Download presentation

Presentation is loading. Please wait.

Published byEzra Fowler Modified over 3 years ago

2
“Finite Element Modeling of Radiofrequency Cardiac and Hepatic Ablation” SUPAN TUNGJITKUSOLMUN “Finite Element Modeling of Radiofrequency Cardiac and Hepatic Ablation” SUPAN TUNGJITKUSOLMUN Dept. Of Electrical and Computer Engineering University of Wisconsin-Madison Advisor: Professor John G. Webster

3
Goal Use Finite Element Modeling (FEM) to Improve the Efficacy of Current RF Ablation Technologies and to Design New Electrodes

4
Introduction: RF ablation & FEM Overview: Finite element modeling process 1. Effects of changes in myocardial properties 2. Needle electrode creates deep lesions 3. Uniform current density electrodes 4. Bipolar phase-shifted multielectrode catheter 5. Use FEM to predict lesion dimensions 6. FEM of hepatic ablation Outline

5
95% success rate in curing Supraventricular tachycardias Low success rate for hepatic ablation Development for VT (Large lesions) Development for AFIB (long thin lesions) Introduction What Is Ablation? Modes of operation ~500 kHz, < 50 W Temperature-controlled Power-controlled Present Technology Heating of cardiac tissue to cure rhythm disturbances and of liver tissue to cure cancer What Is Ablation? Modes of operation

6
System for Cardiac Ablation

7
Common cardiac ablation sites AV Node Above the tricuspid valves Above and underneath the mitral valves Ventricular walls Right ventricular outflow tract Etc.

8
Tip Electrode RF generator

9
Energies Involved in RF Ablation Process

10
Bioheat Equation Heat transfer coefficient Blood temperature Density Specific heat Thermal conductivity Time Temperature Current density Electric field intensity heat loss to blood perfusion VARIABLES Heat Change MATERIAL PROPERTIES Electrical conductivity Density Specific heat Thermal conductivity Time Temperature Current density Electric field intensity heat loss to blood perfusion Heat Conduction Joule Heat

11
Finite Element Analysis Divide the regions of interest into small “elements” Partial differential equations to algebraic equations 2-D (triangular elements, quadrilateral elements, etc.) 3-D (tetrahedral elements, hexahedral elements, etc.) Nonuniform mesh is allowed Software & Hardware PATRAN 7.0 (MacNeal-Schwendler, Los Angeles ) ABAQUS 5.8 (Hibbitt, Karlsson & Sorensen, Inc., Farmington Hills, MI) HP C-180, 1152 MB of RAM, 34 GB Storage

12
Process for FEM Generation Geometry Material Properties Initial Conditions Boundary Cond. Mesh Generation Preprocessing (PATRAN 7.0) Solution (ABAQUS/STANDARD 5.8) Duration Production Adjust Loads Check for desired parameters Postprocessing (ABAQUS/POST 5.8) Temperature Distribution Current Density Determine Lesion Dimensions (from 50 C contour) Convergence test (for optimal number of elements )

13
Modes of RF Energy Applications Maintain the tip temperature at a preset value Adjust voltage applied to the electrode Temperature controlled ablation Power controlled ablation Maintain power delivered at a preset value Adjust voltage applied to the electrode

14
1. Effects of changes in myocardial properties to lesion dimensions* *Tungjitkusolmun, S., Woo, E. J., Cao, H., Tsai, J.-Z., Vorperian, V. R.,and Webster, J. G.., Thermal-electrical finite element modeling for radio-frequency cardiac ablation: effects of changes in myocardialproperties, Med. Biol. Eng. Comput., accepted, 2000. 1.1 Electrical conductivity 1.2 Thermal conductivity 1.3 Specific heat (Density) Material Properties For each case: Temperature independent Temperature dependent Increase by 50%, or 100% Decrease by 50%

15
FEM results Lesion growth over time (Red is 50 C or higher)

16
Temperature distribution after 60 s Maximum temperature ~ 95 C Highest temperature

17
Maximum changes in Lesion Size PropertyCase% Volume Change Electrical conductivity 50% 58.6 Thermal conductivity +100% 60.7 Specific heat 50% +43.2 Power controlled

18
PropertyCase% Volume Change Electrical conductivity 50% +12.9% Thermal conductivity 50% 21.0% Specific heat+100% 29.4% Temperature controlled

19
Conclusion Temperature dependent properties are important Errors in Power-Controlled Mode are higher Better measurement techniques are needed

20
2. Needle electrode design for VT* E. J. Woo, S. Tungjitkusolmun, H. Cao, J.-Z. Tsai, J. G. Webster, V. R. Vorperian, and J. A. Will, “A new catheter design using needle electrode for subendocardial RF ablation of ventricular muscles: finite element analysis and in-vitro experiments,” IEEE Trans. Biomed. Eng., vol. 47, pp. 23 31, 2000.

