat) red is vertical force, cyan is AP force body weight line airborne phase 24Biomechanics Laboratory, uOttawa

25
Centre of Gravity (BioProc3) Squat depth was 1.39 cm Takeoff height was 79.6 cm Jump height was 28.3 cm 25Biomechanics Laboratory, uOttawa

Similar presentations

OK

Accelerating Wind Energy 1 A physical approach to monitor tower base fatigue loads using standard signals Thesis presentation Freark Koopman Committee:

Accelerating Wind Energy 1 A physical approach to monitor tower base fatigue loads using standard signals Thesis presentation Freark Koopman Committee:

© 2017 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Ppt on abo blood grouping principle Ppt on atm machine free download Ppt on eye oscillopsia Ppt on mars rover Ppt on leadership in organizations Ppt on principles of object-oriented programming php Ppt on light travels in a straight line Ppt on 3g technology in india Ppt on akbar's administration Ppt on online library management