Download presentation

Presentation is loading. Please wait.

Published byJonas Hines Modified over 2 years ago

1
Soft x-ray tomography on HT-7 tokamak K.Y. Chen, L.Q. Hu, Y.M. Duan HT-7

2
Outline Experimental set-up Theoretical background Performance of the tomography system A case study Summary HT-7

3
Experimental set-up Major radius: 1.22 m Minor radius: 0.27 m Five cameras 46 channels for each camera Thickness of beryllium filter 12.5 μm. Distance for plasma center to slit: 400 Maximum sampling rate: 250 kHz Spatial resolution : 0.6 ~ 1.5 cm Data acquisition: Photon Be filter Photon diode Preamplifier Amplifier AD Control room HT-7

4
Theoretical background The quantity in brackets can be divided into the left side to define the Chord ‘brightness’, with units of W·m -2. The tomography problem HT-7 Assuming soft x-ray (SXR) emissivity is constant on magnetic surfaces, the contours of SXR emissivity represent the magnetic surface structure. Our task is to invert the line-integrated soft x-ray (SXR) data to reconstruct the SXR emissivity distribution, i.e. magnetic surface structure. Theoretical background

5
Solution of the tomography problem The Fourier–Bessel inversion method is used to generate tomographic reconstructions from the soft X-ray data. The local emissivity can be expanded in terms of Fourier harmonics to yield: If g m (r) is expanded in terms of Bessel series, the chord brightness f(p,φ) can be written as Whereis theth zero of the Bessel function. After numerical Integration along the chords L(p,φ), Eq. (1) can be put in a matrix form and inverted to obtain the a m,l coefficients and consequently the local emissivities. Usually, the radial expansion limit l max is adequate, whereas the angular expansion limit m max is restricted by the number of soft x-ray cameras. For a soft x-ray imaging system with N cameras, m max can be N-1/2 at most, where 1/2 represents the component of cos (Nθ) or sin (Nθ). HT-7 (1)

6
Performance of the tomography system Simulation of emissivity distributions Assuming a pure hydrogen or deuterium plasma, the emissivity is simulated as where equilibrium temperature and density profiles are derived from PHA temperature diagnostic and HCN density diagnostic in a typical discharge. The perturbed temperature and density distributions are simulated with Kadomtsev’s model. HT-7 ξ= 0.3 ξ= 0.2 m max =1 1 / 2 m max =2m max =3

7
A case study HT-7 A typical sawtoothing discharge with LHCD and IBW Temporal evolution of main plasma parameters Expanded view of SXR intensities Mid-oscillation Pre-cursor oscillation

8
Mid-oscillation HT-7 Results of singular value decomposition EigenvalueChronoTopo Tomographic reconstructions A B C

9
Sawtooth crash HT-7 Results of singular value decomposition EigenvalueChronoTopo Tomographic reconstructions D E F

10
Discussion Mid-oscillations are primarily due to hot core displacement and the rotation of m=1 magnetic island. The appearance of mid-oscillations indicate that m=1 during the sawtooth does not necessarily mean sawtooth crash. Mid-oscillations are frequently observed in LHCD plasmas, while seldom observed in ohmic-heating plasmas. This suggests that LHCD may play an important role in resulting in the oscillations. The modified current density profile may be responsible for the growth, saturation and decay of the oscillations. Sawtooth crash is due to the growth of m=1 mode. Sometimes m=2 mode may play a role in the crash, but this is seldom observed. After the crash, the soft x-ray emissivity in the plasma center is relatively cold. This may be due to the large pressure gradient outside the inversion surface. Besides, low SXR intensity does not necessarily mean low temperature. HT-7

11
Summary HT-7 The soft x-ray imaging system on HT-7 tokamak contains five soft x-ray cameras. The temporal and spatial resolution are 4 μs and 0.6-1.5 cm respectively. Fourier-Bessel inversion method was used to reconstruct the distribution of soft x-ray emissivity. The inversion accuracy is associated with the number of angular harmonic and the displacement of plasma hot core. The tomographic results show that mid-oscillation and sawtooth crash are primarily due to the behavior of m=1 mode.

12
Thank you for attention! HT-7

Similar presentations

OK

Plasma MHD Activity Observations via Magnetic Diagnostics Magnetic islands, statistical methods, magnetic diagnostics, tokamak operation.

Plasma MHD Activity Observations via Magnetic Diagnostics Magnetic islands, statistical methods, magnetic diagnostics, tokamak operation.

© 2017 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Ppt on views in dbms definition Ppt on word association test pdf Ppt on oxidation and reduction for class 10 Ppt on brand marketing books Ppt on area of parallelogram vectors Ppt on railway track construction Ppt on lost city of atlantis Ppt on power sharing in india download movies Ppt on national festivals of india Ppt on obesity prevention in schools