Presentation on theme: "Energy Dispersive Spectrometry (EDS)"— Presentation transcript:
1Energy Dispersive Spectrometry (EDS) UW- Madison Geology 777Version Last Revised: 2/4/2014Electron Probe Microanalysis EPMAEnergy Dispersive Spectrometry (EDS)
2UW- Madison Geology 777What’s the point?Using X-rays to produce e-hole pairs (charges proportional to X-ray intensity), which are amplified and then “digitized”, put in a histogram of number of X-rays counts (y axis) versus energy (x axis). A solid state technique with unique artifacts.EDS spectrum for NIST glass K309(Goldstein et al, Fig. 6.12, p. 356)
3UW- Madison Geology 777SummaryX-rays cause small electric pulses in a solid state detector. Associated electronics produce ‘instantaneously’ a spectrum, i.e. a histogram of counts (number=intensity) vs the energy of the X-rayRelatively inexpensive; there are probably EDS detectors in the world for every 1 WDS (electron microprobe)Operator should be aware of the limitations of EDS, mainly the specific spectral artifacts, and the poor spectral resolution for some pairs of elements, and general lack of quality control of results
4Generic EMP/SEM EDS detector Electron gun Column/ Electron optics UW- Madison Geology 777Generic EMP/SEMElectron gunColumn/ Electron opticsOptical microscopeEDS detectorScanning coilsSE,BSE detectorsWDS spectrometersVacuum pumpsFaraday current measurement
5UW- Madison Geology 777EDS assemblageGoldstein et al fig 5.21There are several types of solid state EDS detectors, the most common (cheapest) being the Si-Li detector. Components: thin window (Be, C, B); SiLi crystal, FET (field effect transistor: initial amp), cold finger, preamp, vacuum, amp and electronics (“single channel analyzer”).
6UW- Madison Geology 777EDS WindowsWindows allow X-rays to pass and protect detector from light and oil/ice.Be: The most common EDS detector window has been made of Be foil ~7.6 mm (0.3 mil) thick. It allows good transmission of X-rays above ~ 1 keV. It is strong enough to withstand venting to atmospheric pressure, and opaque to optical photons.Thin - Ultrathin: For transmission of light element X-rays (<1 keV), windows ~0.25 mm thick of BN, SiN, diamond or polymer are used. They must use supporting grids to withstand pressure differentials; the grid (e.g., Si or Ni) takes up about 15% of the area, but the window material is thin enough that low energy X-rays pass through.This plots shows the transmittance of X-rays thru different types of window material. (Quantum [BN] 0.25 um, diamond 0.4 um). The higher the transmission number, the better“Windowless”: Here there is no film, and there is a turret that allows swapping with a Be window. Difficult to use as oil or ice can coat the detector surface. Not used much.Goldstein Fig. 5.41, p. 318
7UW- Madison Geology 777EDS WindowsThis plots shows the transmittance of X-rays thru different types of window material. (Quantum [BN] 0.25 um, diamond 0.4 um). The higher the transmission number, the betterGoldstein Fig. 5.41, p. 318
8How it works: energy gap A semi-conductor like Si has a fully occupied valence band and largely unfilled conduction band, separated by an energy gap (1.1 eV). Incident energy can raise electrons from the valence to the conduction band.X-ray hits the SiLi crystal, producing a specific number of electron-hole pairs proportional to X-ray energy; e.g. one pair for every 3.8* eV, so for incident Fe Ka, 6404 eV, 1685 e-hole pairs are produced. With a bias** applied across the crystal, the holes are swept to one side, the electrons to the other, producing a weak charge. Boron is important acceptor impurity in Si and degrades it (permits thermal excitation: bad); at the factory, Li is drifted in (donor impurity) to counter its effects.* 1.1 eV + energy wasted in lattice vibrations, etc**bias: a voltage is applied between 2 points; e.g. one v, other v.Goldstein et al, Fig 5.19
9How they make the Si(Li) For electronics, dope Si to change electronic structure: Si + Grp 5 (P, As) electrons majority carriers; Si + Grp 3 (B, Ga) holes majority carriers, BUT net effect is to allow current to flow in device inpresence of bias – not good for an x-ray detector where you ONLY want current to flow when x-ray impacts. WANT HIGH PURITY Si (but Boron contaminant); so diffuse in Li (n-type dopant). Then drift with reverse bias. Result: Li-rich surface n-type; center: no-charge carriers (compensated); Low-Li p-type. Remove the ends.Williams 1987
