Download presentation

Presentation is loading. Please wait.

Published byMiguel Larsen Modified over 3 years ago

1
Istituto di Fisica dello Spazio Interplanetario A gradiometric measurement of G XCIV Congresso Nazionale SIF Genova, 22-27 Settembre 2008 V. Iafolla, E. Fiorenza, C. Lefevre, S. Nozzoli, R. Peron, M. Persichini, A. Reale, F. Santoli Istituto di Fisica dello Spazio Interplanetario (IFSI/INAF), Roma, Italy

2
Istituto di Fisica dello Spazio Interplanetario Gundlach, J. H. and Merkowitz, S.M. Phys. Rev. Lett. 85, 2869-2872 (2000) Current knowledge of G with a relative uncertainty: Current experimental results for G are characterized by very small errors, but with big discrepancies among the various results: Gundlach et al. Last best measurement: 600 ppm However, the values obtained in the various experiments differ one from the other by more than 600 ppm

3
Istituto di Fisica dello Spazio Interplanetario Gundlach et al. The currently recommended value of G by CODATA is: with a relative uncertainty: The discrepancies in the results for G of the various experiments are a proof of the presence of significant systematic errors in the experiments, as well as of the difficulty in carrying out the experiment. This also stresses the importance of the use of different techniques in measuring G. The difficulties of the measurement of G are related with the smallness of the gravitational effects, to the presence of spurious effects (vibrational, thermal, electromagnetic), as well as to the implementation of the required metrological precision. Current knowledge of G

4
Istituto di Fisica dello Spazio Interplanetario Torsion balances The test bodies are attached at the opposite ends of a horizontal beam which is suspended with a wire from its midpoint. Any force acting differently on the test masses and perpendicular to the plane formed by the beam and the wire, gives rise to a non zero net torque which will twist the wire by an angle proportional to the differential force itself. Current knowledge of G

5
Istituto di Fisica dello Spazio Interplanetario Current knowledge of G

6
Istituto di Fisica dello Spazio Interplanetario Why measuring G ? (the locally measured gravitational constant G L ) fundamental constant of Nature G represents a fundamental constant of Nature both in Newton theory as well as in Einsteins Geometrodynamics conversion factor G enters in several mathematical expressions with other fundamental constants (G is a conversion factor …) can vary with time or not One very important problem in modern physics is to know whether the quantities known as constants in Nature (G, c, h, e, …) can vary with time or not spatial and a time dependency Indeed, nonstandard theories of gravity predicts both a spatial and a time dependency of G … No more to say about the importance of G measurements Current knowledge of G

7
Istituto di Fisica dello Spazio Interplanetario A characteristic of such gradiometer is the detection of the off–diagonal components of the gravity gradient tensor At the same time the gradiometer rejects the linear acceleration due to the vibrational noise, the so–called Common Mode Rejection (CMR) One–axis gravity gradiometer Al 5056 X Y Z A preliminary experimental work has been performed in order to measure the gravitational (constant) G using a one axis gravity gradiometer

8
Istituto di Fisica dello Spazio Interplanetario The test mass is connected to an external reference frame by means of two torsional arms The torsion axis is also the symmetry axis of the test mass The oscillator is sensitive to the rotations of the horizontal plane The torque produced by the gravity gradient is balanced by the torsional spring of the oscillator One–axis gravity gradiometer Al 5056 X Y Z

9
Istituto di Fisica dello Spazio Interplanetario One–axis gravity gradiometer Mgravitational torque signal deflection torsion elastic constant Imoment of inertia of the test mass coefficient of dissipation The oscillator works in the region where its transfer function is flat, below the resonance frequency f 0 : Al 5056 X Y Z

10
Istituto di Fisica dello Spazio Interplanetario One–axis gravity gradiometer The pick-up system The signal is detected by means of a capacitive transducer in a bridge configuration The bridge is driven by a voltage with high frequency f p trough a transformer The output signal at frequency f r = 10 KHz produced by the unbalancing of the bridge is detected as a modulation of the bias voltage Total torque noise of the oscillator

11
Istituto di Fisica dello Spazio Interplanetario Common mode rejection: CMR Usually, the rejection is quite difficult with different instruments: we need two equal transfer functions The rejection is very important because of the vertical seismic noise Indeed, the limit in the accuracy of measurement is given by the value of the seismic noise on the oscillator around the frequency of the gravitational signal to be detected Of course, the seismic noise that does apply a torque on the system is not rejected Al 5056 X Y Z

12
Istituto di Fisica dello Spazio Interplanetario In order to obtain the CMR we need to take the difference of the two measurements (time span of about 8 days) With the one–axis gradiometer the rejection is automatic and the CMR is valid both for the mechanical signal and the thermal one Common mode rejection: CMR A way to obtain CMR is to take two equal accelerometers (with the same transfer function) with their sensitive axis almost parallel to the horizontal plane Two equal accelerometers

