Download presentation

Presentation is loading. Please wait.

Published byGabriel Donahue Modified over 4 years ago

1
X5/GIF Test Beam: High Rate Analysis Manuela Cirilli

2
Roma, 25 novembre 2002Manuela CirilliX5/2 Detectors * BIS module 0: * 170cm * 30 tubes X 4 layers X 2 ml * parallel gas distribution * Bundle: * 280cm * 6 tubes X 4 layers (1 layer disconnected from gas and HV due to a broken wire) * serial gas distribution (6 tubes in series) * BIS module 0: * 170cm * 30 tubes X 4 layers X 2 ml * parallel gas distribution * Bundle: * 280cm * 6 tubes X 4 layers (1 layer disconnected from gas and HV due to a broken wire) * serial gas distribution (6 tubes in series) BIS BUNDLE

3
Roma, 25 novembre 2002Manuela CirilliX5/3 Glossary * T drift = T max - T 0 * In the TDC spectra, T max and T 0 are in TDC counts * T drift is in ns * T drift is corrected for temperature effect according to H8 2001 results (-2.6 ns/T) * All T drift are @ 19°C * Source intensity is expressed with the absorption factor abs: * abs2 = 1/2 of the full source intensity * abs2 - abs5 corresponds to nominal Atlas flux (100 Hz/cm 2 ) * abs1 corresponds to Atlas X safety factor 5 * T drift = T max - T 0 * In the TDC spectra, T max and T 0 are in TDC counts * T drift is in ns * T drift is corrected for temperature effect according to H8 2001 results (-2.6 ns/T) * All T drift are @ 19°C * Source intensity is expressed with the absorption factor abs: * abs2 = 1/2 of the full source intensity * abs2 - abs5 corresponds to nominal Atlas flux (100 Hz/cm 2 ) * abs1 corresponds to Atlas X safety factor 5

4
Roma, 25 novembre 2002Manuela CirilliX5/4 Study of T max vs. Tube in the Bundle Initial strategy: Initial strategy: * Use only source off data taken in nominal conditions of gas pressure, HV, threshold * Combine source off runs only when there were no runs with the source on in between * Typical run: 500 M Evts * 5-6 tubes in a layer have reasonable statistics (>50K evts) @ high rate: * Lots of bkg on the TDC spectrum: clean it using BIS as an external tracker * At abs1, also distortion of the spectrum: unfolding Initial strategy: Initial strategy: * Use only source off data taken in nominal conditions of gas pressure, HV, threshold * Combine source off runs only when there were no runs with the source on in between * Typical run: 500 M Evts * 5-6 tubes in a layer have reasonable statistics (>50K evts) @ high rate: * Lots of bkg on the TDC spectrum: clean it using BIS as an external tracker * At abs1, also distortion of the spectrum: unfolding

5
Roma, 25 novembre 2002Manuela CirilliX5/5 New wrt Pavia * Different fitting procedure: * Fit the rising and falling edge separately (Fermi-Dirac + bkg) * All parameters of the Fermi-Dirac are left free in the fit * More on fitting in my next talk * Different fitting procedure: * Fit the rising and falling edge separately (Fermi-Dirac + bkg) * All parameters of the Fermi-Dirac are left free in the fit * More on fitting in my next talk T0 Tmax

6
Roma, 25 novembre 2002Manuela CirilliX5/6 Presented in Pavia Layer 1 Layer 3 Layer 4 Layer 2 * Comparison with BIS: * Use 6 tubes in the beam: m Bundle q BIS ML1 H BIS ML2 * ~40 ns difference between the two between the two chambers chambers * Bundle drift time not constant over the constant over the data taking data taking * Comparison with BIS: * Use 6 tubes in the beam: m Bundle q BIS ML1 H BIS ML2 * ~40 ns difference between the two between the two chambers chambers * Bundle drift time not constant over the constant over the data taking data taking Tube nb T drift (ns)

