Download presentation

Presentation is loading. Please wait.

Published byLucas Thomas Modified over 4 years ago

1
Cavity BPMs for Happex and G0 John Musson

2
Triplet Configuration…X, Y, and I TM 010 Mode for I TM 110 Mode for X & Y –Slugs provide proper excitation, reducing TM010 x-talk –Nominal output: 54nV-uA/um (-132 dBm)…per MAFIA simulations

3
Cavity Response ___________________ Courtesy Jürgen Schreiber, ECFA/DESY LC workshop, Amsterdam, April 1-4, 2003

5
I & Q Demodulation I Q DE- MO D I @ 28 Msps ADC 70 MHz LO 90 Degree I & Q 56 Msps ADC 0 2 3 1 + 1 COUNTE R RE G ADC70 MHz LO 56 Msps System Clock 14-Bit 2s Complement 0 2 3 1 + 1 RE G Q @ 28 Msps = +I+Q-I-Q+I+Q-I-Q+I+Q-I-Q+I+Q-I-Q+I+Q-I-Q+I+Q

6
Receiver

7
Functional Description

8
Receiver Parameters Noise Floor: -91 dBm (200 uV) –3690 uA-um per shot (per cavity simulation) Bandwidth = 100 kHz Processing Gain = 6MSPS/100 kHz = 18 dB Best-case Resolution at 50 uA ~ 9 um (at full BW) Additional Integration Gain (16ms): 32 dB DAC Output BW ~ 75 kHz (200 ksps) Calculated resolution for Eff = 50 %: 0.7 um –Hoping for 1 um

9
EPICS Interface

10
Happex Run BCM Crosstalk Helicity-correlated position differences, vs stripline,1nm Resolution BCM DD Glitches BCM Linearity Helicity-correlated position difference. xtalk Same plot, better data!

11
Happex Run Bad News….. Limited Dynamic Range –Required external amps and filters Crosstalk…~45 dB of C-C isolation –BCM signal would corrupt X & Y Software Problems –Register overflow resulted in glitching and Bedposts Synchronous Detection => Dedicated MO –Phase noise and distribution issues (LOL) Data COURTESY l. Kaufman, K. Paschke, R. Michaels

12
In Addition Setup is a learned behavior! –We devised a procedure, which proved to be more difficult than expected with actual beam. Hall personnel eagerly participated….. –More eyes –Technical understanding of benefits and limitations –Fantastic model for future systems

13
G0 Improvements Hardware –Crosstalk path identified. IF traps installed on Local Oscillator lines => > 60 dB –Amplifier removed from BCM (I) channel –Additional bench testing to understand Software –Register rollover identified, corrected, and tested. MO –Try asynchronous operation, due to large Phase Noise in Halls Hall personnel also system-savvy! –Data courtesy R. Suleiman

14
G0 Data

15
Non-Linear Behavior

16
Double Bounce at Zero X-ing

17
Glitching

18
Known Improvements 50 nA Sensitivity –Currently have 74 mm resolution! Need additional 37 dB to achieve 1 mm. LNA? –I-cavity is not a problem…plenty of signal Shore up all hardware fixes (ie. LO-IF traps) Firmware and Additional tests. Setup Procedure (EPICS) and All-Save Return to Synchronous MO –Must improve MO Distribution to Hall(s) Cavity Investigation on Downstream X&Y –How can we duplicate beam+cavity behavior in the lab? Thank you to all for data, feedback, and especially patience!

19
CORDIC Algorithm COordinate Rotation DIgital Computer –Jack E. Volder, The CORDIC Trigonometric Computing Technique, IRE Transactions on Electronic Computers, September 1959 –Ray Andraka, A Survey of CORDIC Algorithms for FPGA Based Computers, FPGA '98. Proceedings of the 1998 ACM/SIGDA sixth international symposium on Field programmable gate arrays, Feb. 22-24, 1998, Monterey, CA. pp191-200. Iterative method for determining magnitude and phase angle –Avoids multiplication and division N bits +1 clock cycles per sample Can also be used for vectoring and linear functions (eg. y = mx + b)

20
Concept Exploits the similarity between 45 o, 22.5 o, 11.125 o, etc. and Arctan of 0.5, 0.25, 0.125, etc. Multiplies are reduced to shift-and-add operations AngleTan ( )Nearest 2 -N Atan ( ) 451.0145 22.50.4140.526.6 11.250.1990.2514.04 5.6250.0950.1257.13 2.81250.0490.06253.58 1.4061250.02460.031251.79 0.7031250.01230.015630.90

21
Y X Binary search, linked to sgn(Y) Successively add angles to produce unique angle vector Resultant lies on X (real) axis Functionally..... with a residual gain of 1.6

Similar presentations

OK

FONT4 Status Report Glenn Christian John Adams Institute, Oxford for FONT collaboration.

FONT4 Status Report Glenn Christian John Adams Institute, Oxford for FONT collaboration.

© 2018 SlidePlayer.com Inc.

All rights reserved.

To ensure the functioning of the site, we use **cookies**. We share information about your activities on the site with our partners and Google partners: social networks and companies engaged in advertising and web analytics. For more information, see the Privacy Policy and Google Privacy & Terms.
Your consent to our cookies if you continue to use this website.

Ads by Google

Ppt on campus recruitment system online Ppt on any one mathematician jobs Ppt on fdi and fii in india Ppt on cross sectional analysis of financial statement information Ppt on body language management Ppt on limits and continuity of functions Ppt on operation research simplex method Ppt on testing of 3 phase induction motor Ppt on solid dielectrics glass Ppt on hydrogen fuel cell technology