Principle of Microcantilevers The key elements in the detection of a mass are the vibrational frequency and the deflection of the cantilever* Deflection* Proportional to mass content Resonance frequency* ω R =(k/m) 1/2 K = spring constant M= mass * Sandeep Kumar Vashist (2007) Review of Microcantilevers for Sensing Applications Journal of Nanotechnology 3: 1-15.
Readout Method There are several methods available to observe the deflection and resonance frequency of the microcantilever* Optical* Piezoelectric* Piezoresistive* * Sandeep Kumar Vashist (2007) Review of Microcantilevers for Sensing Applications Journal of Nanotechnology 3: 1-15.
Optical Optical method requires the use of a low power laser beam* If microcantilever does not deflect, then no biomolecules have been absorbed* Laser beam hits a specific position on the position sensitive detector (PSD)* Major weakness-high cost* * Karolyn M. Hansen, Hai-Feng Ji, Guanghua Wu, Ram Datar, Richard Cote, Arunava Majumdar, and Thomas Thundat (2001) Cantilever-Based Optical Deflection Assay for Discrimination of DNA Single-Nucleotide Mismatches. Analytical Chemistry 73 (7): 1567-1571
Piezoresistive These sensors measure the strain induced resistance change* When the biomolecules are absorbed by the material there is a volumetric change in the sensing material* Volumetric change is measured by resistance change in cantilever* Advantages-Low cost* * Viral detection using an embedded piezoresistive microcantilever sensor. Sensors and Actuators A: Physical 107 (3), 219- 224
Piezoelectric These sensors detect the change in the resonance frequency of microcantilever only* Use microactuator to drive the plate into resonance* Microsensor to the determine the frequency of the plate* * S. Zurn, M. Hsieh, G. Smith, D. Markus, M. Zang, G. Hughes,Y. Nam, M. Arik and D. Polla (2001) Fabrication and structural characterization of a resonant frequency PZT microcantilever. Institute of Physics Publishing 10: 252-263
Applications Microcantilevers may be used to detect the presence against viruses, or even cancerous cells** Mass detection of Vaccina virus particle* Cancer monitoring** * Amit K. Gupta, Pradeep R. Nair, Demir Akin, Michael R. Ladisch, Steve Broyles, Muhammad A. Alam, and Rashid Bashir (2006) Anomalous resonance in a nanomechanical biosensor. PNAS 103 (36): 13362- 13367 ** Mauro Ferrari (2005) Cancer Nanotechnology: Opportunities and Challenges. Nature Publishing Group 5, 161-171 Figure 1* Figure 2**
Simulation (Mode Analysis) f 0 =194,532Hz f 1 =194,483Hz S Morshed and B.C. Prorok (2007) Tailoring beam mechanics towards enhancing detection of hazardous biological species. Experiment Mechanics 47:405-415
Design and optimization Tailoring geometry to improve resonance frequency and shift frequency K m Increase the spring constant Reduce the effective mass at the fee end S Morshed and B.C. Prorok (2007) Tailoring beam mechanics towards enhancing detection of hazardous biological species. Experiment Mechanics 47:405-415 ∆f /∆ m=π k 1/2 m -3/2 f=2π k 1/2 m -1/2
Design and optimization ∆f=41Hz ∆f=49Hz ∆f=69Hz ∆f=36Hz ∆f=31Hz Conclusion: Increase the clamping width; Reduce the width in free end S Morshed and B.C. Prorok (2007) Tailoring beam mechanics towards enhancing detection of hazardous biological species. Experiment Mechanics 47:405-415
Design and optimization ∆f = 506Hz Another advantage is the relatively uniform stress distributions We can put more piezoresistors on Disadvantage: Not enough room at the tip for capturing bioparticles! S Morshed and B.C. Prorok (2007) Tailoring beam mechanics towards enhancing detection of hazardous biological species. Experiment Mechanics 47:405-415
Design and optimization Final Structure Trapezoid-like cantilever ∆f=150Hz Further improve the frequency shift, how? Higher frequency mode! S Morshed and B.C. Prorok (2007) Tailoring beam mechanics towards enhancing detection of hazardous biological species. Experiment Mechanics 47:405-415
Higher frequency mode Element Model Solid187 6163 Elements overall Material propertiesYoung’s modulusDensityPoisson Ratio Value100 GPa2850 kg/m 3 0.24 S Morshed and B.C. Prorok (2007) Tailoring beam mechanics towards enhancing detection of hazardous biological species. Experiment Mechanics 47:405-415
Higher frequency mode Mode 1 Mode 2 ∆f=150Hz∆f=300Hz
Higher frequency mode Mode 5 ∆f=200 Hz Conclusion: Mode 2 has double shift frequency, and its amplitude is big enough for piezoresistors to sense.
Sensitivity Analysis The mass of the applied particle is 0.285 pg; while the frequency shift is 300Hz (using cantilever shape G and operating at the second mode) The sensitivity: S = 300Hz/0.285pg=1.05×10 18 s -1 kg -1
Fabrication: Phase One o The unaltered SOI wafer o Ion implantation to form piezoresistive element (Boron, dose ~10 14 /cm 2 ) o Deposition of photoresist on upper silicon layer (~1µm) Phase one of the fabrication process Photoresist
Fabrication: Phase Two o Photolithography to define tip and electrode o Wet etching to eliminate unexposed photoresist o Further etching to remove exposed photoresist Phase two
Fabrication: Phase Three o E-beam deposition of titanium (~5 nm) o E-beam deposition of Au (~150 nm) o Wet etching of remaining photoresist Phase three
Fabrication: Phase Four o DRIE to define cantilever o Bulk DRIE to eliminate Si substrate o Wet etching for removal of SiO 2 to free cantilever Phase four
Fabrication: Phase Five o Biosensitive film selectively binds to gold, allowing cantilever dipping Cell selectively binding to biosensitive layer* *Images can be found in: Lan, S., Veiseh, M. and Zhang, M. Surface modification of silicon and gold-patterned silicon surfaces for improved biocompatibility and cell patterning selectivity. Biosensors and Bioelectronics, 2005, 20(9), 1697-1708 Cells cultivated on gold with silicon substrate after biosensitive treatment*
Fabrication: Phase Six o Piezoelectric actuator stamped on base of cantilever The final product: a MEMS biosensor
Summary Portable device with convenient readout and external actuation. Optimized geometry and frequency sensitivity Easy fabrication using SOI wafer
Your consent to our cookies if you continue to use this website.