7
Roma, 25 novembre 2002Manuela CirilliX5/7 Hit rejection à la Claudio

8
Roma, 25 novembre 2002Manuela CirilliX5/8 Even better for out-of-beam tubes…

9
Roma, 25 novembre 2002Manuela CirilliX5/9 Typical T 0 and T max fits * T 0 : Fermi-Dirac + exp * T max : Fermi-Dirac + k

10
Roma, 25 novembre 2002Manuela CirilliX5/10 Problems: distortion of the TDC spectra * Already evident without background without background * * If the chambers are really at 0°, then the bias is only in the central part of the TDC spectrum * No bias on T 0 and T max determination T max determination * Already evident without background without background * * If the chambers are really at 0°, then the bias is only in the central part of the TDC spectrum * No bias on T 0 and T max determination T max determination clean spectrum raw spectrum

11
Roma, 25 novembre 2002Manuela CirilliX5/11 Bias on T 0 and T max Counts low-statistics tubes * Mean = 0.16 counts * RMS = 0.29 counts * Mean = 0.16 counts * RMS = 0.29 counts * Mean = -1.49 counts * RMS = 2.1 counts * Mean = -1.49 counts * RMS = 2.1 counts T0 Tmax

12
Roma, 25 novembre 2002Manuela CirilliX5/12 Special case: abs1 clean spectrum raw spectrum * The T max of the clean spectrum is shorter * The T max of the clean spectrum is shorter * But spectra at abs1 need unfolding… * But spectra at abs1 need unfolding…

13
Roma, 25 novembre 2002Manuela CirilliX5/13 Problem: loss of statistics * It is problematic for out-of-beam tubes (already low stat) * It could be overcome if we could combine different runs together… * It is problematic for out-of-beam tubes (already low stat) * It could be overcome if we could combine different runs together… Source off

14
Roma, 25 novembre 2002Manuela CirilliX5/14 Typical T 0 and T max fits at High Rate Abs 2

15
Roma, 25 novembre 2002Manuela CirilliX5/15 Results: abs 10 (10-50 Hz/cm 2 ) Clean spectra

16
Roma, 25 novembre 2002Manuela CirilliX5/16 Results: abs 5 Clean spectra

17
Roma, 25 novembre 2002Manuela CirilliX5/17 Results: abs 5 Clean spectra

18
Roma, 25 novembre 2002Manuela CirilliX5/18 Results: abs 5 Clean spectra Raw spectra

19
Roma, 25 novembre 2002Manuela CirilliX5/19 Results: abs 2 Clean spectra Runs 8-9 29/05/02 Runs 8-9 29/05/02

20
Roma, 25 novembre 2002Manuela CirilliX5/20 Results: abs 2 Clean spectra Run 8 31/05/02 Run 8 31/05/02

21
Roma, 25 novembre 2002Manuela CirilliX5/21 Results: abs 2 Raw spectra Clean spectra Run 6 29/05/02 Run 6 29/05/02

22
Roma, 25 novembre 2002Manuela CirilliX5/22 Results: abs 2, fitting errors Clean spectra Run 6 29/05/02 Run 6 29/05/02

23
Roma, 25 novembre 2002Manuela CirilliX5/23 Another way to look at the data Source off

24
Roma, 25 novembre 2002Manuela CirilliX5/24 Another way to look at the data Abs 2 Run 6 29/05/02 Run 6 29/05/02

25
Roma, 25 novembre 2002Manuela CirilliX5/25 Another way to look at the data Abs 2 Runs 8-9 29/05/02 Runs 8-9 29/05/02

26
Roma, 25 novembre 2002Manuela CirilliX5/26 Provocative: abs 1 Clean spectra

Similar presentations

Presentation is loading. Please wait....

OK

口算小能手.

口算小能手.

© 2019 SlidePlayer.com Inc.

All rights reserved.

To make this website work, we log user data and share it with processors. To use this website, you must agree to our Privacy Policy, including cookie policy.

Ads by Google