Presentation on theme: "Www.ihe.net PCD Infusion Pump WG Containment / Mode / Status Discussion (rev. 2) Updated: 2012.04.23 Monday."— Presentation transcript:
www.ihe.net PCD Infusion Pump WG Containment / Mode / Status Discussion (rev. 2) Updated: 2012.04.23 Monday
www.ihe.net 2 DS-IP-xyz Containment/Mode/Status Discussion Scope: Review proposed containment models for : DS – Infusion Pump – LVP DS – Infusion Pump – Syringe DS – Infusion Pump – PCA Define role and values for pump Mode parameter Define purpose and values for pump Status parameter
www.ihe.net 3 DS-IP-LVP Containment Model VMDs are paired with technology-specific delivery channels Source Channels package semantics that define / drive the behavior of a Delivery Channel Multiple source channels can be active at any point in time.
www.ihe.net 4 DS-IP-Syringe Containment Model Syringe-specific parameters are contained in the Syringe Delivery Channel Source Channels are the same as those available for LVP Delivery Channels
www.ihe.net 5 DS-IP-LVP-PCA Model PCA specific parameters are packaged in a PCA Source Channel PCA Source Channel can be combined with other Source Channels as needed PCA Source Channel can be combined with any Delivery Channel (LVP or Syringe) PCA Source Channels may operate exclusively, sequentially or concurrently with one or more other Source Channels.
www.ihe.net 6 DS-IP-Syringe-PCA Containment Model Only difference is the choice of the Syringe Delivery Channel mechanism
www.ihe.net 7 Pump Mode & Status Parameters Discussion Points: Operational Mode & Status contained in delivery channels Mode reflects … what source channels are actively determining the operation of their associated delivery channel What the user would see as the configured mode All source channels OR only currently active source channels? Status reflects … primarily what is observed in the distal line … ? ISSUE #1: What do we want these values to be for DS-IP-xyz? Including active / concurrent source channel indication ISSUE #2: Harmonization with the 11073-10101:2004 Definitions
www.ihe.net 10 Proposed Resolution Define two Operational Mode parameters + 1 status parameter MDC_PUMP_MODE_SETTING (proposed) Indicates the configured infusion protocol Enumerations include: pump-mode-continuous / -piggyback / - pca / … Multiple mode bits can be set MDC_PUMP_MODE_CURRENT (proposed) Indicates the actual operational – including simultaneous – operational modes Enumerations - same as MDC_PUMP_MODE_SETTING MDC_PUMP_STAT (… can we make this _STATUS?) Indicates what the pumping mechanism is doing Does NOT include alarm status indication (separate parameter) Enumerations include: pump-status-ready / -infusing / -paused / -delayed / -standby / -off / …