Download presentation

Presentation is loading. Please wait.

Published byKolby Shillito Modified over 3 years ago

1
Atomic Clocks for Amateur Radio Astronomy Presented by Shad Nygren at Society of Amateur Radio Astronomers regional meeting Owens Valley CA Feb 28 & 29, 2004 Presentation revised Sept 26, 2007

2
Radio Astronomy Applications of Accurate Time and Frequency VLBI Pulsar timing Doppler shift measurements Zeeman effect Relativity

3
Time scales for VLBI Two time scales are important: Over periods of a few minutes, frequency stability must be adequate to provide sufficient integration time for extracting the VLBI observables. Over periods from several hours to a day, the frequency standard must work as a stable clock for recording signal arrival times. All VLBI observations must be phase-coherent, so a frequency reference standard which is significantly better than 1/(highest observing frequency) is required. The stability should be maximized over a period of 1 to 1000 seconds, the latter being the approximate length of a source scan. (Source: http://newton.jb.man.ac.uk/tmfrq.htm)

4
Coherence Time The coherence time is the temporal interval over which we can reasonably predict the phase of the light wave at a given point in space. Temporal coherence relates to the finite bandwidth of the source

5
What types of Atomic Clocks are available to amateurs Quartz Rubidium Cesium GPS Loran

6
Factors affecting clocks especially Quartz Aging is "the systematic change in frequency with time due to internal changes in the oscillator" Drift is "the systematic change in frequency with time of an oscillator." Drift is due to aging plus changes in the environment and other factors external to the oscillator. Environment, Temperature, Vibration, Atmospheric Pressure?, Voltage, Magnetic Fields Vibration / Acceleration / Gravity (The acceleration can be a steady-state acceleration, vibration, shock, attitude change (2-g tipover), or acoustic noise.) Clock Noise

7
1.Random walk FM (random walk of frequency), S plot goes down as 1/f 4. 2.Flicker FM (flicker of frequency), S plot goes down as 1/f 3. 3.White FM (white of frequency), S plot goes down as 1/f 2. 4.Flicker PM (flicker of phase), S plot goes down as 1/f. 5.White PM (white of phase), S plot is flat. Power-law noise processes are models of precision oscillator noise that produce a particular slope on a spectral density plot. We often classify these noise processes into one of five categories. For plots of S (f), they are:

8
Phase Noise Phase Noise is the noise level relative to the maximum signal. Note that the “distance” from the carrier is actually specified in frequency... Phase Noise is given at any frequency “away” from the carrier in units relative to the maximum signal gain at the carrier. These units have the mysterious name “dBc/Hz” (ratio at a given freq of noise to signal at a 1 Hz bandwidth) dBc is decibels relative to the carrier. dBc/Hz is decibels relative to the carrier per Hertz. These units are used to describe in decibels how far down signals and noise are relative to a known signal. (Source: http://www.radarproblems.com/chapters/ch06.dir/ch06pr.dir/c06p7.dir/ c06p7.htm) Phase noise is a small fraction of undesirable frequency near the output frequency. Phase noise is dependent mostly on the crystal with the circuitry making up the unit playing a small role. The measurement is commonly in the 1 Hz bandwidth. The description of phase noise is "at x Hz offset it is y dBc/Hz SSB (Single Side Band)

9
Phase Modulation

10
Phase Noise Components White Phase: Crystal Q Flicker Phase: Thermal noise White Frequency: Oscillator components Flicker Frequency: Oscillator Q Each segment of phase noise has been identified as being caused by unique characteristics….

11
Random Walk Noise A type of oscillator noise caused by environmental factors such as mechanical shock, vibration and temperature fluctuations which cause random shifts in frequency. As a general rule, random walk noise cannot be removed by averaging.

12
Flicker Noise A type of low frequency noise where the power spectral density is inversely proportional to the frequency. For this reason, it is sometimes referred to as 1/f noise. Flicker noise is one of the least understood. It is usually attributed to manufacturing noise mechanisms or device surface states. Experimental data shows that this type of noise has a dependence on dc current and is similar to shot noise Reminded someone of candle flickering Also called “pink noise”

13
White Noise Also known as Shot Noise Is independent of frequency Shot noise is generated by the random fluctuations in the normal current flow. Because each electron carries a discrete amount of charge and the flow of electrons is subject to small random fluctuations, a noise current is generated.

14
Random White Noise

15
How is clock performance characterized? Allan Variance Drift Phase Noise

16
Allan Variance The advantage of this variance over the classical variance is that it converges for most of the commonly encountered kinds of noise. In contrast the classical variance does not always converge to a finite value. For example, Flicker noise and random walk noise are two types which commonly occur in clocks and in nature where the classical variance does not converge. If the fluctuations are characterized by flicker noise or any other non- white-noise frequency deviations, what happens to the standard deviation for that data set? One can show that the standard deviation is a function of the number of data points in the set; it is also a function of the dead time and of the measurement system bandwidth. For example, using flicker noise frequency modulation as a model, as the number of data points increases, the standard deviation monotonically increases without limit. REFERENCE: http://www.boulder.nist.gov/timefreq/phase/Properties/four.htm

17
Allan Variance Concept Source: http://www.ieee-uffc.org/freqcontrol/tutorials/Parker_2002_files/frame.htm

18
Allan Deviation Noise Source: http://www.boulder.nist.gov/timefreq/general/glossary.htm

19
Modified Allan Variance The Allan variance filter cannot distinguish between white Phase Noise and flicker Phase Noise. Synchronization networks require frequency standards which have good time and phase stability. White PM and flicker PM levels are an important indicator of time/ phase stability. Therefore, another algorithm is necessary to distinguish between the various noise components. Modified Allan variance (MVAR) will discern white PM and flicker PM.