10How it works: inside the detector UW- Madison Geology 777How it works: inside the detectorX-rays are absorbed by Si, with photoelectrons ejected. This photoelectron then creates electron-hole pairs as it scatters inelastically. The Si atom is unstable and will either emit a characteristic Auger electron or Si ka X-ray. If Auger, it scatters inelastically and produces electron-hole pairs. If Si Ka X-ray,it can be reabsorbed, in a similar process, or it can be scattered inelastically. In either case, the energy will end up as electron-hole pairs. The result, in sum, is the conversion of all the X-ray’s energy into electron-hole pairs -- with 2 exceptions.Fig 9.5 Reed; Fig 5.22 Goldstein
11Artifacts: Si-escape peak; Si internal fluorescence peak UW- Madison Geology 777Artifacts: Si-escape peak; Si internal fluorescence peakThere are 2 exceptions to the neat explanation of how the Si(Li) detector works. Si-escape peaks are artifacts that occur in a small % of cases, where the Si ka X-ray generated in the capture of the original X-ray escapes out of the detector (red in figure). Since this X-ray removes 1.74 keV of energy, the signal generated (electron-hole pairs) by the incident X-ray will be 1.74 keV LOW. This will produce a small peak on the EDS spectrum 1.74 keV below the characteristic X-ray peak. Another artifact is the Si internal fluorescence peak, which occurs if an incident X-ray is absorbed in the Si “dead” layer (greenregion). This region is “dead” to production of electron-hole pairs, but Si ka X-rays can be produced here which then end up in the “live” part of the detector, and result in a small Si ka EDS peak.Consider Ti …Fig 5.22 Goldstein et al
12UW- Madison Geology 777Artifacts: Si-escape peaks; Si internal fluorescence peak; extraneous peaksThe figure shows a real spectrum of a sample of pure Ti metal -- but there are 7 peaks besides the Ti Ka and Kb. At 1.74 keV below each, are the respective escape peaks (blue arrows). Also present is a Si internal fluorescence peak (green arrow). The Fe and Cu peaks are from excitation of metal in chamber or sample holder by BSE or Ti X-rays. Note the sharp drop in the background intensity on the high side of the Ti Kb peak (= Ti K absorption edge, red arrow). (2 Ti Ka and Ti Ka+Kb explained shortly.)Note the scale of the spectrum: the Ti Ka max is 1.3 million counts. These effects are generally weak, but evident when you are looking for minor elements.Goldstein et al Fig 5.39,p. 316
13UW- Madison Geology 777Question: Do all characteristic X-rays have Si-escape peaks in a Si(Li) detector? Why or Why Not?Hint 1: Sr La does not, but Os Ma doesHint 2: Look up the characteristic energies of eachHint 3: Look up the absorption edge (critical excitation) energy of Si KaHint 4: Compare the numbers in 2 to number in 3. Which one is greater than the one in #3?Would a Si Ka x-ray produced in the sample, which then makes its way thru the vacuum to the EDS detector, have enough energy to knock out the inner shell (K) electron of the Si detector crystal?Sr La 1805 Sr Lb Os Ma Si edgte 1839
14UW- Madison Geology 777Signal processingSi(Li) detector has no internal gain*; for Ca Ka photon with ~1000 e-hole pairs, the charge is only ~10-16 Coulomb (weak!)Directly coupled to the Si crystal is a field effect transistor (FET) that converts charge to voltage, followed by a preamp. We need low noise, high gain amplification, so theSi detector and FET are cooled to about 100K with liquid nitrogen (LN) to prevent noise (and prevent diffusion of Li in detector--at least in old ones). More signal gain provided by main amplifier (signal now boosted to 1-10 volts) where also RC (resistor-capacitor) circuits are used to shape the pulse, to maximize signal/noise ratio and minimize pulse overlap at high count rates. Then ADC (analog to digital converter) outputs data to the screen as a spectrum display.*gain = electronic multiplication of signal intensity
15The first signals in the EDS detector UW- Madison Geology 777The first signals in the EDS detectorThe set of electron-hole pairs produced by the impact of the X-ray on the Si(Li) detector produces a tiny charge (~10-16 C), very quickly (~150x 10-9 sec). The FET(preamplifier) changes the charge (capacitance) into a tiny voltage (millivolts). These steps are shown in the first half of (a) to the right. The output of the FET is shown below at (b) where the x axis is time and y is voltage. The “jump” represents the presence of a voltage proportional to the number of electron-hole pairs generated by each X-ray, so Photon 2’s jump is of a higher energy than Photon 1’s jump which is higher than Photon 3’s jump. At a certain point the FET reaches the limit of the number of“ramp”charges it can hold, and then there is a reset or zeroing back to some baseline where it starts over. Following this are electronics to shape the voltage into a pulse that can be counted.Goldstein et al (1992), p. 297
