We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byTanner Swanton
Modified about 1 year ago
EE 5340, SMU Electrical Engineering Department, © 1999 1 Carlos E. Davila, Electrical Engineering Dept. Southern Methodist University slides can be viewed at: http:// www.seas.smu.edu/~cd/ee5340.html EE 5340/7340 Introduction to Biomedical Engineering Electromagnetic Flowprobes
EE 5340, SMU Electrical Engineering Department, © 19992 Electromagnetic Flowmeters VoVo + _ electromagnet blood vessel indicator dilution methods assume flow rate is constant, only measure average flow. EM flowmeters enable measurement of instantaneous flow.
EE 5340, SMU Electrical Engineering Department, © 19993 Faraday’s Law voltage induced across electrodes velocity of blood (m/s) magnetic flux density (Wb/m 2 ) vector in direction of electrodes length of -a moving conductor in a (possibly constant) magnetic field will have a voltage induced across it response is maximized when,, and are mutually orthogonal
EE 5340, SMU Electrical Engineering Department, © 19994 Toroidal Cuff Probe
EE 5340, SMU Electrical Engineering Department, © 19995 DC Flowmeter n use DC (constant) magnetic field n half-cell potential results across each sensing electrode, in series with the flow signal, even with non-polarizable potentials n pick up stray ECG n basically doesn’t work well, and DC flowmeters are not used. n flow frequency range: 0 - 30 Hz
EE 5340, SMU Electrical Engineering Department, © 19996 AC Flowmeter n frequency of : about 400 Hz n V o becomes amplitude modulated sine wave: 400 Hz carrier 0 flow need a phase-sensitive demodulator
EE 5340, SMU Electrical Engineering Department, © 19997 Transformer Voltage VtVt + _ blood vessel plane of electrode wires should be parallel to magnetic field. Otherwise, get transformer voltage, V t, proportional to:
EE 5340, SMU Electrical Engineering Department, © 19998 Transformer Voltage (cont.) t t t magnet current, i m (t) transformer voltage, v t (t) flow voltage, v f (t) 90 o out of phase 0 or 180 o out of phase, depending on flow direction
EE 5340, SMU Electrical Engineering Department, © 19999 Removal of Transformer Voltage n Phantom Electrode n Gating Flow Voltage n Quadrature Suppression
EE 5340, SMU Electrical Engineering Department, © 199910 Phantom Electrode VtVt + _ blood vessel adjust until transformer voltage = 0
EE 5340, SMU Electrical Engineering Department, © 199911 Gating Flow Voltage t t t flow voltage, v f (t) magnet current, i m (t) transformer voltage, v t (t) sample flow voltage when transformer voltage = 0
EE 5340, SMU Electrical Engineering Department, © 199912 Quadrature Suppression Discussed in Chapter 8 of text. To understand it fully, we must go over several modulation/demodulation methods: n Amplitude Modulation/Demodulation n Double Sideband Modulation /Demodulation n Quadrature Multiplexing/Demultiplexing
EE 5340, SMU Electrical Engineering Department, © 199913 Amplitude Modulation/Demodulation A Modulation: Demodulation (envelope detector): C R + _ + _ : information-bearing signal carrier frequency
EE 5340, SMU Electrical Engineering Department, © 199914 Double Sideband (DSB) Modulation/Demodulation modulation: demodulation: LPF m(t) can be bipolar carrier frequency and phase must be known carrier frequency this demodulator is phase sensitive
EE 5340, SMU Electrical Engineering Department, © 199915 DSB Modulation/Demodulation (cont.) trigonometric identity: LPF
EE 5340, SMU Electrical Engineering Department, © 199916 DSB Modulation/Demodulation (cont.) Frequency Domain: from frequency shifting property of the Fourier Transform: USB LSB 0
EE 5340, SMU Electrical Engineering Department, © 199917 DSB Modulation/Demodulation (cont.) 0 0 LPF =
EE 5340, SMU Electrical Engineering Department, © 199918 Quadrature DSB (QDSB) Modulation -allows one to transmit two different information signals, m 1 (t) and m 2 (t) using the same carrier frequency, this enables more efficient bandwidth utilization.
EE 5340, SMU Electrical Engineering Department, © 199919 QDSB Demodulation LPF
EE 5340, SMU Electrical Engineering Department, © 199920 QDSB Demodulation (cont.) Trigonometric Identities:
EE 5340, SMU Electrical Engineering Department, © 199921 QDSB Demodulation (cont.) LPF
EE 5340, SMU Electrical Engineering Department, © 199922 Quadrature Suppression -used to suppress transformer voltage vessel amp LPF oscillator magnet current generator 90 o phase shift LPF
EE 5340, SMU Electrical Engineering Department, © 199923 Electromagnetic Flowprobe: Case Study- Cliniflow II, Carolina Medical SPECIFICATIONS ACCURACY Electrical Zero --- Automatic zero for occlusive or non-occlusive zero reference. Calibrate Signal --- -1V to +1V in 0.1V steps @ 0.2 sec/step. Flowmeter Calibration Accuracy --- +/-3% of full scale after a 5 second warm-up. (Includes the effect of gain and excitation variation.) DC Drift --- +/-5mV after a 5 second warm-up. Linearity --- +/-1% maximum full scale.
