Quality Assurance Program Screening of filter material roll stock SEM –Visual evaluation –Pore size uniformity: subjective determination Alpha Spectrometry –Am-241/243; 50µg Ce –FWHM each peak <= 50 keV
A little background on filters Design and specification set on “removal” of particles from liquids. Nominal “pore size” related to the size of particles removed from the liquid. In alpha spec source prep the major consideration should be a uniform pore size to give the best source. “Shiny side” apparent on some types of filters indicating different performance each side.
“0.1 micron” pore size filter 1µ Surface pores > 0.1µ will occur Designed to remove 99.98% of particles >0.1µ Goal of Eichrom product is uniform pore size 15,000x
Two sides of the filter “Inside” and “Outside” of the Roll Stock (5,000x magnification) Inside Lot 292218Outside Lot 292218
Peak Resolution Evaluation FWHM (keV) Am-243 Outside Am-243 Inside Am-241 Outside Am-241 Inside 36294126 28223337 31383245 32353744 43423742 4341 Ave =34Ave =33Ave = 36Ave = 39 Std dev = 7Std dev = 8Std dev = 4Std dev = 8
External Product Evaluation Barry Stewart, Carlsbad EMRC –“These filters are superb for use in making sources for alpha spectroscopy.” Am 241/243 With and without substrate ASTM C1163-98, LaF 3 ppt., 50 µg La carrier “Glossy side” up and down
Summary Eichrom now supplies a 0.1µ polypropylene filter for alpha spec source prep Pore size uniformity is monitored by SEM analysis and performance is measured by FWHM of Am-243/241 (specification <= 50 keV.) Performance demonstrated at Eichrom and externally. Both sides of filter are equivalent: no need to “put the shiny side up.”
Your consent to our cookies if you continue to use this website.