16Processing Time and Pulse Pileup Rejection UW- Madison Geology 777Processing Time and Pulse Pileup RejectionThe user can ‘tweak’ the time constant (T.C.) which sets the time allocated in the electronics to process each pulse (=X-ray). In the top figure, a short T.C. (1 ms) permits each pulse to be counted correctly. A longer T.C. (10 ms) means the “gate” is open longer and a second pulse can enter and be incorrectly added; this is “pileup” and causes distorted spectra. Therefore, circuits are added (#4, bottom figure) to sense when pileup occurs and to ignore that pulse.Goldstein et al (1992), Fig and 5.25, p. 300
17UW- Madison Geology 777Dead Time“Deadtime” is the period during which the detector is “busy” and cannot accept/process pulses. This can introduce error unless it is accounted for, either by extending counting time, or correcting for it in the software. In most systems, the user sets the “live time” which is the time during which counts are actually counted, and the “real time” is automatically determined by the electronics or software.Optimal deadtime is in the 35-45% range. This optimizes both user/machine time and moderate to high throughput of counts.80%40%60%DeadtiimeReal time = live time + dead time seconds = 80 seconds counting + 20 seconds dead for 20% deadtimeGoldstein et al (1992), Fig (p. 300) andGoldstein et al (2003), Fig. 7.9 (p. 307)
18Detector performance: peak resolution (FWHM) UW- Madison Geology 777Detector performance: peak resolution (FWHM)The characteristic X-rays generated in the specimens are very close to lines, i.e. only a few eV wide at most. However, the conversion of X-ray to a pulse in the detector has several variables (imperfections) that broaden the peak to between maybe eV, depending upon the type of detector and how well maintained it is. The narrowness of the peak is measured by the width of the peak at one half the maximum intensity of the peak -- this is what is termed the FWHM.In EDS detectors, it is usually measured at the Mn Ka position, with values of 160 eV and below. Modern (2005) one are quoted at <130 eV.Goldstein et al, Fig 5.34, p. 311
19Why Mn Ka for EDS resolution? UW- Madison Geology 777Why Mn Ka for EDS resolution?EDS companies (their engineers mainly) do not want to have to carry around an SEM or EMP to be able to test, repair and calibrate an EDS system. Instead they carry a small 1” diameter x 2” long tube that fits over the end of the EDS “snout”. Inside it is an Fe-55 isotope source (half life 2.7 yr) which emits an intense x-ray at keV which is only a few eV different than Mn Ka.
20Time Constant+Beam Current--> Dead Time UW- Madison Geology 777CurrentTCDTCtsResCommentLowShort10%2000150 eVTakes a long time to get a decent spectrumLong20-30%1000135 eVOptimal for phase IDHigh40-60%10-20K180 eVOptimal for X-ray mapping100%___Good for nothingThe above values are approximate, and meant only to show the relation of the variables.
21Spectral processing: background correction UW- Madison Geology 777Spectral processing: background correctionThe characteristic X-rays that we need to quantify “ride” atop the continuum, and the continuum contribution to the characteristic counts must be subtracted. (Top) Linear interpolation (B-D) will be in error due to the abrupt drop of continuum at the Cr K-absorption edge (5.989 keV). B-C is possible but critically dependent upon having good spectral resolution (<160 eV). A-B would be preferable. (Below) Doing background fit of a complex stainless steel.Goldstein et al Fig. 7.1,2, p. 367
22Spectral processing: background modeling or filtering UW- Madison Geology 777Spectral processing: background modeling or filteringCorrecting for the background is done by either of 2 methods: developing a physical model for the continuum, or using signal/noise filtering.Modeling is based upon Kramers Law: there is a function describing the continuum at each energy level, that is a function of mean atomic number, and measured “detector response”.
