Download presentation

Presentation is loading. Please wait.

Published byOswaldo Scragg Modified over 3 years ago

1
Prof. David R. Jackson Dept. of ECE Notes 16 ECE 5317-6351 Microwave Engineering Fall 2011 S -Parameter Measurements 1

2
2

3
Test cables Hewlett-Packard 8510 DUT Port 1 Port 2 DUT Vector Network Analyzer (VNA) 3

4
Network Analysis of VNA Measurement 4

5
S-Parameter Measurements Error boxes contain effects of test cables, connectors, couplers,… 5

6
S-Parameter Measurements (cont.) This is called “de-embedding.” 6

7
“Short, open, match” calibration procedure Calibration loads ConnectCalibration SC OC +1 Z00Z00 shortopenmatch Recall from Notes 15: 7

8
“ Thru-Reflect-Line (TRL)” calibration procedure Calibration (cont.) 8 This is an improved calibration method that involves three types of connections: 1)The “thru” connection, in which port 1 is directly connected to port 2. 2) The “reflect” connection, in which a load with an (ideally) large (but not necessarily precisely known) reflection coefficient is connected. 3) The “line” connection, in which a length of matched transmission line (with an unknown length) is connected between ports 1 and 2. The advantage of the TRL calibration is that is does not requires precise short, open, and matched loads. This method is discussed in the Pozar book (pp. 193-196).

9
Z -Parameter Extraction Assume a reciprocal and symmetrical waveguide or transmission-line discontinuity. Examples Waveguide post Microstrip gap We want to find Z 1 and Z 2 to model the discontinuity. 9

10
Plane of symmetry (POS) POS Z -Parameter Extraction (cont.) 10

11
Assume that we place a short or an open along the plane of symmetry. Z -Parameter Extraction (cont.) 11

12
The short or open can be realized by using odd- or even-mode excitation. Z -Parameter Extraction (cont.) T + - + + Odd mode excitation Even mode excitation Incident voltage waves 12 The even/odd-mode analysis is very useful in analyzing devices (e.g., using HFSS).

13
De-embeding of a Line Length We wish the know the reflection coefficient of a device under test (DUT), but the DUT is not assessable directly – it has an extra length of transmission line connected to it. 13

Similar presentations

Presentation is loading. Please wait....

OK

Notes 15 ECE Microwave Engineering

Notes 15 ECE Microwave Engineering

© 2018 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Ppt on sea level rise due Ppt on business environment nature concept and significance of names Ppt on 555 timer calculator Ppt on case study method of training Ppt on credit default swaps greece Ppt on power sharing in india download movies How to run ppt on ipad Ppt on as 14 amalgamation theory Ppt on carbon cycle in nature Ppt on cartesian product example