Presentation on theme: "Embedded Skin Diagnosis System SDME USER Manual For your beautiful and comfortable life, BOMTECH EMBEDDED SKIN DIAGNOSIS SYSTEM SDME www.bomtech.co.kr."— Presentation transcript:
Embedded Skin Diagnosis System SDME USER Manual For your beautiful and comfortable life, BOMTECH EMBEDDED SKIN DIAGNOSIS SYSTEM SDME www.bomtech.co.kr Copyright ⓒ 2001 BOMTECH.Co., Ltd. All right reserved.
[ Contents ] Ⅱ. How to use Ⅰ. Overview of SDME 1. Strengths 2. Composition 3. Connection 1. Date and Time setting 2. Registration of customer information 3. Start new test 4. Moisture and Sebum test 5. Pore size test 6. Wrinkle test 7. Pigmentation test 8. View the test result Ⅲ. Result of SDME 1. General result 2. Detailed result IV. Product Recommendation V. Purchase Record 1. Registration of product 2. Update of product information 3. Shopping cart
Ⅰ. Overview of SDME 1. Strengths 1. Accurate skin diagnosis is available, even though on make-up condition. 2. Vivid magnified skin photos by the scope 3. Accurate diagnosis through vivid skin photos 4. Dermis layer as well as the skin surface can be checked by the E-CAM. 5. Suitable treatment tips according to skin types are recommended with analyzed photos and detailed explanation of skin condition. 6. Powerful graphic image processing chart with detailed results. 7. User friendly man machine interface simple hardware. 8. Automatic sebum quantity calibration function with day time. 2. Composition ▣ New Embedded (built-in Skin Diagnosis) with adapter ▣ E-CAM (X1, X80) ▣ P-Sensor ▣ a Stylus pen
12 3. Connection Initial screen – Prepare a stylus pen. Then Select “SDME” with it Connect adapter to “DC 12V” After connection, switch ”ON” 1. Plug the camera (E-CAM) in “AV IN” 2. Plug the adapter in “DC 12V” 3. Switch on and PRESS the Button of P-sensor and E-cam. (if not, you can not see any test result and might lead malfunction of Embedded computer.) Connect camera to “AV IN” Press the buttons both of P-sensor and E-cam before operation of SDME
II. How to operate SDME 12 1. Date & Time setting Start screen – Click “Date / Time setting” button Date/Time setting – Check on the Numeric keypad. Start screen – Click “Diagnosis” button *** NOTICE Please check if you press the buttons both of P-sensor and E-cam before operation of SDME.
1221 2. Registration of Customer information Main screen – Check “Search” button Search screen – Check “ADD” button Automatically keyboard pops up, input the name, gender, telephone, mobile, birthday and memo on each line using the keyboard. – Then click “SAVE” button.
3. Start New Test 1. After customer registration, customer name creates automatically. Then, check the “New Test” button and enter the program. 2. To get an image, select a screen on the 4 divided screen and capture the images pointing on “CAM “ button. 1223 4. Sebum & Moisture test: U-ZONE TEST by P-Sensor 1. Check the make-up thickness before test. 2. Select the “U-ZONE TEST”. 3. Put the P-Sensor to U-ZONE on the face 4. Press the sensor button “Test” and wait until it beeps. 5. Bar Graph appears on the screen. Make-up checking None; Condition of no makeup Light : Applied Tonic, Essence, Moisturizer only (No make-up) Thick: Applied any kinds of Cream or full make-up.
1.Select the :T-ZONE TEST” 2.Put the P-Sensor to T-ZONE on the face. 3.Press the sensor button “Test” and wait until it beeps. 4.Bar graph appears on the screen. (Green bar –standard / Red bar – customer) 1223 4. Sebum & Moisture test: Sebum test by E-Cam 1. Select the screen and activate the E-Cam by clicking the freezer button. 2. Check the AUTO ‘ON’ 3. Select the “U-ZONE” and put the E-cam on the cheek of the face. 4. Press the “Mode” button (UV light) on the E-cam and press the “Freezer” button. 5. Then the value appears on the bottom part ‘SEBUM, U-ZONE’ 6. Select the “T-ZONE” and put the E-cam on the forehead on the face to check he sebum quantity. 7. Move to the next step “PORE” 4. Sebum & Moisture test: T-ZONE TEST by P-Sensor Focus control Image capture UV light conversion
5. Pore size test 1. Put the E-cam on the face and take a picture of pore (around the nose). 2. The captured Image is saved and scanned the pore size automatically 3. The value of pore appears with scanned image 12 6. Wrinkle test 1. Put the E-cam on the face and take a picture of wrinkle (around the eye zone or forehead). 2. The captured Image is saved and scanned the wrinkle size automatically 3. The value of wrinkle appears with scanned image
12 7. Pigmentation test 8. View the test result 1. Click the “Result” button after completed every test. 2. Pop-up window comes up, “Do you want to save the test result?” comes up- click “OK” 1. Put the E-cam on the face and take a picture of pigmentation (around the cheek). 2. The captured Image is saved and scanned the pigmentation automatically 3. The value of wrinkle appears with scanned image
III. Result 12 1. General result 1. Result I - Each result comes out according to the various skin condition. 2. Result II – Skin type displays on the map as picture
12 2. Detailed result Result III ~ Result VII – every test result comes out and Click “Cart” button
12 1. Registration of products 1. Products display on Cart screen according to the test result. 2. Click “Update of products” if you want to register products more. Click “ADD” button and open the pop-up window of product registration. IV. Product Recommendation
122 2. Update of Product information 1. Enter the descriptions of the products and save it. 2. You can add the picture with “OPEN” f or take an image with E-Cam by clicking the “CAM” 1.After product registration, select suitable products for each skin type and click the “Add to recommendation” (Each skin type has different products list as you registered). 2. Click “BACK” button and go to the shopping cart screen. Product registration is completed !
12 3. Shopping cart function 1. Products display on Cart screen and select products according to the skin type. 2. You can increase the quantity – double click on the “qty”. 3. “Edit Qty” pop-up window opens and increase the quantity of selected items. 4. Click “OK” button.
12 1. Purchase Record function 1. After complete the shopping of products, go back to main screen. 2. Click “Search” button and select the customer 3. Purchased products display and you can check all selected items for customer management. Ⅴ. Purchase Record
Manufacture and Sales BOMTECH Electronics Co.,Ltd. 2~3F, Yeon-san Bldg., 1547-15, Seocho 3-dong, Seocho –ku, Seoul, 137-073, Korea Global Business Dept. TEL: 822-523-8293 FAX: 822-523-8297 E-mail: email@example.com firstname.lastname@example.org Homepage: www.bomtech.co.kr www.bomtech.net Copyright ⓒ 2001 BOMTECH.Co., Ltd. All right reserved. For your beautiful and comfortable life, BOMTECH skin diagnosis system SD-Pro