We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byYolanda Lancaster
Modified about 1 year ago
48th International Conference EIPBN 2004, San Diego CA Gas Delivery and Virtual Process Chamber Concept for Gas Assisted Material Processing in Focused Ion Beam System ©Valery Ray VALERYRAY@AOL.COM 1 EIPBN 2004 Presentation, San Diego CA
©Valery Ray 2 EIPBN 2004 Presentation, San Diego CA Gas Assisted Etching (GAE) Rate Versus Pressure Relationship in FIB GAE Process Optimal precursor gas pressure for Cl 2 -assisted etching of GaAs by FIB is in 20 mTorr range. Optimal precursor gas pressure for XeF 2 -assisted etching of Si by FIB is in 1 mTorr range. Etching rate with Cl 2 declines rather rapidly as pressure drops below or rises above the optimal. Results by Y. Ochiai et al., “Pressure and Irradiation Angle Dependence of Maskless Ion Beam Assisted Etching of GaAs and Si”, J. Vac. Sci. Technol. B 3(1), 1985, pp. 67 - 70 FIB systems operate at 10 -5 Torr (or less) pressure level in main chamber (N. Bassom et al., “Modeling and Optimizing XeF2-enhanced milling of Silicon”, Proceedings ISTFA 1999, p. 257).
©Valery Ray 3 EIPBN 2004 Presentation, San Diego CA Gas Delivery Within High Aspect Ratio (HAR) Via Round Via Gas Transmission Probability Aspect Ratio of the Via Plot of Ref.  data
©Valery Ray 4 EIPBN 2004 Presentation, San Diego CA Historical Methods of Gas Delivery in Focused Particle Beam Systems Hypodermic needles (J. Orloff, “Charged Particle Optics”, CRC Press, 1997, pp. 441 - 453, and elsewhere) Separately pumped gas-filled sample chamber (Y. Ochiai et al, “Pressure and Irradiation Angle Dependence of Maskless Ion Beam Assisted Etching of GaAs and Si” J. Vac. Sci. Technol. B 3(1) 1985) EBD cell with vapor source (A. Filch et al, “High-vacuum versus “environmental” electron beam deposition” J. Vac. Sci. Technol. B 14(4) 1996) “ Shroud ” or “ Beehive ” gas concentrator (Casella et al, 1998, US Patent 5,851,413 and Libbi et al, 2002, US Patent 6,497,194)
©Valery Ray 5 EIPBN 2004 Presentation, San Diego CA Limitations of Shroud (“ Beehive ”) Concentrator Long inverted funnel-like shroud absorbs most of secondary electron signal, resulting in poor image S/N.
©Valery Ray 6 EIPBN 2004 Presentation, San Diego CA Electrostatic Bias of Shroud Concentrator Introduction of shroud concentrator significantly reduces level of secondary electron signal available for detection. Experiments demonstrated that image contrast is improved by factor of eight by positive electrostatic bias applied to shroud (“ Beehive ”) concentrator, however S/N remains low. Results by V. Ray et al., “Improvements of Secondary Electron Imaging and Endpoint Detection…” ISTFA 2003 Conference Proceedings, Santa Clara, CA pp. 338-342.
©Valery Ray 7 EIPBN 2004 Presentation, San Diego CA Effects of Vacuum Vessel Geometry on Gas Concentration Theoretical calculations predict localized gas concentration in the center of a spherical vacuum vessel by factor of seven larger then at the vessel walls. Attempt of experimental measurements of the distribution indicated uniform pressure distribution within the vessel. Results by S. B. Nesterov et al., “Influence of the vacuum chamber shape on the non-uniformity of gas distribution”, Vacuum 53 (1999) pp. 193-196.
©Valery Ray 8 EIPBN 2004 Presentation, San Diego CA Virtual Processing Chamber Concept Image From the EIPBN 2004 Abstract Gas Delivery Tube Virtual Chamber Gap Sample surface
©Valery Ray 9 EIPBN 2004 Presentation, San Diego CA Proposed “ Cupola ” Nozzle Sample surface Gas Delivery Tube Virtual Chamber Gap 250 m 1 mm 500 m
©Valery Ray 10 EIPBN 2004 Presentation, San Diego CA Electron-Optics Simulations of “ Cupola ” and “ Shroud ” or “ Beehive ” Nozzles “Cupola” nozzle biased +200V transmits 94.5% of secondary electrons. “Shroud” or “Beehive” nozzle biased +200V transmits 30% of secondary electrons. Same 2 mm width of bottom opening is simulated for both nozzles.
©Valery Ray 11 EIPBN 2004 Presentation, San Diego CA Electron-Optics Simulations of “ Cupola ” and “ Shroud - Beehive ” Nozzles
©Valery Ray 12 EIPBN 2004 Presentation, San Diego CA Conclusions 1. Proposed virtual processing chamber concept and “CUPOLA” nozzle geometry provide viable advancement of gas delivery techniques for charged particle beam processing apparatus. The concept is equally applicable to Focused Ion Beam (FIB) and Focused Electron Beam (FEB) systems. 2. Secondary electron trajectory simulations of “CUPOLA” geometry suggest that ~80% of the signal could be extracted through the “CUPOLA” under 100V positive bias conditions, a X4 improvement comparatively to the existing “Beehive” nozzles. 3. Secondary electron trajectory simulations suggest that a significant fraction of the secondary electrons are absorbed by the gas delivery nozzles and set a basis for experimental investigation of possibility to monitor the nozzle current for the purposes of detecting material-dependent transitions in a secondary electron emission.
