3 Measurement Of High DC Voltage Series Resistance MicrometerResistance Potential DividerGenerating VoltmeterSphere and Other Gaps
4 Sphere Gaps Applicatios: Arrangements: Voltage Measurement (Peak) - Peak values of voltages may be measured from 2 kV up to about 2500 kV by means of spheres.Arrangements:Vertically with lower sphere grounded (For Higher Voltages)Horizontally with both spheres connected to the source voltage or one sphere grounded (For Lower Voltages).
6 Fig C :Breakdown voltage characteristic of sphere gaps The arrangement is selected based on the relation between the peak voltage, determined by sparkover between the spheres, and the reading of a voltmeter on the primary or input side of the high-voltage source. This relation should be within 3% (IEC, 1973).Standard values of sphere diameter are 6.25, 12.5, 25, 50, 75, 100, 150, and 200 cm.The Clearance around the sphere gaps:Fig C :Breakdown voltage characteristic of sphere gaps
7 Sphere GapsThe effect of humidity is to increase the breakdown voltage of sphere gaps by up to 3%.Temperature and pressure, however, havea significant influenceo n breakdown voltage.Breakdown Voltage under normal atmospheric conditions is, Vs=kVn where k is a factor related to the relative air density (RAD) δ.The relation between the RAD(δ) and the correction factor k:Under impulse voltages, the voltage at which there is a 50% breakdown probability is recognized as the breakdown level.
8 Sphere Gaps Factors Influencing the Sparkover Voltage of Sphere Gaps Nearby earthed objects,Atmospheric conditions and humidity,Irradiation, andPolarity and rise time of voltage waveforms.The limits of accuracy are dependant on the ratio of the spacing d to the sphere diameter D, as follows:d < 0.5 D Accuracy = ± 3 %0.75 D > d > 0.5 D Accuracy = ± 5 %For accurate measurement purposes, gap distances in excess of 0.75D are not used
10 High Ohmic Series Resistance with Microammeter Resistance (R) :Constructed with large wire woundValue: Few hundreds of Mega ohms –Selected to give (1-10μA) for FSD.Voltage drop in each element is chosen to avoid surface flashovers and discharges (5kV/cm in air, 20kV/cm in oil is allowed)Provided with corona free terminals.Material: Carbon alloy with temperature coefficient of 10-4/oC .Resistance chain located in air tight oil filled PVC tube for 100kV operation with good temp stability.Mircoammeter – MC typeVoltage of source, V=IR
11 High Ohmic Series Resistance with Microammeter Impedance of the meter is few ohms. i.e, very less compared to R so the drop across the meter is negligible.Protection: Paper gap, Neon Glow tube, a zener diode with series resistance – Gives protection when R fails.Maximum voltage: 500kV with 0.2% accuracy.Limitations:Power dissipation & source loadingTemp effects & long time stabilityVoltage dependence of resistive elementsSensitivity to mechanical stresses
12 Resistance Potential Divider It uses electrostatic voltmeter or high impedance voltmeter.Can be placed near the test object which might not always be confined to one locationLet, V2-Voltage across R2Sudden voltage changes during transients due to:Switching operationFlashover of test objectsDamage due to stray capacitance across the elements & ground capacitanceTo avoid sudden changes in voltages, voltage controlling capacitors are connected across the elements
13 Resistance Potential Divider At high voltage ends, corona free termination is used to avoid unnecessary discharges.Accuracy:0.05% accuracy up to 100 kV0.1% accuracy up to 300 kV0.5% accuracy for 500 kV
14 Generating VoltmeterGenerating voltmeter: A variable capacitor electrostatic voltage generator.It generates current proportional to voltage under measurement.This arrangement provides loss free measurement of DC and AC voltagesIt is driven by synch. motor, so doesn’t observe power from the voltage measuring sourceThe high voltage electrode and the grounded electrode in fact constitute a capacitance system.The capacitance is a function of time as the area A varies with time and, therefore, the charge q(t) is given as,
15 Generating Voltmeter and, For d.c. Voltages, Hence If the capacitance C varies sinusoidally between the limits C0 and (C0 + Cm) thenC = C0 + Cm sin ωtand the current ‘i' is then given as, i(t) = im cos ω t , where im = VCmωHere ω is the angular frequency of variation of the capacitance.Generally the current is rectified and measured by a moving coil meterGenerating voltmeters can be used for a.c. voltage measurement also provided the angular frequency ω is the same or equal to half that of the voltage being measured.Above fig. shows the variations of C as a function of time together with a.c. voltage, the frequency of which is twice the frequency of C (t).
16 Generating Voltmeter Instantaneous value of current i(t) = Cm fvV(t) where fv = 1/Tv the frequency of voltage.Since fv = 2fc and fc =1/( 60/n) we obtain,I(t) = (n/30) CmV(t)Fig. shows a schematic diagram of a generating voltmeter which employs rotating vanes for variation of capacitanceHigh voltage electrode is connected to a disc electrode D3 which is kept at a fixed distance on the axis of the other low voltage electrodes D2, D1, and D0.The rotor D0 is driven at a suitable constant speed by a synchronous motor.Rotor vanes of D0 cause periodic change in capacitance between the insulated disc D2 and the high voltage electrode D3.Number and shape of vanes are so designed that a suitable variation of capacitance (sinusodial or linear) is achieved.The a.c. current is rectified and is measured using moving coil meters. If the current is small an amplifier may be used before the current is measured.
17 Generating Voltmeter Limitations: Generating voltmeters are linear scale instruments and applicable over a wide range of voltages.The sensitivity can be increased by increasing the area of the pick up electrode and by using amplifier circuitsAdvantages:Scale is linear and can be extrapolatedSource loading is practically zeroNo direct connection to the high voltage electrodeVery convenient instrument for electrostatic devicesLimitations:They require calibrationCareful construction is needed and is a cumbersome instrument requiring an auxiliary driveDisturbance in position and mounting of the electrodes make the calibration invalid.