20
Modified Allan Noise

21
Quartz Crystal AT cut or SC (Stress Compensated) cut XO gives 5 ppm to 10 ppm per year. TCXO (Temperature Compensated) gives 0.5 ppm to 2 ppm per year. OCXO (Oven Controlled) 0.05 ppm to 0.1 ppm per year. The highest precision OCXOs can age a few parts in 10 12 per day, i.e., less than 0.01 ppm per year.

22
Quartz Crystal Cont The frequency accuracy of resonators is affected by many variables, some of which are temperature, aging, drive level, retrace and vibration. The SC-cut, Silicon Dioxide Crystal, is the only known resonator which has the inherent compensation required to minimize frequency shift versus all of the above factors. Prior to the SC discovery, the most versatile design was the single rotation AT-Cut, which was discovered by groups from America, Germany and Japan in 1929, and still remains the most widely used design today. Although the doubly rotated SC-cut (Stress Compensated) crystal was developed in the 1970's(4,&5) it has only been in recent years that Piezo Crystal and a few other companies have refined the manufacturing processes to the point where the crystal is being produced in substantial quantities with respectable yields over a wide range of frequencies. Source: http://www.corningfrequency.com/library/dasc.html

23
Rubidium Resonance 6,834,682,608 Hz Optical pumping relies on the natural coincidence of optical resonant frequencies between 85 Rb and 87 Rb, both at 795nm. Good short term stability Poor long term stability Used in VLBI until superseded by Hydrogen Maser

24
Cesium Resonance at 9,192,631,770 Hz by definition of the second C-Field Tube life approx 3 years Cost of used HP 5061A on eBay approx $1500 Cost of new tube is about $10K While cesium clocks have excellent long-term stability, the greatest weakness is the short-term stability caused by shot noise in the cesium beam. Not used for VLBI because of poor short term stability

25
Hydrogen Maser Resonance at 1420MHz Active and Passive designs. Currently the most stable frequency standard for averaging times between 10 3 and 10 4 seconds where stabilities as good as 7x10 -16 are reached. Hydrogen masers are used as VLBI standards because of their excellent short term stability. Their long-term accuracy, however, is subject to drift. H-Maser Oscillators have a wear out mechanism, mainly from Hydrogen depletion and Ion Pump failures, which begin to show up in 3 to 5 years for active masers and 5 to 7 years for passive masers. Cost >$60K to >$200K

26
Cesium Fountain NIST-F1 Laser Cooled The nation's primary time and frequency standard Uncertainty about 5 x 10 -16

27
Time Scales TAI UTC UT1 GPS Loran TDB – Barycentric Dynamical Time

28
TAI timescale The reference used is International Atomic Time (TAI), a time scale calculated at the BIPM using data from some two hundred atomic clocks in over fifty national laboratories. TAI is a uniform and stable scale which does not, therefore, keep in step with the slightly irregular rotation of the Earth.

29
Leap Seconds Currently 13 leap seconds (+1 in 2007) As of 2007 these leaps have always been positive

30
UT1 timescale is the principal form of Universal Time. It is computed from the raw observed UT0 by correcting UT0 for the effect of polar motion on the longitude of the observing site. UT1 is the same everywhere on Earth, and is proportional to the true rotation angle of the Earth with respect to a fixed frame of reference. Since the rotational speed of the earth is not uniform, UT1 has an uncertainty of plus or minus 3 milliseconds per day. The ratio of UT1 to mean sidereal time is defined to be 0.997269566329084 − 5.8684×10 −11 T + 5.9×10 −15 T², where T is the number of Julian centuries of 36525 days each that have elapsed since JD 2451545.0 (J2000)JDJ2000

31
UTC Such a scale is Coordinated Universal Time (UTC), which is identical with TAI except that from time to time a leap second is added to ensure that, when averaged over a year, the Sun crosses the Greenwich meridian at noon UTC to within 0.9 s (Coordinated Universal Time) is an atomic timescale that approximates UT1. It is the international standard on which civil time is based. It ticks SI seconds, in step with TAI. It usually has 86400 SI seconds per day, but is kept within 0.9 seconds of UT1 by the introduction of occasional intercalary leap seconds. When an accuracy better than one second is not required, UTC can be used as an approximation of UT1. The difference between UT1 and UTC is known as DUT1

32
UTC Official international time is Coordinated Universal Time, UTC UTC is a paper clock generated once a month by the Bureau International des Poids et Mesures, BIPM with data collected from about 230 atomic clocks in 50 labs around the world and distributed in BIPM Circular T. The method of time transfer is primarily common-view GPS and some Two-Way Satellite Time and Freq Transfer. Local representations of UTC are hardware clocks with ticks generated in real time, UTC(NIST) for example. Local UTC’s are estimations of UTC.

