Presentation on theme: "Astronomical Observational Techniques and Instrumentation"— Presentation transcript:
1 Astronomical Observational Techniques and Instrumentation RIT Course NumberProfessor Don FigerSpectral resolution, wavelength coverage, the atmosphere and background sources
2 Aims and outline for this lecture describe system requirements for spectral resolution and wavelength coveragedescribe atmospheric effects on observationssummarize primary background sources and effects on observations
3 Need for Spectral Information “A picture is worth a thousand words.” --Barnard“A spectrum is worth a thousand pictures.” --an astronomerA spectrum is the distribution of flux versus wavelength.Spectra are critical for making measurements of physical properties of astronomical objects.low resolution spectra reveals spectral energy distribution (SED)high resolution spectra reveals emission/absorption features
4 Spectrum of ArcturusThis illustration shows the complete visible spectrum of the giant star Arcturus. By passing the star's light through a prism, the light is spread out into its component colors, displayed here wrapped around row after row from red at top left to blue at bottom right. Each of the 50 rows covers 8 nanometers in wavelength, for complete coverage of the visible spectrum from 300 to 700 microns. The many dark lines are caused by light from the hot star being absorbed by specfic atoms and ions in the star's cooler atmoshpere. (NOAO)
5 High Resolution Spectroscopy High resolution spectroscopy can yieldvelocityabundancestemperaturevirial mass (through integrated light)molecular excitationinterstellar absorptionionization stateothersThe example on the right shows high resolution infrared spectra of red supergiants.Note the many CO rotational-vibrational absorption lines.The slight translations in the wavelength axis from star to star shows velocity differences.Weaker features to the left reflect atomic absorption and can be used for abundance analysis.velocity differenceCO bandheadatomic absorption
6 Low Resolution Spectroscopy Low resolution spectroscopy can yieldemission mechanism (via spectral index)temperaturemolecular contentinterstellar molecular absorptionThe example on the right shows low resolution infrared spectra of red supergiants.Note the CO bandhead.The depth of the bandhead indicates that these stars are cool.For this application, it might be just as well to obtain a high resolution spectrum over this waveband.In some cases, a low resolution spectrum can reveal curvature over a broad range of wavelengths – something difficult to detect in high resolution spectra.CO bandhead
7 Spectrograph/spectrometer A spectrograph/spectrometer is an instrument that can measure intensity versus wavelength.dispersiveprismgratingnondispersivefilters, circular variable filter (CVF)Fabry-Perot
8 Spectroscopy System Design Criteria Efficiency: amount of light directed into the spectrumResolution: minimum spatial and spectral separation between resolved featuresScattering: light directed out of the spectrum or in undesired locations in the spectrumStability and calibration accuracy: ability of system to maintain or reproduce wavelength versus pixel relationshipBackground: light from all sources other than target, might be sky, telescope, optics, baffles, etc.Packaging: compactness of layout
9 Spectrograph CartoonSpectroscope Diagram of a simple spectroscope. A small slit in the mask on the left allows a narrow beam of light to pass. The light passes through a prism and is split up into its component colors. The resulting spectrum can be viewed through an eyepiece or simply projected onto a screen.
10 Continuum and Emission Lines Continuous and Emission Spectra When passed through a slit and split up by a prism, light from a source of continuous radiation (a) gives rise to the familiar rainbow of colors. By contrast, the light from excited hydrogen gas (b) consists of a series of distinct bright spectral lines called emission lines. (The focusing lenses have been omitted for clarity.)Elemental Emission The emission spectra of some well-known elements. In accordance with the convention adopted throughout this text, frequency increases to the right. (Wabash Instrument Corp.)
11 Absorption LinesAbsorption Spectrum When cool gas is placed between a source of continuous radiation, such as a hot lightbulb, and a detector, the resulting spectrum consists of a continuous spectrum crossed by a series of dark absorption lines. These lines are formed when the intervening gas absorbs certain wavelengths (colors) from the original beam. The absorption lines appear at precisely the same wavelengths as the emission lines that would be produced if the gas were heated to high temperatures.Sodium Spectrum (a) The characteristic emission lines of sodium. The two bright lines in the center appear in the yellow part of the spectrum. (b) The absorption spectrum of sodium. The two dark lines appear at exactly the same wavelengths as the bright lines in the sodium emission spectrum.
12 Equivalent Width Equivalent width is a measure of flux in a line. It is independent of instrument parameters.Absorption Spectrum When cool gas is placed between a source of continuous radiation, such as a hot lightbulb, and a detector, the resulting spectrum consists of a continuous spectrum crossed by a series of dark absorption lines. These lines are formed when the intervening gas absorbs certain wavelengths (colors) from the original beam. The absorption lines appear at precisely the same wavelengths as the emission lines that would be produced if the gas were heated to high temperatures.Sodium Spectrum (a) The characteristic emission lines of sodium. The two bright lines in the center appear in the yellow part of the spectrum. (b) The absorption spectrum of sodium. The two dark lines appear at exactly the same wavelengths as the bright lines in the sodium emission spectrum.
