Download presentation

Presentation is loading. Please wait.

Published byMariela Ridings Modified over 3 years ago

1
Pulse Techniques

2
Off-Resonance Effects Initial magnetization along z x-pulse ( = 0) On-resonance: M z -> -M y Off-resonance: phase

3
Off-Resonance Effects – 90º Pulse 90º hard pulse works well for < 1 Phase shift is approximately linear Effective flip angle increases

4
Off-Resonance Effects – 90º Pulse Assume: Close to resonance 90º pulse ( = /2) First-order approximation Phase angle is proportional to offset First-order phase correction

5
Off-Resonance Effects – 90º Pulse Initial magnetization state is irrelevant Result: z - x - z

6
Off-Resonance Effects - Compensation Not constant-time Subtract 2 90 / for each 90º pulse, or Add a short echo Constant-time No compensation needed if 90º pulses are identical Shaped 90º pulses?

7
Pulse calibration – 180º or 360º? 180º pulse: Poor off-resonance performance M xy ~ Linear phase twist Saturation in repeated experiments 360º pulse Better off-resonance performance < 0.1 1 Uniform phase Minimal saturation

8
Off-Resonance Effects – 180º Pulse Inversion: Poor off-resonance performance Refocusing: Orthogonal better than parallel Phase twist is the same Both work only for < 0.2 1

9
Composite Pulses Trains of rectangular pulses instead of a single pulse Off-resonance performance Tolerance to B 1 inhomogeneity 90º – historic 180º inversion – most common

10
90 x -180 y -90 x Composite Pulse 90 x -180 y -90 x replaces 180 x for inversion The middle 180º compensates for off- resonance behavior of the outer 90º pulses

11
x -2 y - x Composite Pulse 90 x -180 y -90 x has better broadband inversion performance < 1 90 x -225 y -90 x has smoother, but narrower profile < 0.7 1 x -2 y - x composite pulse is less sensitive to mis- calibration of 90º pulse Cannot be used for refocusing

12
Selective Pulses Requirements: Uniform excitation (inversion) profile Minimal perturbation outside the target frequency range Uniform resulting phase (for excitation) Short pulse length Low peak power Easy to implement shape

13
Soft Rectangular Pulses Find 1, which produces a null in the excitation (inversion) profile at a certain offset off-resonance

14
Shaped Pulses Approximated by a series of short rectangular pulses Every point i : Amplitude i Phase i

15
Examples - Gaussian Pulse Truncated at 1% of max amplitude Excitation profile - gaussian

16
Examples - Sinc Pulse One-lobe of the sinc function Needs less power for the same length Similar excitation profile

17
Examples - SEDUCE-1 Essentially a gaussian, smoothed at both ends Used mostly for decoupling

18
Examples - Q5 Gaussian Cascade ~300 s pulse used as 90º 13C excitation pulse in Bruker pulse sequences Better than a soft rectangular 90º pulse For I y -> I z application needs to be time-reversed

19
Q5 Gaussian Cascade – Excitation Profile

20
Examples - Q3 Gaussian Cascade ~200 s pulse used as 180º 13C refocusing pulse in Bruker pulse sequences

21
Q3 Gaussian Cascade – Refocusing Profile

22
Q3 Gaussian Cascade – Inversion Profile It also provides good inversion

23
Examples - E-BURP2 Used in L-optimized experiments

24
E-BURP2 – Excitation Profile

25
Phase-Modulated Pulses – Frequency shifting Linear phase ramp 0 controls the phase of the resulting pulse: 90º ( I z -> I y ): N = 0 90º ( I y -> I z ): 0 = 0 180º ( I y -> -I y ): N/2 = 0 180º ( I z -> -I z ): arbitrary

26
Phase-Modulated Pulses – Resolution Issues Time resolution More points = better shape approximation Minimal pulse delay Phase resolution More points = smaller phase steps Minimal phase step

27
Phase-Modulated Pulses – Multi-band excitation Cosine modulation 10 kHz Needs 2x more power

28
Adiabatic Pulses “Conventional” pulses: Stationary B eff orientation Magnitude may be variable Precession around B eff Adiabatic pulses: Non-stationary B eff orientation Magnetization follows B eff, while precessing around it

29
Adiabatic Pulses – Performance Conditions B eff magnitude should be changed gradually

30
Adiabatic Pulses – Performance Conditions A second rotating frame {x r, y r, z r } within the first rotating frame B eff magnitude should be tilted slowly

31
Adiabatic Pulses – Performance Conditions Adiabatic condition Adiabaticity factor

32
Adiabatic Pulses - Chirp 180 13C inversion on Bruker 60 kHz sweep 500 us 20% smoothing

33
Chirp – Inversion Profile Very broad inversion profile Low peak power ~50% of high power

34
Adiabatic Pulses – Tolerance B 1 Inhomogeneity Power 3 dB higher than calibrated Power 3 dB lower than calibrated

35
Adiabatic Pulses - Wurst Smoothed ramp-up and ramp-down

36
Adiabatic Pulses - sech

37
Pseudo Bloch-Siegert Shifts Assume: x-pulse ( = 0) Very far off-resonance Magnetization along y Third order approximation

38
Pseudo Bloch-Siegert Shifts The result is a z-rotation by an angle In the absence of 1 the rotation = p – the chemical shift evolution! Pulse phase and initial magnetization phase are irrelevant

39
Pseudo Bloch-Siegert Shifts An off-resonance pulse incurs a phase shift PBS It is proportional to time- averaged square of 1 Also valid for shaped pulses Continuous application of 1 leads to a frequency shift PBS SEDUCE decoupling during chemical shift evolution

40
PBS Phase Shifts - Compensation Small-phase adjustment of a 90º pulse or phase correction in the frequency domain Depends on whether there is chemical shift evolution Works only for a narrow range of resonances far off-resonance Partial compensation Used in BioPack

41
PBS Phase Shifts - Compensation Using a second identical pulse and an echo Perfect compensation over a broad spectral range Limited by the broadband performance of refocusing 180º pulse Default for Bruker sequences

42
PBS Phase Shifts - Compensation A second pulse with opposite offset Must be far off- resonance Imperfect compensation – first- order phase correction required Not widely used Cosine modulation Requires twice the power

43
PBS Frequency Shifts – SEDUCE decoupling Train of cosine modulated SEDUCE-1 shaped pulses Modulation frequency mod Signals of interest are near the carrier Decoupling is very far off resonance The result is a scaling of the spectrum by a factor Compression around the center

44
PBS Frequency Shifts – SEDUCE decoupling 600 MHz example 15 s hard 13C pulse 252 s SEDUCE-1 1 max = 2.1 kHz Scaling factor = 0.36 For squared shapes, relative to a rectangle of the same amplitude mod = 17.7 kHz For 80 ppm spectral width 0.2 ppm shift at edges Unsuitable for 15N-, 13C ali -, 13C aro -NOESY

Similar presentations

OK

3F4 Optimal Transmit and Receive Filtering Dr. I. J. Wassell.

3F4 Optimal Transmit and Receive Filtering Dr. I. J. Wassell.

© 2018 SlidePlayer.com Inc.

All rights reserved.

To make this website work, we log user data and share it with processors. To use this website, you must agree to our Privacy Policy, including cookie policy.

Ads by Google

Ppt on difference between mission and vision Ppt on programmable logic array design Ppt on microcontroller based traffic light controller Ppt on electric meter testing michigan Ppt on education system in india during british rule Maths ppt on binomial theorem Ppt on gujarati culture family Ppt on tens and ones for grade 1 Ppt on group life insurance Ppt on raving fans book