Download presentation

Presentation is loading. Please wait.

Published byKarlee Massengill Modified over 2 years ago

2
Institute for Reference Materials and Measurements (IRMM) Geel, Belgium http://www.irmm.jrc.be http://www.jrc.cec.eu.int 230th ACS NM, Washington D.C., Aug 2005 Problems with the uncertainty budget of half-life measurements S. Pommé

3
230th ACS NM - T1/2 2 Half-life Measurements Half-life determination by following the decay of a radioactive source The problem of data discrepancy; examples Autocorrelation causes underestimation of the uncertainty Example: the decay of Fe-55

4
230th ACS NM - T1/2 3 Half-life of 109 Cd a selection of the ‘best’ data

5
230th ACS NM - T1/2 4 Half-life of 134 Cs

6
230th ACS NM - T1/2 5 Half-life of 54 Mn a selection of the ‘best’ data

7
230th ACS NM - T1/2 6 970 980 990 1000 1010 1020 Houtermans Lagoutine Hoppes Morel Karmalitsyn Schötzig Van Ammel Days Half-life of 55 Fe a selection of the ‘best’ data

8
230th ACS NM - T1/2 7 Why are uncertainties underestimated? ** by analogy with ‘It’s the economy, stupid!’ by James Carvil, advisor of Bill Clinton in the elections of 1992

9
230th ACS NM - T1/2 8 Example: decay of 55 Fe fitted decay curve measured activity T 1/2 =1005.0 d ± 1.4d

10
230th ACS NM - T1/2 9 Procedure ‘uncertainty from fit’ 1) calculate the 2 –value, ensuring that it assumes its expectation value = the number of degrees of freedom 2) adjust the half-life so that 2 increases by a value of one 3) adopt as the standard deviation the square root of the amount by which the half-life was varied Bevington, P.R. Data Reduction and Error Analysis for the Physical Sciences; McGraw-Hill Book Company, New York, USA, 1969

11
230th ACS NM - T1/2 10 Residuals 55 Fe add uncertainty component of 0.0175% 2 / =3.5 2 / =1.0

12
230th ACS NM - T1/2 11 Check of distribution of residuals BEFORE introducing extra uncertainty component AFTER 2 / =3.5 2 / =1.0

13
230th ACS NM - T1/2 12 Half-life and uncertainty from least-squares fit => fit underestimates uncertainty!

14
230th ACS NM - T1/2 13 Why?

15
230th ACS NM - T1/2 14 Residuals 55 Fe: blow up uncertainty only including counting statistics => does not fully account for spread of data

16
230th ACS NM - T1/2 15 Autocorrelated data are not random autocorrelation plot of the residuals

17
230th ACS NM - T1/2 16 Alternative Procedure Subdivide instabilities according to frequency (low, medium, high) Apply new (simplified) uncertainty formula: n= the number of occurances of the effect n=1 for medium and low frequencies Add all components as square sum

18
230th ACS NM - T1/2 17 Subdivide instabilities according to frequency noise, counting statistics, … => random effects geometrical reproducibility, ‘seasonal’ effects, short-lived impurity, … dead time, detector/source degradation, background subtraction, …

19
230th ACS NM - T1/2 18 T 1/2 from combination of all data T 1/2

20
230th ACS NM - T1/2 19 Realistic uncertainty

21
230th ACS NM - T1/2 20

22
230th ACS NM - T1/2 21 Residuals Ba-133 NIM A390 (1997) 267-273 data scatter exceeds uncertainty => unidentified HIGH FREQUENCY component

23
230th ACS NM - T1/2 22 Residuals Cs-134 NIM A390 (1997) 267-273 MEDIUM FREQUENCY instability => fit underestimates uncertainty

24
230th ACS NM - T1/2 23 Residuals Ce-144 NIM A390 (1997) 267-273 sign of LOW FREQUENCY deviation => fit tends to hide it

25
230th ACS NM - T1/2 24 Residuals Y-88 NIM A390 (1997) 267-273 only a few measurement data => LACK OF INFORMATION

26
230th ACS NM - T1/2 25 Residuals Cd-109 NIM A390 (1997) 267-273 OVERESTIMATION of measurement uncertainty => overestimates uncertainty of half-life?

27
230th ACS NM - T1/2 26 Residuals 99 Tc m electrometer range switching systematic deviation for short times ARI 60 (2004) 317-323

28
230th ACS NM - T1/2 27 Conclusions perform many measurements with good statistical accuracy and carefully study the randomness of the residuals identify and quantify medium and low frequency instabilities separately and apply a conservative propagation factor to T 1/2 report in sufficient detail, for traceability Half-life measurements are NOT EASY !

29
230th ACS NM - T1/2 28 Half-life of 93m Nb

30
230th ACS NM - T1/2 29 Half-life of 152 Eu a selection of the ‘best’ data

31
230th ACS NM - T1/2 30 Residuals Co-57 NIM A390 (1997) 267-273 decreasing accuracy with time => extension of experiment becomes ineffective

32
230th ACS NM - T1/2 31 Residuals Ba-133 and Eu-152 NIM A390 (1997) 267-273 ‘independent’ measurements showing correlations => ‘medium term’ detector instability

Similar presentations

Presentation is loading. Please wait....

OK

Using uncertainty to test model complexity Barry Croke.

Using uncertainty to test model complexity Barry Croke.

© 2017 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Ppt on earth leakage circuit breaker Ppt on branches of biology Ppt on earthquakes and tsunamis Resource allocation ppt on cloud computing Ppt on conservation of forest and wildlife in india Ppt on centering prayer Ppt on french culture in french language Ppt on do's and don'ts of group discussion strategies Ppt on volatility of stock market Ppt on media and democracy in india