13
Istituto di Fisica dello Spazio Interplanetario System parameters: experimental measurements… The main parameters to be measured in the laboratory are: 1.The system resonance frequency 2.The mechanical quality factor 3.The system transfer function (ambient conditions) 4.The electrical parameters 5.The transducer factor (calibration) 6.The seismic noise at the frequencies of interest of the gravitational signal

14
Istituto di Fisica dello Spazio Interplanetario System parameters: experimental measurements … The measurement has been performed under vacuum conditions in order to have a higher Q(=3), but low enough in order to excite the system with an electric signal The experimental value for the frequency is 1.01 Hz Q=3 Amplitude Phase Transfer function and resonance frequency

15
Istituto di Fisica dello Spazio Interplanetario At ambient pressure the oscillator is an overdamped system because of the gas between the capacitor plates We are interested to measure gravitational signals around 10 1 Hz, therefore a smaller Q is possible Again, the system has been excited with an electric signal Q=0.1 Amplitude Phase System parameters: experimental measurements … Transfer function in ambient conditions

16
Istituto di Fisica dello Spazio Interplanetario System parameters: experimental measurements… M ΔVΔV Calibration procedure for the transducer factor The response of the system is linear

17
Istituto di Fisica dello Spazio Interplanetario Measurement of the gravitational effect The rotor is made by suspending a system of two masses on a wire The one–axis gradiometer detector is placed inside a metallic box in order to provide electric isolation of the system The sources are two cylinder filled with fused Pb: 29 kg each mass The frequency of the gravitational signal is twice the rotation frequency of the rotor: in this way noise effects induced by the rotation of the system are avoided An electric motor produces the desired rotation Source masses: D = 18 cm H= 11.5 cm The experimental apparatus

18
Istituto di Fisica dello Spazio Interplanetario Measurement of the gravitational effect In order to maximize the gravitational effect, the two centers-of-mass of the gradiometer (in black) were placed as close as possible to the two sources (in red) Blue = gravitational signal Red = noise of the rotor Spectrum of the signal Signal The experimental apparatus

19
Istituto di Fisica dello Spazio Interplanetario Measurement of the gravitational effect Numerical evaluations with a finite elements method Experimental results Comparison of the experimental results with the theoretical (numerical) predictions Metrological errors: Masses Distances

20
Istituto di Fisica dello Spazio Interplanetario Measurement of the gravitational effect Electronic noise Rotor noise Rotor + signal noise Accuracy of the measurement At IFSI laboratory 5 hours integration time At Gran Sasso laboratory: lower differential noise

21
Istituto di Fisica dello Spazio Interplanetario Recent results New design for the gradiometer

22
Istituto di Fisica dello Spazio Interplanetario Recent results Resonance frequency

23
Istituto di Fisica dello Spazio Interplanetario Conclusions 1.The one–axis gradiometer is able to measure G with a fractional accuracy of about 10 4, i.e., 100 ppm 2.This accuracy is competitive with other measurements in the literature 3.However, such a result can be reached if and only if all the other parameters entering in the gravitational law are measured with the same accuracy: 3.1 Homogeneity and density of the masses, their shape 3.2 Center–of–mass positions 3.3 Distances 4.A laboratory with a very low seismic noise and stable temperature is necessary (possibly no variations at the gravitational signal frequency) 5.At first, because of its higher frequency, the one–axis gradiometer seems unfavourable when compared with torsion balances

24
Istituto di Fisica dello Spazio Interplanetario Conclusions 6.On the contrary this instrument has several advantages: 6.1The system is excited at a specific frequency in a frequency band where the seismic noise is very low 6.2The CMR is high, only differential noise will matter 6.3A capacitive bridge is used for the detection of the signal 6.4The system undergoes small displacements and it is not subjected to the hysteresis of the spring 6.5The system can be calibrated very easily 6.6With the use of electric fields we are able to lower its resonance frequency 6.7The system works in free fall conditions, therefore it can be used in space

Similar presentations

OK

Oscillations. Periodic Motion Periodic motion is motion of an object that regularly returns to a given position after a fixed time interval A special.

Oscillations. Periodic Motion Periodic motion is motion of an object that regularly returns to a given position after a fixed time interval A special.

© 2018 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Ppt on abstract art Ppt on financial statement of sole proprietorship Ppt on memory management in operating system Ppt on bluetooth hacking softwares Download ppt on indus valley civilization for kids Download ppt on oxidation and reduction definitions Ppt on company profile of hdfc bank Ppt on cloud computing benefits Ppt on natural numbers whole numbers Ppt on power sector reforms in india