EE 5340, SMU Electrical Engineering Department, © 199924 Case Study (cont.) SAFETY Patient Isolation --- Isolated patient ground. 2500V RMS. Equipment Isolation --- External connections to recorders, etc, are optically isolated to preserve patient protection even when connected to external equipment. Electrical Isolation --- Designed to comply with UL544 specifications. No exposed, non-isolated metal surfaces available to the operator or patient.
EE 5340, SMU Electrical Engineering Department, © 199925 Case Study (cont.) INPUT CHARACTERISTICS Autoranging --- Overall gain, full scale recorder output amplitude, flow rate range indicator and decimal point location are automatically programmed by the selected probe. Probe Excitation --- 450 or 475Hz square-wave, 0.5 Ampere +/-l%. Amplifier Input --- Differential >30 megohm plus 50pF. CMRR >/- or =80dB @ 60Hz. Defibrillator protected.
EE 5340, SMU Electrical Engineering Department, © 199926 Case Study (cont.) OUTPUT CHARACTERISTICS Flow Range --- 5 milliliters/min to 19.99 liters/min depending on probe selected. Gain --- Automatically preset by the probe used. Flow Indicator --- 3.5 digit red L.E.D. display, automatic calibration, automatic flow direction indicator. Outputs PULSATILE: Single ended, +/-lOV (20Vp-p) full scale. MEAN: single ended, +/-1.999V (4Vp-p) full scale. BOTH: capable of driving 1 kohm minimum load. Short circuit protected. Isolated from power or chassis ground.
EE 5340, SMU Electrical Engineering Department, © 199927 Case Study (cont.) Frequency Response --- Front panel selectable, 3dB down @ 12Hz, 25Hz, 50Hz or 100Hz. Output Noise PULSATILE: 11OmV typical @ 100Hz response, 30mV typical @ 12Hz response. (Varies with the probe used and the frequency response setting.) MEAN: 5mV maximum.
EE 5340, SMU Electrical Engineering Department, © 199928 Case Study (cont.) courtesy of Carolina Medical examples of electromagnetic flowprobes
EE 5340, SMU Electrical Engineering Department, © 199929 Case Study (cont.): example of EM flowmeter courtesy of Carolina Medical
Electromagnetic Blood Flow Meters Measures instantaneous pulsatile flow of blood Works based on the principle of electromagnetic induction The voltage.
EE 5340/7340, SMU Electrical Engineering Department, © Carlos E. Davila, Electrical Engineering Dept. Southern Methodist University slides can be.
Prof. Brian L. Evans Dept. of Electrical and Computer Engineering The University of Texas at Austin EE445S Real-Time Digital Signal Processing Lab Fall.
By: Engr. Hinesh Kumar Lecturer. Amplifiers used to process biopotentials are called biopotential amplifier. Biopotential Signals (e.g., ECG, EMG, EEG,
Oscilloscope Tutorial The oscilloscope is basically a graph- displaying device It draws a graph of an electrical signal. In most applications the graph.
Vital Signs Monitor UConn BME 4900 Vital Signs Monitor Purpose As the population ages, many people are required by their doctors to take vital signs.
Prof. Brian L. Evans Dept. of Electrical and Computer Engineering The University of Texas at Austin EE345S Real-Time Digital Signal Processing Lab Spring.
Et438b-4.pptx 1 ET 483b Sequential Control and Data Acquisition.
Prof. Brian L. Evans Dept. of Electrical and Computer Engineering The University of Texas at Austin EE345S Real-Time Digital Signal Processing Lab Fall.
Biomedical Instrumentation I Chapter 7 Bioelectric Amplifiers from Introduction to Biomedical Equipment Technology By Joseph Carr and John Brown.
Project by Santiago Yeomans, Chad Cummins, Gboyega Adeola Guitar Signal Transmitter.
Electronics Involves the use of devices and circuits to control the flow of electric current to achieve some purpose. These circuits contain: Resistors,
Chapter 3 Amplitude Modulation Circuits AMPLITUDE MODULATORS AM is the process of multiplying the carrier by the modulating signal. Non-linear mixing.
1 Lecture 7 AM and FM Signal Demodulation Introduction Demodulation of AM signals Demodulation of FM Signals Regeneration of Digital Signals and Bias Distortion.
OUTLINE Multimeters Analog Multimeters Digital Multimeters Breadboard Oscilloscope Signal Generator Clamp Meter Voltage Transformer Current.