23UW- Madison Geology 777Background ModelingThe spectrum of Kakanui hornblende (top left), with superimposed calculated (modeled) background, based upon Kramers Law*. Bottom figure shows after the background has been subtracted. Cu is artifact (stray X-rays). Mn is actually present at <700 ppm.Ij = constant x Z (E0 - Ej) / Ejat each energy channel jGoldstein et al Fig 7.4, p. 372
24UW- Madison Geology 777Background FilteringTheoretically Fourier analysis will separate out the low frequency continuum signal and high frequency ‘noise’ from the medium frequency characteristic peaks; however, there is overlap and the result is a poor fit. A better filter is the “top hat filter”, where no assumptions are made about the spectrum, and only the mathematical aspects of signal vs noise are considered.
25UW- Madison Geology 777Top Hat FilteringThis filter (top right) moves across the EDS spectrum (with an optimally defined window, ~ 2 FWHM* Mn Ka;~320 eV), and assigns a new value for the center channel based upon subtracting the values in the left and right channel from the center (value hk chosen to total area =0). Thus, in the simple spectrum (bottom right), the center channel (+), when the left and right channels are subtracted, leaves a value ~0.*FWHM: full width at half maximum.Reed Fig 12.7 p. 174,Goldstein et al Fig 7.6, p. 374
26More Artifacts: Sum Peaks UW- Madison Geology 777More Artifacts: Sum Peakstrue single X-ray peak, and includes it with all the other peaks from the elements actually present. For 2 major elements, could be 3 sum peaks; for 3, 6. In reality,you only see 1 or 2 unless you zoom in to the background level. Always consider their possible presence.There is a short period of time (t0) during each X-ray capture by the EDS detector, when the detector can capture a second X-ray “by mistake”. The electronics cannot distinguish this “sum peak” from a
27UW- Madison Geology 777…Fools even the prosEven the fanciest, slickest EDS setup can fool newcomers, not to mention experienced users. Above, is a partial spectrum (major peaks) of a commercial glass that has a lot of Si and O, plus Na and Al. Notice the S (Sulfur) label over a peak around 2.3 keV … sure looks like it might be Sulfur, right? It is NOT, rather it is a sum peak of O Ka (.525 keV)+ Si Ka (1.74 keV). Previous experience with this “fake” peak had taught me to be skeptical
28UW- Madison Geology 777Sum PeaksIn qualitative analysis of silicates, there are some combinations of element Ka peaks that fall close to Ka peaks of elements possibly present, as indicated in the table below:Some of more advanced EDS software now contain algorithms to recognize and automatically remove sum peaks. But you must always be on guard for them, particularly ones which “could really be there”. In many of those cases, WDS is the solution.
29UW- Madison Geology 777And More ArtifactsThere is always a potential for ‘stray’ X-rays being detected. It thus pays for the EDS operator to understand what the path is for the electron beam and for the X-rays, and know what ‘other’ elements might show up unintentionally.This is particularly true for EDS associated with TEM, where specimens routinely sit on grids (Cu?) and the high energy (200 keV?) electrons can go through the specimen and hit a metal part of column or chamber, with the resulting X-rays finding a way back to the detector.
30And More Artifacts Family of Pd L lines UW- Madison Geology 777And More ArtifactsAnother thing: many SEM labs use gold or palladium coating on specimens. These very thin coats will produce definite x-ray peaks!Family of Pd L lines
31…And beware of lazy peak IDs UW- Madison Geology 777…And beware of lazy peak IDsEven the fanciest, slickest EDS setup can make misidentifications…so the analyst cannot get lazy and assume just because the expensive software said something was there, it was there. I knew Arsenic was possible (As La identified), but unlikely, and rather Mg Ka was more likely. To confirm it was NOT As, I cranked the accelerating voltage up to 20 keV (the K shell binding energy is 11.9 keV) and found there was NO As Ka x-ray. Ergo, not As.
