We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byYolanda Lancaster
Modified about 1 year ago
48th International Conference EIPBN 2004, San Diego CA Gas Delivery and Virtual Process Chamber Concept for Gas Assisted Material Processing in Focused Ion Beam System ©Valery Ray 1 EIPBN 2004 Presentation, San Diego CA
©Valery Ray 2 EIPBN 2004 Presentation, San Diego CA Gas Assisted Etching (GAE) Rate Versus Pressure Relationship in FIB GAE Process Optimal precursor gas pressure for Cl 2 -assisted etching of GaAs by FIB is in 20 mTorr range. Optimal precursor gas pressure for XeF 2 -assisted etching of Si by FIB is in 1 mTorr range. Etching rate with Cl 2 declines rather rapidly as pressure drops below or rises above the optimal. Results by Y. Ochiai et al., “Pressure and Irradiation Angle Dependence of Maskless Ion Beam Assisted Etching of GaAs and Si”, J. Vac. Sci. Technol. B 3(1), 1985, pp FIB systems operate at Torr (or less) pressure level in main chamber (N. Bassom et al., “Modeling and Optimizing XeF2-enhanced milling of Silicon”, Proceedings ISTFA 1999, p. 257).
©Valery Ray 3 EIPBN 2004 Presentation, San Diego CA Gas Delivery Within High Aspect Ratio (HAR) Via Round Via Gas Transmission Probability Aspect Ratio of the Via Plot of Ref.  data
©Valery Ray 4 EIPBN 2004 Presentation, San Diego CA Historical Methods of Gas Delivery in Focused Particle Beam Systems Hypodermic needles (J. Orloff, “Charged Particle Optics”, CRC Press, 1997, pp , and elsewhere) Separately pumped gas-filled sample chamber (Y. Ochiai et al, “Pressure and Irradiation Angle Dependence of Maskless Ion Beam Assisted Etching of GaAs and Si” J. Vac. Sci. Technol. B 3(1) 1985) EBD cell with vapor source (A. Filch et al, “High-vacuum versus “environmental” electron beam deposition” J. Vac. Sci. Technol. B 14(4) 1996) “ Shroud ” or “ Beehive ” gas concentrator (Casella et al, 1998, US Patent 5,851,413 and Libbi et al, 2002, US Patent 6,497,194)
©Valery Ray 5 EIPBN 2004 Presentation, San Diego CA Limitations of Shroud (“ Beehive ”) Concentrator Long inverted funnel-like shroud absorbs most of secondary electron signal, resulting in poor image S/N.
©Valery Ray 6 EIPBN 2004 Presentation, San Diego CA Electrostatic Bias of Shroud Concentrator Introduction of shroud concentrator significantly reduces level of secondary electron signal available for detection. Experiments demonstrated that image contrast is improved by factor of eight by positive electrostatic bias applied to shroud (“ Beehive ”) concentrator, however S/N remains low. Results by V. Ray et al., “Improvements of Secondary Electron Imaging and Endpoint Detection…” ISTFA 2003 Conference Proceedings, Santa Clara, CA pp
©Valery Ray 7 EIPBN 2004 Presentation, San Diego CA Effects of Vacuum Vessel Geometry on Gas Concentration Theoretical calculations predict localized gas concentration in the center of a spherical vacuum vessel by factor of seven larger then at the vessel walls. Attempt of experimental measurements of the distribution indicated uniform pressure distribution within the vessel. Results by S. B. Nesterov et al., “Influence of the vacuum chamber shape on the non-uniformity of gas distribution”, Vacuum 53 (1999) pp
©Valery Ray 8 EIPBN 2004 Presentation, San Diego CA Virtual Processing Chamber Concept Image From the EIPBN 2004 Abstract Gas Delivery Tube Virtual Chamber Gap Sample surface
©Valery Ray 9 EIPBN 2004 Presentation, San Diego CA Proposed “ Cupola ” Nozzle Sample surface Gas Delivery Tube Virtual Chamber Gap 250 m 1 mm 500 m
©Valery Ray 10 EIPBN 2004 Presentation, San Diego CA Electron-Optics Simulations of “ Cupola ” and “ Shroud ” or “ Beehive ” Nozzles “Cupola” nozzle biased +200V transmits 94.5% of secondary electrons. “Shroud” or “Beehive” nozzle biased +200V transmits 30% of secondary electrons. Same 2 mm width of bottom opening is simulated for both nozzles.