©Valery Ray 13 EIPBN 2004 Presentation, San Diego CA Ongoing and Future work 3D Simulations of “ CUPOLA” nozzle, including reduced dimensions and effects of bias on primary ion and electron beams. Technology suitable for manufacturing of “ CUPOLA ” nozzle geometry? Micro-machining, electroforming, laser, or FIB processing? Experimental monitoring of nozzle current during the milling of HAR via could be of high practical interest for detection of material-dependent transitions in secondary electron emission.
©Valery Ray 14 EIPBN 2004 Presentation, San Diego CA R E F E R E N C E S 1. J.F. O’Hanlon “A Users Guide to Vacuum Technology” John Wiley & Sons, 1989. 2. J. Orloff, M. Utlaut, L. Swanson “High Resolution Focused Ion Beams, FIB and Its Applications” Kluwer Academic Publishers, 2003. 3. Casella et al., US Patent 5, 851, 413 “Gas Delivery Systems for Particle Beam Processing”, 1998 4. Other publications as referenced throughout the presentation
©Valery Ray 15 EIPBN 2004 Presentation, San Diego CA Acknowledgements Author would like to thank Mr. Nicholas Antoniou, Dr. Neil Bassom, Mr. Gavin Hartigan and Dr. Kun Liu, all of FEI Company, for there’s contributions to the abstract of this presentation.
©Valery Ray 16 EIPBN 2004 Presentation, San Diego CA Acknowledgements Special thanks are due to INTEGRATED ENGINEERING SOFTWARE company from Winnipeg, Canada, for providing evaluation copy of LORENZ-2E particle trajectory simulation software. WWW.INTEGRATEDSOFT.COM
Valery Ray Particle Beam Systems & Technology, Methuen, USA High Aspect Ratio Via Milling Endpoint Phenomena in Focused Ion Beam.
Valery Ray Particle Beam Systems & Technology, Methuen, USA High-Throughput Milling of HAR Vias with Concentrated Gas Delivery.
Valery Ray Particle Beam Systems & Technology, Methuen, USA Fluorocarbon Precursor for High Aspect Ratio Via Milling in Focused.
02/24/ th FIB/SEM User Group Meeting, Washington DC Valery Ray Developing FIB GAE Recipes: Practical Application of “Unfinished.
Valery Ray Particle Beam Systems & Technology, Methuen, USA CAD – less Blind Navigation in Focused Ion Beam System PBS&T Chris.
Design and Simulation of a MEMS Piezoelectric Micropump Alarbi Elhashmi, Salah Al-Zghoul, Advisor: Prof. Xingguo Xiong Department of Biomedical Engineering,
Quantification Techniques Under Dynamic SIMS 1 Secondary Ion Mass Spectrometer Start-Up Project Overview Secondary Ion Mass Spectrometry Overview MiniSIMS:
J. Tinslay 1, B. Faddegon 2, J. Perl 1 and M. Asai 1 (1) Stanford Linear Accelerator Center, Menlo Park, CA, (2) UC San Francisco, San Francisco, CA Verification.
Effect of Re Alloying in W on Surface Morphology Changes After He + Bombardment at High Temperatures R.F. Radel, G.L. Kulcinski, J. F. Santarius, G. A.
High Intensity Polarized Electron Gun Studies at MIT-Bates 10/01/2008 PESP Evgeni Tsentalovich MIT.
Abstract A time resolved radial profile neutron diagnostic is being designed for the National Spherical Torus Experiment (NSTX). The design goal is to.
ILE, Osaka Concept and preliminary experiment on protection of final optics in wet-wall laser fusion reactor T. Norimatsu, K. Nagai, T. Yamanaka and Y.
Focused ion beam (FIB) 1.Overview. 2.Ion source and optics. 3.Ion-solid interaction, damage. 4.Scanning ion beam imaging. ECE 730: Fabrication in the nanoscale:
Copyright © 2004 by Chem-Space Associates, Inc. ALL RIGHTS RESERVED Atmospheric Pressure Focusing Edward W Sheehan, Ross C Willoughby, David F. Fries Chem-Space.
Tomsk Polytechnic University1 A.S. Gogolev A. P. Potylitsyn A.M. Taratin.
05/20/ High-Current Polarized Source Developments Evgeni Tsentalovich MIT.
L.B. Begrambekov Plasma Physics Department, Moscow Engineering and Physics Institute, Moscow, Russia Peculiarities, Sources and Driving Forces of.
