Presentation on theme: "Shoulder Arthroplasty Daniel Penello Upper Extremity Rounds April 26, 2006."— Presentation transcript:
Shoulder Arthroplasty Daniel Penello Upper Extremity Rounds April 26, 2006
Lesions of the shoulder requiring arthroplasty are much less common than lesions involving the weight- bearing joints of the body, such as the hip and knee.
The Shoulder Greatest ROM No inherent bony stability Relies on soft tissues for stability Many injuries involve the soft tissues (rotator cuff, labrum) Little glenoid bone stock
Indications for Shoulder Arthroplasty Osteoarthritis Rheumatoid arthritis Rotator cuff tear arthropathy Avascular necrosis Post-traumatic arthritis Severe proximal humeral fractures
Hemiarthroplasty Total Shoulder Reverse Total Shoulder Arthroplasty Options
Surgical Approach Deltopectoral Coracoid
A little history French surgeon Pean inserted platinum and rubber components to replace a shoulder joint destroyed by tuberculosis Neer I, Vitallium Hemiarthroplasty prosthesis which resulted in pain relief and good function compared to previous options.
1974- Neer II Prosthesis. Modified Neer I to conform to a glenoid component. Courtesy of Smith & Nephew
1970s - constrained components were popular, but follow-up reports demonstrated high rates of loosening, particularly of the glenoid component.
1980s – Modular humeral components were developed, along with cementless glenoid fixation using polyethylene on a metal backing.
Cemented polyethylene versus uncemented metal- backed glenoid components in total shoulder arthroplasty: a prospective, double-blind, randomized study. Boileau PBoileau P, Avidor C, J Shoulder Elbow Surg Jul- Aug;11(4):351-9.Avidor C 40 Shoulders with 3 year follow up. Metal-backed – 2% radiolucent lines, 100% progressive, 25% loose in 3 years. Associated with shift and osteolysis. Cemented – 80% radiolucent lines, 25% progressive. None loose in 3 years.
Other Problems with Metal-Backed Glenoid Components Metal-backing increased the thickness of the component and often lead to over- stuffing of the joint. To avoid over-stuffing the joint, the polyethylene thickness had to be reduced, resulting in accelerated poly wear & failure Poly-metal disassociation occurred with unacceptable frequency.
Humeral Components CEMENTED PROX POROUS COATED FULLY POROUS COATED Good for osteopenic bone Lower risk of intra-operative fracture More stress- shielding Hard to revise Higher risk of intra-operative fracture Less stress- shielding Easier to revise Need good bone stock Higher risk intra-operative fracture More stress shielding Hard to revise
Cemented vs Press-fit Humeral Components Harris, Jobe and Dai reported less micro- motion with proximally-cemented stems. Fully cemented stems provide no additional benefit or stability over proximally- cemented stems. Sanchez-Sotelo reported a low rate of stem loosening regardless of fixation, but press- fit prostheses developed more radiolucent lines in the first 4 years.
The Need for Modularity F-H Offset B-C Head thickness D-E = 8mm Top of humeral head is higher than greater tuberosity
The Need for Modularity Reestablishing normal glenohumeral anatomic relationships is important to ensure optimal results. Iannotti JP; JBJS 74A 1992
Other Anatomic Variables to Consider Glenoid : 2° anteversion to 7° retroversion Humeral Head: 20° - 40° retroversion Axial CT of the glenohumeral joint is a valuable pre-op planning tool.
Contraindications to Shoulder Arthroplasty Active or recent shoulder joint infection Paralysis with complete loss of rotator cuff and deltoid function A neuropathic arthropathy Irreparable rotator cuff tear is a contraindication to glenoid resurfacing.
Osteoarthritis In addition to the universal features of osteoarthritic joints ( joint space narrowing, cyts, osteophytes …), the shoulder can also demonstrate Posterior glenoid erosion Flattening of the humeral head Enlargement of the humeral head Rotator cuff tears are uncommon in OA
Hemi vs Total Shoulder Easy procedure Short Operating time Less risk of instability Can be revised to TSA Less reliable pain relief Progressive Glenoid erosion may cause results to deteriorate over time Need concentric glenoid More consistent pain relief Better fulcrum for active motion Difficult procedure Longer OR time Poly wear can cause loosening of both components More Glenoid bone loss
Recommendation based on Experience Neer, 1998 When the articular surface of the glenoid is good, the results of hemiarthroplasty are similar to those of TSA. Wear on the glenoid has not been a problem if the articular surface was good at the time of surgery and glenohumeral motion was re-established
Recommendations based on Evidence Kirkley et al, pts, 3 surgeons (stratified) One year follow-up No significant difference in WOSI, ASES, DASH Constant Score or ROM. Trend towards better pain relief with TSA. 2 Hemi patients crossed over to TSA after 1 year follow-up.