13 Spectroscopy System Design Form Example graphic shows simple layout for spectrographgrating normalgratingcollimatorslittelescopedetectorlens
14 Spectral ResolutionSpectral resolution is a measure of the ability to separate nearby features in wavelength space.Delta lambdaoften set to the full-width at half-maximum of an unresolved linecan be measured in the datadepends on data analysiscan be limited by diffraction, slit width, detector sampling
15 Spectral Resolution: Prism A prism disperses light with a transmissive optic that has chromatic index of refraction.
16 Gratings: Grating Equation The grating equation gives the geometry for constructive interference between facets of a grating. The equation can be derived by setting the optical path difference between light scattering from two adjacent facets to an integer multiple of the wavelength.In more standard nomenclature,where, m is the order number and T is the groove density.Note that the equation reduces to the law of reflection for m=0.
17 Gratings: Angular Resolution Angular resolution describes the variation in output angle versus wavelength. It can be found by differentiating the grating equation.Note that the angular resolution can be increased by increasing the output angle.For a fixed wavelength, angular resolution is dependent on geometry.
18 Gratings: Diffraction-limited Resolution The diffraction-limited resolution of a grating is simply Nm, where N is the number of illuminated grooves and m is the order number.
19 Gratings: Slit-limited Resolution The resolution, or resolving power, is usually limited by slit width. (“primed” quantity is at image plane)
20 Gratings: Diffraction-limited Image Imagine that we set the slit width to be equal to the full width at half maximum of a diffraction-limited spot. (Assume Littrow).Note that the resolution can be increased by increasing the size of the collimated beam (and size of grating), increasing the output angle, or decreasing the wavelength (assuming that the decrease in wavelength is accommodated by a decrease in the spot size and decrease in slit width).
21 Gratings: Increasing Resolution From the previous equations, it appears that resolution can be increased by:decreasing the slit width (which might decrease the transmitted flux)increase the collimator focal length (which will increase the beam size and required size of the grating)increase the output angle (which will increase the required size of the grating)There is another possibility that is not apparent from the grating equation, given that its derivation assumed that the index of refraction of the material surrounding the grating was one.A more general derivation would show that the realized angular dispersion is multiplied by n.This effect is used in an immersion gratingSilicon is a particularly useful material for this purpose because it has a high index (n~4), although it is only transmissive in the infrared.
22 Gratings: Immersion Grating Resolution The resolution can be increased by immersing the grating surface in a high index material.Note that angular dispersion is n times greater than for case without immersion.
23 Gratings: Echelle Grating Angular dispersion is dramatically increased when using an “echelle” grating.In this case, the output angle is typically >60 degrees.Because an echelle is used at a large angle, it must be rectangular.The grating facets must be ruled at a large angle, because:In order to maintain high efficiency, the input and output angles should be the same (“Littrow condition”).In order to provide high resolution, the output angle must be large.
24 Gratings: Echellogram This “echellogram” shows the data format on the detector in the HIRES optical spectrograph on Keck. The solar spectrum shows numerous absorption lines. Note in the inset photograph the very large dimension of the grating. The resolution depends on slit width and is in the range of 25, ,000.HIRES order format showing the solar spectrum(note that color was artificially added for illustrative purposes)
26 Fabry-Perot EtalonA Fabry-Perot etalon is a filter that selectively transmits a narrow range of wavelengths.Like a thin film, it transmits the most flux at wavelengths that satisfy the thin film condition.Finesse is a function of reflectivity.Path length for T1 is t/n and that for T2 is 3*t/n. So, the path difference is 2t. Setting this to m*lambda for constructive interference for the transmitted beam, gives 2t/n=m*lambda/n, or lambda=2tn/m.F=finesse= 4*R/(1-R)^2.
27 Wavelength Coverage: Free Spectral Range Multiple orders of the same wavelength range are dispersed to either side of the order of interest.The non-overlapping wavelength region in the order of interest is defined as the “free spectral range.” Outside of this range, light of different wavelengths in multiple orders overlap.order=ml1l2l1l2m-1mm+1FSRoverlapoverlapFSR=l/m
28 Atmospheric effects Absorption Emission Turbulence reduced flux from sourcedifficult calibrationsEmissionincreased background noisereduced integration timesdifficult calibrations (subtracting time-varying components)Turbulenceincreased object size (“seeing”)All effects vary with wavelength, time, altitude, line-of-sight
29 Atmospheric absorption Molecules are the dominant absorbers (H2O, CO2, Ox)Strong function of:wavelength, atmospheric wavebandstime, frequent calibration (hour timescales)altitude, mountain-top observing sitesline-of-sight, limited target access and frequent calibration
30 Atmospheric absorption versus l Sharp cutoffsdefined primarily by H2Oshape wavebandsHigher transmission between lines with higher resolutionCan introduce large calibration errors for low resolution observations (MNRAS, 1994, 266, 497)
31 Wavelength Regimes: near/mid-infrared: Atmospheric Transmission Like all other forms of electromagnetic radiation, infrared is utilised by astronomers to learn more about the universe. As infrared is essentially heat radiation, infrared telescopes (which include most major optical telescopes as well as a few dedicated infrared telescopes) need to have their detectors shielded from heat and chilled with liquid nitrogen in order to actually form images. This is particularly important in the mid infrared and far infrared regions of the spectrum. The principle limitation on infrared sensitivity from ground-based telescopes is the water vapour in the Earth's atmosphere, which absorbs a significant amount of infrared radiation. For this reason most infrared telescopes are built in very dry places at high altitude (above most of the water vapour in the atmosphere).
