Download presentation

Presentation is loading. Please wait.

1
New test result of MPPC M.Taguchi(kyoto)

2
**Contents New MPPC sample Noise rate,gain,crosstalk rate PDE**

Long term stability

3
**New MPPC sample recently we got**

HPK32(400pixel) ×3 Operating voltage : 68~69V 1p.e signal width:~20ns HPK53(100pixel) ×3 Operating voltage:68~69V 1p.e signal width:~50ns HPK say these have higher PDE than old samples The test result of HPK14(100pixel) that we used at beamtest will be presented for comparison

4
**Noise rate measurement**

■ 0.5p.e threshold ▲ 1.5p.e threshold HPK53(100pixel) HPK32(400pixel) 1MHz 100kHz 15℃ 15℃ 20℃ 20℃ 25℃ 25℃ 68.4 69.6 68.4 69.4 Bias voltage Bias voltage

5
**Gain measurement@T=20℃ gain 3×106 HPK53(100pixel)**

Gain of new samples >105,so satisfy T2K requirement HPK14(100pixel) 1×106 HPK32(100pixel) 0.5p.e threshold noise rate 500kHz 1MHz

6
**Crosstalk rate measurement@T=20℃**

HPK14(100pixel) HPK53(100pixel) 0.2 Crosstalk rate of new samples is still high (T2K requires crosstalk rate<0.1) 0.1 HPK32(100pixel) 0.5p.e threshold noise rate 500kHz 1MHz

7
**PDE measurement@T=15℃ Blue LED MPPC 1mmφfiber 1mmφslit Blue LED MPPC**

With this set up all the light going through 1mmφslit inject to the active area of MPPC MPPC ½inch PMT 1mmφfiber Ratio of MPPC p.e to PMT p.e measured by this set up is defined as PDE scan 1mmφslit Blue LED With this set up some light from a fiber don`t reach to the active area of MPPC because there is a distance(~0.8mm) between surface cover and active area MPPC ½inchPMT 1mmφfiber Ratio of MPPC p.e to PMT p.e measured by this set up is defined as effective PDE scan

8
**PDE measurement@T=15° PDE is showed with relative value to PMT**

■ effective PDE HPK53(100pixel) HPK32(400pixel) HPK14(100pixel) 2 2 1 500 900 ノイズレート(kHz) ノイズレート(kHz) 500 900 550 900 400 900 Noise rate(kHz) Noise rate(kHz) Noise rate(kHz) PDE of new sample is 2 times larger than that of old one Effective PDE is 50~60% of PDE We requested the improvement of MPPC packaging to HPK

9
**Raw signal of new sample**

HPK53(100pixel) Noiserate=1345kHz HPK32(400pixel) Noiserate=1246kHz Noise with large pulse height (～10p.e) is observed when noise rate becomes >1MHz There is no problem because T2K requires noise rate <1MHz

10
**Long term stability test@T=20℃**

Test the stability of gain,noise rate,p.e for MIP Cosmic ray Trigger counter 4 layers HPK32(400pixel) Blue LED ½ inch PMT MPPC parameter Bias voltage 68.8V gain 5.6×105 0.5p.e th noiserate 800kHz Crosstalk rate 0.1 Effective PDE(including crosstalk) 1.2 3 different triggers Cosmic trigger LED trigger(for monitaring MPPC gain) Pedestal trigger(for monitaring noise rate)

11
Cosmic ray event MPPC PMT 14.8p.e 16.2p.e 20 20 60 60 p.e p.e

12
**Stability of MPPC gain and noiserate**

5% 5% 839kHz 5.7×105 5% 5% 40 hour 40 hour RMS/mean=2.5% RMS/mean=4.3%

13
**Stability of MPPC Light yield for MIP**

Peak adc count for MIP Peak adc count/gain for MIP 5% 5% 14.7p.e 5% 5% 40 hour 40 hour RMS/mean=4.0% RMS/mean=2.9% Correction by gain

14
summary New MPPC samples have about 2 times larger PDE than that of old ones MPPC light yield for MIP~15p.e (at the point where noise rate and gain satisfy T2K requirement) Gain stability ~3% ,noise rate stability~4% Stability of light yield for MIP~3% (about 2 days)

15
Back up

16
LED event MPPC PMT

17
Staibility of PMT gain 1p.e adc count 5% 5% 40hour RMS/mean = 4.7%

18
**Stability of PMT light yield for MIP**

Peak adc count for MIP Peak adc count/gain for MIP 5% 5% 16.5p.e 5% 5% 40hour 40hour RMS/mean=4.6% RMS/mean=3.1% Correction by gain

19
**Operating voltage range which satisfy T2K requirement**

HPK32(400pixel) Upper limit：1.5p.e th noise rate<50kHz 68.2～68.8V Lower limit ：PDE >15% HPK53(100pixel) Upper limit：1.5p.eth noise rate<50kHz 68.3～69.1V Lower limit：PDE>15%

20
**Recovery time LED1 400p.e LED1 LED2 HPK32 (400pixel) LED2 160ns**

Fire all pixels with LED1 and study the response of LED2 with the delay to LED1 Saturation signal was observed by LED2 even with the 120ns(signal width) delay 400p.e LED1 LED2 recovery time is about the same as signal width 120ns

Similar presentations

OK

Performance of MPPC using laser system Photon sensor KEK 2005.12.26-27 Niigata university, ILC calorimeter group Sayaka IBA, Hiroaki ONO, Paul.

Performance of MPPC using laser system Photon sensor KEK 2005.12.26-27 Niigata university, ILC calorimeter group Sayaka IBA, Hiroaki ONO, Paul.

© 2017 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Ppt on high level languages vs low level Download ppt on automatic street light control Ppt on model view controller explained Download ppt on energy conservation and management Ppt on health insurance in india Ppt on success and failure articles Ppt on cattle farming for class 9 Ppt on cost accounting Ppt on question tags on youtube Ppt on nature of human resource management