Presentation on theme: "General Approach of Haemostasis"— Presentation transcript:
1General Approach of Haemostasis Lecture 3:Coagulation Time of whole blood
2Coagulation Time ( Clotting Time) CT. Clotting Time is the time required for blood to form a clot in vitro.The basis for the test is that whole blood will form a solid clot when exposed to a foreign surface such as a glass tube.It is a rough measure of all intrinsic clotting factors in the absence of tissue factors. Variations are wide and the test sensitivity is limited.As the test is the least effective test in the diagnosis of actual haemostasis failure, it has been replaced by PTT.Clotting time was used as a screening test to measure all stages in the intrinsic coagulation system and to monitor heparin therapy .Clotting is the formation of a jelly like substance over the valves of the vessels.
3It is a complex process involving over 30 substances It is however, a time-consuming test, has poor reproducibility, is sensitive only extreme factor deficiencies. It is therefore, of limited use in today’s laboratory .It is a complex process involving over 30 substancesConditions accompanied by increased Clotting Time:Factors V, VII, VIII, IX, XI, XII Deficiencies.Hemorrhagic disease of NewbornVitamin K deficiency.Heparin Therapy.Presence of Circulating antibodies (inhibitors)Anemia and leukemia.Afibinogenemia and Pneumonia.and heparin.Reproducibility is the ability of an experiment or study to be accurately reproduced, or replicated, by someone else working independently. It is one of the main principles of the scientific method.Heparin: Works by markedly enhancing activity of antithrombin, which inhibits activated factors II, IX, X, XI, XII, kallikrein and probably VII, but doesn’t cause a true decrease in factor levelsVitamin K: prothrombin (factor II), factors VII, IX, X, protein C, protein S, and protein Z
5Tube Method (Lee-White method) Reagent & equipmentWater bath, 37O C.Glass test tube (10 x 75 mm)Stopwatch.Plastic syringe and 20-gauge needle.SpecimenFresh whole blood , 4 ml .Etymology: Roger I. Lee, American physician, b. 1881; Paul D. White, American physician, ; Gk, meta, beyond, hodos, waya method of determining the length of time required for a clot to form in a test tube of venous blood. It is not specific for any coagulation disorder but is often used to monitor coagulation during heparin therapy. Because normal values and precise methodology vary, instructions are provided by most laboratories.
6ProcedureLabel 3 glass test tube with patient name and number them, 1, 2, and 3.Perform a clean, Untraumatic venipucture using a 20-gauge needle and drawn 4 mL of blood.Remove the needle from the syringe, and fill each of the three tubes with 1 ml blood.The last 1 ml of blood may be discarded.Start the stopwatch as soon as the blood enters the syringe.Place the three test tubes in a 37°C water bath.At exactly 3 min., Remove the first tube form water bath and tilt gently to a 45° angle to see whether the blood has clotted.If Blood not clotted return it to the water bath and examine it at 30 second intervals.
7Normal Range: 5 – 12 Minutes*** After the blood in the first tube has clotted, examine the second tube immediately.Then examine the 3rd one.Record the time it took the blood in the 3rd test tube to clot.Normal Range: 5 – 12 Minutes***Since agitation and handling speed up coagulation, the clotting time test tube #3 is the reported result.
8DiscussionPoor venipucture technique, causing hemolysis or tissue thromboplastin to mix with the blood, shortens the clotting time.Bubbles entering the syringe when the blood sample is being obtained increase the rate of coagulation.Always tilt the tube in the same direction and at the same angle so that the blood is moving in the same pathway up the side of the tube each time.Excessive agitation of blood (occur during the transfer of the blood from the syringe to the test tube). The blood should be allowed to flow gently down the inside of the test tube and not forcefully squirted into the test tube.
9Dirty Test Tubes will affect the result. At the completion of the clotting time, one tube should remain in the 37°C water bath to be checked for clot retraction. Also, this same tube may be allowed to remain in the water bath overnight and checked the next day for clot lysis.Dirty Test Tubes will affect the result.Coagulation will be retarded by the following:Temperature Below 35O C.Temperature above 45O C.Diameter of the test tube (the smaller the diameter, the more rapid the clot formation is).
10ACT (Activated clotting Time) Fresh whole blood is added to a tube containing negatively charged particles and timed for the formation of a clot.The type of negatively charged particle affects the "normal" length of ACTCelite = Diatomaceous Earth: normal is secondsKaolin: normal is secondsGlass particles: normal secondsType of machine affects normal and therapeutic valuesDaily calibration checks are imperativeeTherapeutic Range: seconds.Diatomaceous Earth also known as diatomite or kieselgur, is a naturally occurring, soft, siliceous sedimentary rock that is easily crumbled into a fine white to off-white powder. The typical chemical composition of oven dried diatomaceous earth is 80 to 90% silica, with 2 to 4% alumina (attributed mostly to clay minerals) and 0.5 to 2% iron oxide.Kaolin It is a layered silicate mineral, with one tetrahedral sheet linked through oxygen atoms to one octahedral sheet of alumina octahedra. Rocks that are rich in kaolinite are known as china clay, white clay, or kaolin.