32 Wavelength Regimes: near-infrared: Atmospheric Transmission JHKLMLike all other forms of electromagnetic radiation, infrared is utilised by astronomers to learn more about the universe. As infrared is essentially heat radiation, infrared telescopes (which include most major optical telescopes as well as a few dedicated infrared telescopes) need to have their detectors shielded from heat and chilled with liquid nitrogen in order to actually form images. This is particularly important in the mid infrared and far infrared regions of the spectrum. The principle limitation on infrared sensitivity from ground-based telescopes is the water vapour in the Earth's atmosphere, which absorbs a significant amount of infrared radiation. For this reason most infrared telescopes are built in very dry places at high altitude (above most of the water vapour in the atmosphere).
33 Atmospheric absorption versus l - high res Array defectsþüCO2 absorption linesR = l/Dl ~ 23,000+l+lKeck II 10-m
34 Atmospheric absorption versus altitude Particle number densities (n) for most absorbers fall off rapidly with increasing altitude.Density of atmospheric constituents modeled as exponential.x0,H20 ~ 2 km, x0,CO2 ~ 7 km, x0,O3 ~ kmSo, 95% of atmospheric water vapor is below the altitude of Mauna Kea.
35 Atmospheric absorption versus airmass The amount of absorbed radiation depends upon the number of absorbers along the line of sightAM=1AM=2Atmosphere
37 Thermal emission spectral radiance, brightness, specific intensity: In = e cos q Bn(T) W m-2 Hz-1 sr-1qe º emissivity (dimensionless)Planck (blackbody) function:Peak in nBn or lBl:2hnc2 exp(hn/kT) - 1lmax(mm) =3674 KTBn(T) =2hcl5 exp(hc/lkT) - 1Bl(T) =lmax~10 mm for room temp.
38 Atmospheric emission: Blackbody Total power onto a detector:P = h AW Dn esky Bn(Tsky)h: transmission of all optics x Q.E.esky: emissivity of skyA: telescope areaW: solid angle subtended by focal plane apertureDn: bandwidthBn(Tsky): Planck functionAt 10 mm, typically: h ~ 0.2, e ~ 0.1, AW ~ 3x10-10 m2 SrDn ~ 1.5 x 1013 Hz (10 mm filter), T ~ 270 KP ~ 10-9 W or ~ 4 x 1010 g s-1
39 Atmospheric emission: A Note on Solid Angle Background from the sky can be calculated by multiplying the blackbody equation by the area of the sky seen by the detector and the solid angle subtended by the telescope aperture.The AW product is invariant in a system.AtelescopeAskydetectorDxdFq0q1It is easier to use area of telescope and solid angle of sky.
43 Atmospheric Turbulence Static atmosphere bends light (n= at 2 mm)Dynamic atmosphere distorts the wavefronta Orionis (“Betelguese”)Data from the William Herschel 4.2 metre Telescope in La Palma. Wavelength 689.3nm, Exposure time 30ms per frame. The data were taken with a simple Speckle Camera situated on the GHRIL platform of the telescope. The data sets were taken with the Speckle Imaging group at Cardiff University.
44 Atmospheric Turbulence A diffraction-limited point spread function (PSF) has a full-width at half-maximum (FWHM) of:In reality, atmospheric turbulence smears the image:At Mauna Kea, r0=0.2 m at 0.5 mm.“Isoplanatic patch” is area on sky over which phase is relatively constant.
51 Background - sources: Atmosphere ThermalOHThe average OH line intensity is approximately 25,000 g s-1 m-2 asec-2 mm-1. See Maihara et alThe continuum between lines is about 50 times lower than this value (in the H band).
52 Background - sources: Telescope - scattering MirrorsBaffle edges and wallsSecondary support
53 Background - sources: Astronomical Astronomical objects can be objects of interest or noise contributors, depending on the project.Sunlight, moonlightLight scattered by solar system dust (“zodiacal”)Light emitted (thermal) by solar system dust (“zodiacal”)Stars (especially in a crowded field)Light emitted by interstellar dust (“cirrus”)
56 Background and Signal-to-Noise Ratio Recall the expression for SNR.The noise is the quadrature sum of uncorrelated noise sources.Background noise is the quadrature sum of shot noise for each background source.