F. Cheung, A. Samarian, W. Tsang, B. James School of Physics, University of Sydney, NSW 2006, Australia.
MOLIBDENUM MIRRORS WITH COLUMN NANOGRAIN REFLECTING COATING AND EFFECT OF ION- STIMULATED DIFFUSION BLISTERRING RRC «Кurchatov Institute» А.V. Rogov, К.Yu.Vukolov.
DESIGN OF 2.2-INCH LED BACKLIGHT UNITS FOR BRIGHTNESS UNIFORMITY AND REDUCTION OF HOT-SPOT EFFECT Jeng-Feng Lin, Chin-Chieh Kang, and Chih-Yang Liu Department.
March 28-April, Particle Acceleratior Conference - New York, U.S.A. Comparison of back-scattering properties of electron emission materials Abstract.
CJ Barton Department of Physics INTAG Meeting – GSI – May 2007 Large Acceptance Bragg Detector at ISOLDE.
Laser etching of GaN Jonathan Winterstein Dr. Tim Sands, Advisor.
III. Results and Discussion In scanning laser microscopy, the detected voltage signal V(x,y) is given by where j b (x,y) is the local current density,
OS12A-141 Comparison of Monte Carlo Model Predictions with Tank Beam Spread Experiments Using a Maalox Phase Function Obtained with Volume Scattering Function.
Who are we? Charged Particle Optics Group. What we want 1. pattern dimensions: nm-scale single atom! 2. any material.
ARIES-IFE Study L. M. Waganer, June 7, 2000 Page 1 Considerations of HI Beam and Vacuum System Arrangement L. Waganer The Boeing Company 7 June 2001 ARIES.
High-Resolution Visible Spectroscopy of H 3 + Christopher P. Morong, Christopher F. Neese and Takeshi Oka Department of Chemistry, Department of Astronomy.
Submillimeter absorption spectroscopy in semiconductor manufacturing plasmas and comparison to theoretical models Yaser H. Helal, Christopher F. Neese,
Development of a Gamma-Ray Beam Profile Monitor for the High-Intensity Gamma-Ray Source Thomas Regier, Department of Physics and Engineering Physics University.
STOCHASTIC DEFECT DETECTION FOR MONTE-CARLO FEATURE PROFILE MODEL * MIPSE Graduate Symposium 2015 Chad Huard and Mark Kushner University of Michigan, Dept.
Mg Films Grown by Pulsed Laser Deposition as Photocathodes: QE and surface adsorbates L. Cultrera INFN – National Laboratories of Frascati.
Corial 200 COSMA Software with: Edit menu for process recipe edition, Adjust menu for process optimizing, Maintenance menus for complete equipment.
BASIC CONCEPT reflecting layer glass capillary wall CENTER LINE optically active layers polyimide.
2. RUTHERFORD BACKSCATTERING SPECTROMETRY Basic Principles.
MiniSIMS Secondary Ion Mass Spectrometer Dr Clive Jones Millbrook Instruments Limited Blackburn Technology Centre, England
1 14 Design of Experiments with Several Factors 14-1 Introduction 14-2 Factorial Experiments 14-3 Two-Factor Factorial Experiments Statistical analysis.
Questions/Problems on SEM microcharacterization Explain why Field Emission Gun (FEG) SEM is preferred in SEM? How is a contrast generated in an SEM? What.
1 Ab initio and Infrared Studies of Carbon Dioxide Containing Complex Zheng Su and Yunjie Xu Department of Chemistry, University of Alberta, Edmonton,
Influence of oxygen content on the 1.54 μm luminescenceof Er-doped amorphous SiO x thin films G.WoraAdeola,H.Rinnert *, M.Vergnat LaboratoiredePhysiquedesMate´riaux.
Other modes associated with SEM: EBIC In this technique, the incident electron beam is used to create electron-hole pairs that constitute the current under.
Scanning Electron Microscope (SEM) Major components: 1. Vacuum system 2. Electron beam generation system 3. Electron beam manipulation system 4. Beam specimen.
Fusion Technology Institute 4/5/2002HAPL1 Ion Driven Fireballs: Calculations and Experiments R.R. Peterson, G.A. Moses, and J.F. Santarius University of.
Characterization of the Spatial Resolution and the Gamma Discrimination of Helium-3 Tubes ESS Detector Group Seminar Edoardo Rossi 14 th August 2015.
Dr. Hao Zhang RECENT DEVELOPMENT OF THE SUPERSONIC GAS-JET BEAM PROFILE MONITOR.
Report 8 Liquid light guide 2016/6/3 Reporter: 儲君宇 Joshua Jun-Yu Chu 0.
Conclusions Despite being able to acquire successfully high resolution images of radiation dose, image quality is limited by hardware problems. Upgrading.
MICRO-STRIP METAL DETECTOR FOR BEAM DIAGNOSTICS PRINCIPLE OF OPERATION Passing through metal strips a beam of charged particles or synchrotron radiation.
FIB User Group, Washington DC, USA Valery Ray PBS&T FREUD Applications of FIB Invasive FIB Attacks and Countermeasures in Hardware.
© 2017 SlidePlayer.com Inc. All rights reserved.