11Hemochron® Tests measured Heparin monitoring with or without Aprotinin.Sample and volume requirementsFresh whole blood- 2mL. Elite: 200µL to fill cuvette well.ControlsElectronic QCLiquid quality control- normal and abnormalReference RangesNormal range- healthy donorsRFTCA 510 tubes: 105 – 167 secondsFTK-ACT tubes: 91 – 151 secondsNormal range- cardiopulmonary bypass patientsHRFTCA 510 tubes: secondsFTK-ACT tubes: secondsHemochron : A warmed test tube is rotated inside the machine. As the blood clots, it displaces the magnet within the test tube. The clotting time is determined when the magnet has displaced enough to activate a proximity switchMedtronic HemoTec: A mechanical plunger is dipped in and out of a kaolin activated blood sample. The machine optically senses the time it takes the plunger to move through the specimen. Clotting is defined by the "drop time" threshold for the plungerThe machines do not correlate with one another but in general the times for the Hemochron are 30% greater than the Medtronic HemoTec
13Clotting Time - Capillary method MaterialSterile disposable pricking needle or lancet.Stop watchDry glass capillary tube (narrow diameter top 2 mm, minimum 10 cm long.)Cotton Swab of absorbent cotton.Spirit wetted, cotton swab.70 % v/v ethyl alcoholCapillary action is the movement of a liquid into a tube based on its surface tension.
14PROCEDUREApply alcoholic 70 % v/v to the clean finger with cotton swab. Allow it to dry naturally.Prick the finger with usual aseptic precautions. Remove the first drop.Dip one end of the capillary into blood drop gently without pressure. (use 3-4 capillary tubes)The Timer is started when the first blood start to enter the first capillary tube.Allow to fill the capillary with blood by lowering the end of fitted capillary. (Do not suck the blood) around ¾th of its length undipped.After every 30 seconds, using stopwatch, break a small piece of capillary.Repeat breaking at regular time intervals, till fibrin thread appears at the broken end of capillary tube. Do not pull away the cut pieces ling apart and bristly.Record time interval between pricking finger and first appearance of fibrin thread at the broken ends of capillary tube. That is clotting time of blood.Don't forget to dispose of the broken tube in the SHARPS CONTAINER.
16Clotting Time - Slide Method The surface of the glass tube initiates the clotting process. This test is sensitive to the factors involved in the intrinsic pathwayThe expected range for clotting time is 4-10 min.
17Clot retraction (Obsolete test) This test measures the amount of time it takes for a blood clot to pull away from the walls of a test tube (Shrinking).The edges of the blood vessel wall at the point of injury are slowly brought together again to repair the damage.It is used to evaluate and manage blood platelet disorders, including Glanzmann's thrombastheniaSo Clot retraction depends primarily on the number and activity of the blood platelets.This test is reliable only when the concentration of fibrinogen within normal limits.Normally a blood clot will begin to retract from the walls of the tube and express serum within one hour.
18Glanzmann's thrombasthenia Is an abnormality of the plateletsIt is an extremely rare coagulopathy in which the platelets lack glycoprotein IIb/IIIa.Reduced glycoprotein IIb/IIIa causes reduced platelet aggregation and clot retraction - no fibrinogen bridging can occur, and bleeding time is significantly prolonged);Glycoprotein IIb/IIIa is an aggregation receptor expressed by thrombocytes. This receptor is activated when the thrombocyte is stimulated by ADP, epinephrine, collagen and thrombin. The GpIIb/IIIa integrin is essential to the blood coagulation since it has the ability to bind fibrinogen, fibronectin and vitronectin. Glycoprotein Ib enables the platelet to be activated by contact with the collagen-von Willebrand-complex that is exposed when the endothelial blood vessel lining is damaged and then aggregate with other thrombocytes via fibrinogen.Glanzmann's thrombasthenia can be inherited in an autosomal recessive manner or acquired as an autoimmune disorder.
19Clot retraction testThis is a measure of platelet function. The test is done in blood to which no anticoagulant has been added and is allowed to clot. Clot retraction is then looked for.Clot retraction becomes abnormal in conditions likeFibrinogen deficiency (congenital or acquired)Thrombocytopenia < 1oooooThrombostheniaPolycythaemiaReduced clot formation: Glanzmann thrombasthenia , DIC, hypofibrinogenemia, dysfibrinogenemia (small clot with increased red blood cell “fall-out”)Disseminated intravascular coagulation (DIC) is a serious disorder in which the proteins that control blood clotting become abnormally active.
20ProcedureObtain 2 ml of blood by venipucture. Place one ml in a glass test tube in a 37oC water bath.Inspect clot at 1, 2, 4 and 24 hours after clotting. Observe for retraction, quality, and lysis of the clot.Remaining 40-60% consists of serum and red blood cell “fall-out” from clotA visual examination of the clot is made. Usually the clot retracts and expresses serum within two hours.
21Interpretation of results The clot will retract from the walls of the tube until the red cell mass occupies 50% of the total volume of blood in the tubeThere is a variable degree of retraction or there is no retraction at all. Grades as follows:0 : no serum extruded1+ : 5-10% serum extruded2+ : 10-20% serum extruded3+ : 20-35% serum extruded4+ : 35-50% serum extrudedResults are reported as the length of time it took for the clot to retract2-4 hrs is reported as normalAfter 4 hrs is reported as poorAfter 24 hrs is reported as no retraction
22Clot lysis testThe whole blood clot lysis time is used to detect increase fibrinolysisThis test is only able to detect high increase in fibrinolytic activityClot lysis testing is a measure of the presence of the clot after 24 hours (blood fibrinolysis).
23ProcedureAt the completion of the clotting time one tube should remain in the 37 water bathA second tube is placed in the refrigerator as soon as clotted as a controlThe tube is test for the disappearance of clot after 4, 8 and 24 hrsIf the sample becomes fluid in less than 48 hrs , the blood is poured out onto a piece of filter paper to be sure for clot disappearance
24InterpretationThe disappearance of clot before 48 hrs means increase fibrinolysisAfter 48 hrs clot may beCommentRefrigerated tube37 C tubeLow fibrinogenDisappearAbnormal fibrinolysis(No clot lysis after 48 hrs)Still intactNomal fibrinolysis
25Euglobulin Clot Lysis Time ActivityWhat is the test mentioned below and how w we can do it?.Euglobulin Clot Lysis Time