33
BIPM

34
USNO Source of GPS time 61 Atomic Clocks, 11 Hydrogen Masers and 50 HP-5071 Cesium

35
NIST http://www.boulder.nist.gov/timefreq/ Traceability

36
Time and Frequency Transfer One-Way Common-View Two-Way

37
Common View Satellite GPS GOES (obsolete) GLONASS (partial) GALILEO (future) Fixed plus variable delay components

38
Two-Way Satellite Time and Frequency Transfer This technique provides stable and accurate time transfer since nearly all of the propagation delay cancels out due to symmetry. Recently the BIPM has started using two-way transfer via a commercial Ku-band communication satellite as the primary time transfer technique for some European and transatlantic links, since it performs significantly better than GPS common view.

39
Time Sources GPS Loran WWV / WWVB

40
GPS The Block IIF satellite, the latest generation of GPS vehicles that began replacing older satellites in 2001, feature four advanced atomic clocks composed of both cesium and rubidium technologies. These extremely accurate GPS atomic clocks keep time to within 8 nanoseconds a day. The atomic clocks aboard the satellites maintain their time to a very high degree of accuracy. However, there will always be a slight variation in clock rates from satellite to satellite. Close monitoring of the clock of each satellite from the ground permits the control station to insert a message in the signal of each satellite which precisely describes the drift rate of that satellite's clock. The insertion of the drift rate effectively synchronizes all of the GPS satellite clocks.

41
GPS Cont Excellent Long Term Stability Poor Short Term Stability Short term stability is degraded by satellite orbital variations, ionosphere delay, troposphere delay, signal multipath from nearby terrain and buildings, receiver electronics, installation (antenna cable) and selective availability (presently turned off)

42
GPS Cont 2 Receiver electronics have more or less delay Best units < 30ns rms ref to UTC Worst units > 1us rms ref to UTC

43
Loran Long Range Navigation time, is an atomic time scale implemented by the atomic clocks in Loran- C chain transmitter sites. Loran time was zero at 0h 1-Jan-1958 and since it is not perturbed by leap seconds it is now ahead of UTC by 23 seconds. Making a comeback after almost being cancelled as GPS backup System being upgraded to enhanced version known as eLoran

44
Loran Cont US Public Law 100-223, Section 310 requires the control the transmission of LORAN signals to within +/- 100 nanoseconds of UTC at times of coincidence (TOC). The Coast Guard has worked with USNO to steer LORAN transmissions to UTC USNO monitors & measures LORAN time differences (TD) in far field to determine difference between LORAN Chain & UTC

45
PRS10 Rubidium + GPS Source: http://www.thinksrs.com/products/PRS10.htm

46
HP Z3801A Stability Source: http://www.leapsecond.com/pages/z3801a-osc/

47
Comparison of Frequency Standards Source: http://www.ieee-uffc.org/freqcontrol/quartz/vig/vigcomp.htm

48
Allan Deviation Comparison Source: http://www.leapsecond.com/museum/manyadev.gif

49
Equipment Sources eBay Stanford Research PRS10 Garmin Rubidium Efratom FRS-C GPS / Motorola Oncore

50
Information Sources http://www.leapsecond.com http://www.boulder.nist.gov/timefreq/

51
Conclusion Atomic Clocks are practical and affordable for Amateur Radio Astronomy. The combination using a Rubidium disciplined to GPS gives good short and long term performance. Opens up the possibility of amateur VLBI and other scientifically meaningful measurements.

52
Recommendations Use equipment with external 10MHz reference inputs Mfg / Vendors should design equipment to accept external 10MHz reference

53
Questions?

54
Clock History Christian Huygens’ (Galileo’s?) pendulum clock (1582) was stable to 1 min/day, but useless for moving platforms John Harrison’s “Marine Chronometer” (1761) kept time on a rolling ship to 0.2 s/day; determined longitude to 0.5 deg Quartz clocks have evolved from 1930’s to be stable to better than 1 part in 10 13

55
HP 58540A

56
Brooks Shera's GPS Controller

Similar presentations

Presentation is loading. Please wait....

OK

SURVEYING II UNIT IV PRESENTATION II.

SURVEYING II UNIT IV PRESENTATION II.

© 2018 SlidePlayer.com Inc.

All rights reserved.

To ensure the functioning of the site, we use **cookies**. We share information about your activities on the site with our partners and Google partners: social networks and companies engaged in advertising and web analytics. For more information, see the Privacy Policy and Google Privacy & Terms.
Your consent to our cookies if you continue to use this website.

Ads by Google

Ppt on pre ignition and detonation Ppt on acid-base titration problems and solutions Ppt on sources of energy for class 8th sample Ppt on seven segment display pin Ppt on conceptual art movement Ppt on cartesian product symbol Ppt on topic search engine Ppt on bluetooth applications for pc Ppt on finfet technology Ppt on standing order act score