21
Methods Both FEM & in vitro experiments Vary needle diameters Vary insertion depths Vary RF ablation duration Change temperature settings Compare lesion dimensions

22
FEM Results Insertion depth (mm)Lesion width (mm)Lesion depth (mm) 2.03.242.80 4.04.524.90 6.05.306.90 8.05.609.10 Needle Diameter (insertion = 8 mm) Insertion Depth (diameter = 0.5 mm) Diameter of needle (mm)Lesion width (mm)Lesion depth (mm) 0.55.609.1 0.66.069.1 0.76.249.1 0.86.509.1 0.96.779.2 1.07.049.3

23
Conclusion Lesion depths are 1 mm deeper than the insertion depth Lesion width increases with increasing diameter and duration Confirmed by in vitro experiments Good contact

24
Needle electrode designs

25
3. Uniform current density electrodes* *Tungjitkusolmun, S., Woo, E. J., Cao, H., Tsai, J.-Z., Vorperian, V. R., and Webster, J. G., Finite element analyses of uniform current density electrodes for radio-frequency cardiac ablation, IEEE Trans. Biomed. Eng., 47, pp. 32-40, January 2000. “hot spot” at the edge of the conventional electrode Uniform current density electrode by – Recession depth – contour on the surface of the electrode ( is the parameter for the shape function). – Filled with coating material

26
FEM results Hot spot at the edge of the metal electrode

27
Current densities at the edge of the tip electrode is the shape function

28
Cylindrical electrodes Changing conductivities Changing the curvatures (S/m) is for the shape function)

29
Current density distributions Cardiac tissue Catheter body Electrode Highest current density +0.00E+00 +2.50E 01 +5.00E 01 +7.50E 01 +1.00E+00 ECDM VALUE C SCALE = 144. Flat Catheter body Cardiac tissue Coating Uniform current density +0.00E + 00 +2.50E 01 +5.00E 01 +7.50E 01 +1.00E + 00 C SCALE = 582. ECDMVALUE Recessed

30
4. Bipolar phase-shifted multielectrode catheter ablation* *S. Tungjitkusolmun, H. Cao, D. Haemmerich, J.-Z. Tsai, Y. B. Choy, V. R. Vorperian, and J. G. Webster, “Modeling bipolar phase-shifted multielectrode catheter ablation,” in preparation, IEEE Trans. Biomed. Eng., 2000 TeTe TmTm

31
Method A. 3-D Unipolar Multielectrode Catheter (MEC) B. Optimal phase-shifted for a system with fixed myocardial properties Optimal phase-shift Optimal phase-shift: T e / T m = 1 C. Effects of changes in myocardial properties on the optimal phase-shift D. Optimal phase-shift for MEC with 3 mm spacing

32
FEM results Phase = 0 Phase = 26.5 Phase = 45

33
Phase vs. T e /T m Changes in electrical conductivity

34
Changes in thermal conductivity

35
Electrode spacing (2mm vs. 3mm)

36
Simplified Control system

37
5. FEM predicts lesion size* Ablation over the mitral valve annulus Ablation underneath the mitral valve leaflets *S. Tungjitkusolmun, V. R. Vorperian, N. C. Bhavaraju, H. Cao, J.-Z. Tsai, and J. G. Webster, “Guidelines for predicting lesion size at common endocardial locations during radio-frequency ablation,” submitted to IEEE.Trans. Biomed. Eng., 1999.

38
Physical conditions LocationBlood velocity (cm/s) h b at blood myocardium interface [(W/(m 2 K)] h be at blood electrode interface [W/(m 2 K)] Position 1 11.014174191 Position 2 2.75442197 PositionContactBlood flow 1. Above the mitral valve1.3 mm embeddedHigh 2. Underneath the mitral valve3.0 mm embeddedLow

39
Temperature Controlled RF Lesion volume vs. time

40
Power controlled RF Lesion volume vs. time

41
6. FEM for Hepatic Ablation* *S. Tungjitkusolmun, S. T. Staelin, D. Haemmerich, J.-Z. Tsai, H. Cao, V. R. Vorperian, F. T. Lee, D. M. Mahvi, and J. G. Webster, “Three-dimensional finite element analyses for radio-frequency hepatic tumor ablation,” submitted to IEEE. Trans. Biomed.Eng., 2000. Hepatic Ablation: Use RF probe to destroy tumor cancer, or cirrhosis Minimally invasive Present: -High recurrence rate -Small lesions

42
Models 4-tine RF Probe Geometry for FEM, 352,353 tetrahedral elements

43
Effect of Blood Vessel Location No Blood VesselBlood Vessel at 1 mm

44
Blood vessel at 5 mm

45
Bifurcated blood vessel

46
Summary 1. Outline a process for FEM creation for RF ablation 2. Show that needle electrode catheter design can create deep lesions by FEM & in vitro studies 3. Uniform current density electrodes reduce “hot spots” 4. Bipolar phase-shifted multielectrode catheter can create long and contiguous lesions 5. We can use FEM to predict lesion formations 6. Apply FEM for RF ablation to hepatic ablation

47
Bipolar Hepatic Ablation BipolarUnipolar

Similar presentations

OK

Introduction to Finite Element Modeling in Biomechanics Dr. N. Fatouraee Biomedical Engineering Faculty December, 2004.

Introduction to Finite Element Modeling in Biomechanics Dr. N. Fatouraee Biomedical Engineering Faculty December, 2004.

© 2018 SlidePlayer.com Inc.

All rights reserved.

To ensure the functioning of the site, we use **cookies**. We share information about your activities on the site with our partners and Google partners: social networks and companies engaged in advertising and web analytics. For more information, see the Privacy Policy and Google Privacy & Terms.
Your consent to our cookies if you continue to use this website.

Ads by Google

Ppt on wireless intelligent network Ppt on surface chemistry Ppt on computer languages to learn Ppt on water pollution pdf Ppt on aircraft landing gear systems Ppt on integration problems and solutions Ppt on business plan presentation Short ppt on green revolution Ppt on latest technology in electronics and instrumentation Ppt on asian continent outline