Biomedical Electrodes, Sensors, and Transducers from: Chaterjee, Biomedical Instrumentation, chapter 6 © D. J. McMahon 2014 rev
Example Problem You are measuring the EEG of a patient and accidently choose two different types of electrodes for EEG lead. One of them has a source impedance.
10/11/2015 Operational Amplifier Characterization Chapter 3.
Biopotential Amplifier Speaker: Sun Shih-Yu 3/20, 2006.
Halliday/Resnick/Walker Fundamentals of Physics 8 th edition Classroom Response System Questions Chapter 31 Electromagnetic Oscillations and AC Current.
Introduction to Op Amps ENGI 242 ELEC 222. April 2004ENGI 242/ELEC 222 Op Amps2 Basic Op-Amp The op-amp is a differential amplifier with a very high open.
1 Complex Gain: Basic amplifier concepts Q: What is a Phasor? A Phasor is an ac voltage or current that maintains the same phase with respect to a standard.
Chapter 2 : Amplitude Modulation (AM) BENG 2413 Communication Principles Faculty of Electrical Engineering 1 Chapter 2 : Amplitude Modulation (AM) Transmission.
General Licensing Class Your HF Transmitter Your organization and dates here.
Alternating Voltage and Current Topics Covered in Chapter : Alternating Current Applications 15-2: Alternating-Voltage Generator 15-3: The Sine.
Self-Induction Transducers ~ AC Supply v ref Inductance Measuring Circuit x (Measurand) Ferromagnetic Target Object Coil is activated by the supply and.
Electrical Production of Sound 1Electric Circuits 2Electric Current 3Resistance 4Voltage 5Ohm’s Law 6Series and Parallel Circuits 7Electric Energy and.
Interferometric Residual Phase Noise Measurement System Pakpoom Buabthong Lee Teng Internship Program Advanced Photon Source Accelerator System Division,
Amplitude/Phase Modulation. Baseband and carrier communications The term baseband refers to a band of frequencies of signal generated by a information.
Module 4 Operational Amplifier. Contents Introduction Introduction OPAMP Symbol OPAMP Symbol Internal block diagram Internal block diagram Open-loop configuration.
Chapters 17 through 23 Midterm Review. Midterm Exam ~ 1 hr, in class 15 questions 6 calculation questions One from each chapter with Ch. 17 and 18 combine.
1 Discussion about the mid-term 4. A high voltage generator is made of a metal sphere with a radius of 6 cm sits on an insulating post. A wire connects.
Department of Electronics and CommunicationsEngineeringYANSAHAN UNIVERSITY Department of Electronics and Communications Engineering YANSAHAN UNIVERSITY.
1 Chelmsford Amateur Radio Society Advanced Licence Course Anthony Martin M1FDE Slide Set 12: v1.4, 2-Dec-2012 (4) Receiver Demodulation Chelmsford Amateur.
Closed Circuits In applications requiring the use of current, electrical components are arranged in the form of a circuit. A circuit is defined as a path.
1 ELECTRICAL CIRCUIT ET 201 Define and explain characteristics of sinusoidal wave, phase relationships and phase shifting.
CHAPTER 3: SPECIAL PURPOSE OP-AMP CIRCUITS. Objectives: Explain and analyze the operation of an instrumentation amplifier. Explain and analyze the operation.
BASIC INSTRUMENTS - OSCILLOSCOPES An oscilloscope is an instrument that allows observation of constantly varying signal voltages, usually as a two- dimensional.
S Transmission Methods in Telecommunication Systems (4 cr) Carrier Wave Modulation Systems.
Chapter 27 Current and Resistance Scalar Sense determined by the movement of the positive charge carrier Average Electric Current Instantaneous Electric.
1 Chapter 20 Circuits. 2 1) Electric current and emf a)Potential difference and charge flow Battery produces potential difference causing flow of charge.
COMMUNICATION SYSTEM EEEB453 Chapter 2 AMPLITUDE MODULATION Dept of Electrical Engineering Universiti Tenaga Nasional.
Measuring Phase Difference Dr. Len Trombetta 1 ECE 2100.
School of Electrical, Electronics and Computer Engineering University of Newcastle-upon-Tyne Baseband Digital Modulation Baseband Digital Modulation Prof.
CHAPTER4: CONTINUOUS-WAVE (CW) MODULATION First semester King Saud University College of Applied studies and Community Service 1301CT.
Eeng Chapter 5 AM, FM, and Digital Modulated Systems Amplitude Modulation (AM) Double Sideband Suppressed carrier (DSSC) Assymetric Sideband.
7th Lecture Dimitar Stefanov. Recapping Three types electrodes are used for sensing of EMG signals: 1.indwelling (intramuscular) electrodes (single fiber.
Integrator Op Amp Amplifier M.S.P.V.L. Polytechnic College, Department of ECE & EEE, Pavoorchatram.
© 2017 SlidePlayer.com Inc. All rights reserved.