©Valery Ray 11 EIPBN 2004 Presentation, San Diego CA Electron-Optics Simulations of “ Cupola ” and “ Shroud - Beehive ” Nozzles
©Valery Ray 12 EIPBN 2004 Presentation, San Diego CA Conclusions 1. Proposed virtual processing chamber concept and “CUPOLA” nozzle geometry provide viable advancement of gas delivery techniques for charged particle beam processing apparatus. The concept is equally applicable to Focused Ion Beam (FIB) and Focused Electron Beam (FEB) systems. 2. Secondary electron trajectory simulations of “CUPOLA” geometry suggest that ~80% of the signal could be extracted through the “CUPOLA” under 100V positive bias conditions, a X4 improvement comparatively to the existing “Beehive” nozzles. 3. Secondary electron trajectory simulations suggest that a significant fraction of the secondary electrons are absorbed by the gas delivery nozzles and set a basis for experimental investigation of possibility to monitor the nozzle current for the purposes of detecting material-dependent transitions in a secondary electron emission.
©Valery Ray 13 EIPBN 2004 Presentation, San Diego CA Ongoing and Future work 3D Simulations of “ CUPOLA” nozzle, including reduced dimensions and effects of bias on primary ion and electron beams. Technology suitable for manufacturing of “ CUPOLA ” nozzle geometry? Micro-machining, electroforming, laser, or FIB processing? Experimental monitoring of nozzle current during the milling of HAR via could be of high practical interest for detection of material-dependent transitions in secondary electron emission.
©Valery Ray 14 EIPBN 2004 Presentation, San Diego CA R E F E R E N C E S 1. J.F. O’Hanlon “A Users Guide to Vacuum Technology” John Wiley & Sons, 2. J. Orloff, M. Utlaut, L. Swanson “High Resolution Focused Ion Beams, FIB and Its Applications” Kluwer Academic Publishers, 3. Casella et al., US Patent 5, 851, 413 “Gas Delivery Systems for Particle Beam Processing”, 1998 4. Other publications as referenced throughout the presentation
©Valery Ray 15 EIPBN 2004 Presentation, San Diego CA Acknowledgements Author would like to thank Mr. Nicholas Antoniou, Dr. Neil Bassom, Mr. Gavin Hartigan and Dr. Kun Liu, all of FEI Company, for there’s contributions to the abstract of this presentation.
©Valery Ray 16 EIPBN 2004 Presentation, San Diego CA Acknowledgements Special thanks are due to INTEGRATED ENGINEERING SOFTWARE company from Winnipeg, Canada, for providing evaluation copy of LORENZ-2E particle trajectory simulation software.
Joint meeting of SEWG 05 and 07 of EU TF PWI, Ljubljana, Slovenia, October 2009 In-situ studies of H/D interaction with W, Ta and Cu Progress report.
X – Ray methods of Analysis : X – Absorption spectroscopy. X – Ray Fluorescence Spectroscopy. X – Ray Diffractometry.
1 Pier Francesco Manfredi Lawrence Berkeley National Laboratory - Berkeley, California (USA) Dipartimento di Fisica dellUniversità di.
ITRS Summer Public Conference, July 15, San Francisco, CA, USA 1 Modeling and Simulation ITWG Jürgen Lorenz – Fraunhofer IISB – Chairperson M&S ITWG ITWG/TWG.
Plasma Window Options and Opportunities for Inertial Fusion Applications Leslie Bromberg Ady Herskovitch* MIT Plasma Science and Fusion Center ARIES meeting.