Recommendations based on Evidence Gartsman, shoulders Average f/u of 35 months No difference in ASES or UCLA scores. Significantly better pain relief with TSA 3 pts crossed over to TSA by 35 months
A comparison of pain, strength, range of motion, and functional outcomes after hemiarthroplasty and total shoulder arthroplasty in patients with osteoarthritis of the shoulder. A systematic review and meta-analysis. Bryant D, Litchfield R; J Bone Joint Surg Am Sep;87(9): Included 4 RCTs Average 2 year follow-up. TSA resulted in significantly improved UCLA scores, pain relief and increased forward elevation (by 13°). This meta-analysis concluded that at 2 years of follow-p, TSA provided a better functional outcome, however the problems of glenoid component loosening in the TSA group and progressive glenoid erosion in the hemi group may affect the eventual long-term outcome. Longer follow-up is necessary
Recommendations based on Evidence The results of arthroplasty in osteoarthritis of the shoulder. Haines JF et al. J Bone Joint Surg Br Apr;88(4): Prospective study of 124 shoulder arthroplasties for OA (Hemi and TSA) Similar improvement in pain and function in both groups if rotator cuff was intact. Better results with Hemi if + rotator cuff tear Hemi Revision at mean of 1.5 years for glenoid pain TSA Revision at mean of 4.5 years for glenoid loosening
Technical Issues to Consider OA tends to result in posterior glenoid wear/erosion, which, if accepted, will lead to a retroverted glenoid component. Compensate by anterior reaming or placing the humeral component in LESS retroversion. Failure to do so will result in Posterior Instability
Rheumatoid Arthritis Cemented short-stemmed prosthesis Gill, Cofield et al recommend at least 60mm between the cement mantles of ipsilateral shoulder and elbow arthroplasties. If this cannot be achieved, join both cement mantles together.
Rheumatoid Arthritis Generally, TSA performed due to destruction of the glenoid articular surface by the disease. Glenoid erosion may require bone grafting, however, if glenoid is eroded to the level of the coracoid process, glenoid resurfacing is contraindicated
Rotator Cuff Arthropathy Described by Neer, Craig and Fukada in A distinct form of osteoarthritis associated with a massive chronic rotator cuff tear. Generally, rotator cuff tears occur in less than 10% of shoulders with OA
Rotator Cuff Arthropathy A function of the rotator cuff is to depress the humeral head and keep it centered on the glenoid fossa. Massive rotator cuff tears result in proximal migration of the humeral head. This is a contraindication to glenoid resurfacing as it results in eccentric (superior) glenoid loading and early component loosening.
Surgical Options Hemiarthroplasty with a large head Repair of rotator cuff and TSA Reverse TSA Clayton Spacer
Outcomes of Hemiarthroplasty Rockwood: 86% satisfactory results after 4 years Zuckerman: 93% adequate pain relief and 90% had improved function for ADLs. Sanches-Sotelo: 75% modest improvements in ROM and strength for ADLs. Good pain relief.
Outcomes of Hemiarthroplasty Field et al, and Sanchez-Sotelo reported that impaired deltoid function and previous subacromial decompression (loss of coracoacromial ligament) were significantly associated with clinical shoulder instability post hemiarthroplasty.
Reverse Total Shoulder Arthroplasty Lateralizes the centre of rotation and places the deltoid at a mechanical advantage. More inherent stability and prevents proximal migration of humeral head.
Outcomes of the Reverse Total Shoulder The Reverse Shoulder Prosthesis for glenohumeral arthritis associated with severe rotator cuff deficiency. A minimum two- year follow-up study of sixty patients. Frankle M, Siegel S, J Bone Joint Surg Am Aug;87(8): Average age = 70 Improved ASES scores Improved ROM Flex: ° Abd: ° 17% Complication rate 7 failures 5 revised to new Reverse TSA 2 revised to Hemiarthroplasties
Outcomes of the Reverse TSA (Delta III prosthesis) Treatment of painful pseudoparesis due to irreparable rotator cuff dysfunction with the Delta III reverse-ball- and-socket total shoulder prosthesis. Werner CM, Glbart M, J Bone Joint Surg Am Jul;87(7): consecutive patients, average age = cases were revisions Follow up = 38 months Improved Constant Score, Pain reduction and improved ROM. ROM: Flex: ° Abd: 43 90° 50% complication rate (including minor) If a 1° surgery = 18% re-operation rate If a Revision surgery= 39% re-operation rate
Reverse Total Shoulder Arthroplasty is Hard to Revise Little Glenoid bone stock once component is removed.
Osteonecrosis Usually young patients with adequate bone stock. Prefer proximally porous-coated, press-fit humeral prosthesis. less stress-shielding easier to revise if necessary Only resurface glenoid in stage V osteonecrosis (glenoid erosion).
Post-Traumatic Arthritis Due to fractures treated conservatively May have mal-union of tuberosities, distorting normal anatomic landmarks 12% of patients have axillary nerve palsies ( Neer ). Many have soft-tissue contractures and muscle weakness
Choice of Prosthesis Consider Patient age Condition of glenoid surface and bone stock Axillary nerve palsy is a relative contraindication to arthroplasty
Complications Instability 1.2% Excessive Retro/Anteversion Head too small Head too low (post fracture) Subscap rupture
Complications Rotator Cuff Tear 2% Results in superior migration of humerus and glenoid loosening Glenoid loosening
Complications Infection 0.5% Staph Aureus More common after revision surgery
Complications Heterotopic Ossification % Males Dx = osteoarthitis Low grade Non-progressive Does not affect outcome Sperling, Cofield et al
Complications Stiffness Depends on indication for arthroplasty Subscap shortening Oversized components Inappropriate rehab
Complications Periprosthetic Fracture Intra-op 1% Post-op % Most common in RA 85% women Glenoid fractures are rare
Complications Axillary nerve injury Rare Higher risk during revision surgery Usually a neuropraxia