International Technology Roadmap for Semiconductors Metrology Roadmap 2001 Update EuropeAlain Deleporte (ST)4/01 Alec Reader (Philips Analytical) Vincent.
ITRS Defect Reduction Technology Christopher Long, ISMT/IBM Milt Godwin, AMAT ITRS 2000 Conference July16-18, 2001 San Francisco,CA International Technology.
Part 1 :Light and material بسم الله الر حمن الرحيم
Annual CAVIAR Science Meeting Dec 2009 Annual CAVIAR Science Meeting Dec 2009 Update on Leicester Lab Studies Mark Watkins, Alex Shillings.
tool to prioritize other instrument campaigns (eg, Incoherent Scatter Radar). The final reconstruction used a quiet profile a priori.
An acoustic resonator to enhance the sensing of cells within micro-fluidic chambers R. J. Townsend 1, M. Hill 1, N. R. Harris 2, S. P. Stacey 3, L. A.
ITRS Winter Public Conference, December 9, Seoul, Korea 1 Modeling and Simulation ITWG Jürgen Lorenz – Fraunhofer IISB – Chairperson M&S ITWG ITWG/TWG.
FRCR: Physics Lectures Diagnostic Radiology Lecture 5 Digital detectors, tomography and dual energy imaging Dr Tim Wood Clinical Scientist.
IAEA Radiation Protection in Paediatric Radiology Radiation Protection of Children in Digital Radiography L04.
Introduction to XRF LearnXRF.com Introduction to X-Ray Fluorescence X-Ray FluorescenceAnalysis.
R.Zagórski, PWI-TF Annual Meeting, Frascati, October 2008 EFDA Plasma Edge Technology Programme Monitoring of 2008 Activities Roman Zagórski European Fusion.
ITRS Summer Public Conference, July 14, 2010, San Francisco, CA, USA 1 Modeling and Simulation ITWG Bert Huizing – NXP – European M&S ITWG and IRC member.
DR. Halima Hawesa RAD 466 Lecture 6. Presentation Out Lines PET/CT System PET/CT System Design PET/CT Instrumentations PET/CT Imaging Method New Advances.
DESIGN FOR SEMICONDUCTOR RELIABILITY George Denes, Dipl.Eng. Senior Semiconductor Reliability Consultant.
© 2007, TSI Incorporated Future Diesel Emission Limits: Particle Measurement Programme (PMP) Update 2008 Channel Partner Sales Meeting.
An Image Filtering Technique for SPIDER Visible Tomography N. Fonnesu M. Agostini, M. Brombin, R.Pasqualotto, G.Serianni 3rd PhD Event- York- 24th-26th.
Transmission Electron Microscopy of Mineralogy Wen-An Chiou Materials Characterization Center (MC 2 ) and Department of Chemical Engineering and Materials.
Measuring innovation TRAINING WORKSHOP ON SCIENCE, TECHNOLOGY AND INNOVATION INDICATORS Cairo, Egypt September 2009.
INFM e Physics Department, University of di Padova, Via Marzolo 8, Padova, Italy SIMS Secondary Ion Mass Spectrometry Spettrometria di massa a ioni.
Measuring the Effects of Unit Nonresponse in Establishment Surveys Clyde Tucker and John Dixon U.S. Bureau of Labor Statistics David Cantor Westat.
Copper Germanium Nanowires: Synthesis, Fabrication, and Applications Jimmy Kan, Keith Chan, Zhonglin Wang, Deli Wang UCSD Department of Electrical and.
Techniques For Troubleshooting FCC Regenerator Problems.
University of Notre Dame Chemical & Biomolecular Engineering Laboratories Senior and Junior laboratory courses.
1 PROBING THE GAS PHASE CHEMISTRY INVOLVED IN DIAMOND CHEMICAL VAPOUR DEPOSITION (CVD) Mike Ashfold School of Chemistry University of Bristol Bristol BS8.
Putting it all together - Particle Detectors Writeup for 3 rd section:
© 2016 SlidePlayer.com Inc. All rights reserved.