32…and trusting software UW- Madison Geology 777…and trusting softwareAs Newbury (2005 and 2006 reply) pointed out, an EDS operator is a fool to believe that the automatic peak ID will be correct 99.9% of the time.From Newbury (2006)
33Artifical EDS spectrum UW- Madison Geology 777Artifical EDS spectrumArtificial: no background, no artifacts, and assumes EACHelement at 100% concentration. Why, then, the two slopes?? Peak intensities of elements from Si to Na decrease, and also from Si to Zn -- why? (Hint: 2 physical phenomena)
34UW- Madison Geology 777Artificial spectrumThe actual spectrum of pure elements, as generated at the point of impact, would be one steady decreasing curve from Na down to Zn, following the red curve superimposed here.Slope down from Si to Zn: there are less and less X-rays being produced because the accelerating voltage is constant (e.g. 20 keV) and the overvoltage is lower.Slope down from Si to Na: X-ray energies are increasingly weaker, and are absorbed both within the specimen and by the window.
35Evolution of EDS spectrum: from the specimen to the monitor - 1 UW- Madison Geology 777Evolution of EDS spectrum: from the specimen to the monitor - 1The spectrum on our monitor (d) is a result of many things impacting the real spectrum generated within the specimen (a). At instant of generation within the specimen, there is only the Ka, Kb and continuum. An instant later (b), as the X-rays leave the specimen, two things can happen: some of the continuum X-rays above keV are absorbed, producing the drop in the continuum there.Simulation of element (say V) X-ray generation and displayGoldstein et al Fig 5.53 (by R. Bolon) p. 330
36Evolution of EDS spectrum: from the specimen to the monitor - 2 UW- Madison Geology 777Evolution of EDS spectrum: from the specimen to the monitor - 2Also in (b) the lower energy continuum is absorbed, causing the dropoff in the spectrum there. When the X-rays hit the detector (c), Si fluorescence peaks can result. And after signal processing (d), the display will show peak broadening, sum peaks, Si-escape peaks, further decrease of intensity and low energy noise.Simulation of element (say V) X-ray generation and displayGoldstein et al Fig 5.53 (by R. Bolon) p. 330
37Comments about LN2 and EDS UW- Madison Geology 777Comments about LN2 and EDSThat big tank of liquid nitrogen cools the SiLi crystal and the FET, so the very low charge generated by the electrons-holes can be detected with minimal noise. But what about letting the thing warm up when you’re on vacation? There is a lot of misunderstanding about this… Modern systems “can” be allowed to warm up without damage to the crystal (if the bias on it is turned off) -- BUT that is not the only thing to be concerned about when it warms up. Another important ingredient is thevacuum within the snout that extends from the bottom of the dewar to the end where the detector sits -- there is a “getter” (zeolites or Al wool) inside that absorb yucky contaminants. But if the getter warms up, they are released inside the snout, creating a poor vacuum, which then means the LN usage increases significantly as the vacuum is poor. Bottom line: keep it cold all the time.
39UW- Madison Geology 777Recent DevelopmentsOver the past years, 2 new “spins” off the ‘old school’ Si(Li) EDS detector have entered the microanalysis world:The microcalorimeterThe Silicon Drift Detector
40UW- Madison Geology 777MicrocalorimeterThe principal behind the microcalorimeter is that an x-ray hitting a very sensitive thermal absorber will register a very small temperature increase. However, this requires a very cold absorber, with liquid helium cooling required. It would provide the best of both EDS and WDS, with simultaneous capture of all x-ray energies AND with very tight spectral resolution (like with WDS).However, there apparently have been major engineering stumbling blocks and none have made it to the market.
41Silicon Drift Detector UW- Madison Geology 777Silicon Drift DetectorThe SDD is similar to SiLi Detector in that electron-hole pairs are generated, but the physical design is radically different. There is a lower capacitance, and also a lower leakage current (high leakage current in SiLi is what requires LN cooling). And because the SDD has the FET “built in”, created during the lithography of the Si crystal, wires are eliminated, reducing capacitance more.Resulting advantages:LN not needed (use a simple Peltier cooler)Can handle high count rates >100,000 up to ~106 cpsSpectral resolution at 100,000 cps still good (~ eV)
42The SDD is created from a single Si crystal using micro-lithography. Image from Bruker web pageThe SDD is created from a single Si crystal using micro-lithography.“The major distinguishing feature of an SDD is the transversal field generated by a series of ring electrodes that causes charge carriers to 'drift' to a small collection electrode. The 'drift' concept of the SDD (which was imported from particle physics) allows significantly higher count rates.” - Wikipedia
43Silicon Drift Detector Simulation For the full simulation, go tourl updated 2/3/14
44UW- Madison Geology 777Further EDS detailsThere are several modern EDS companies, with most producing very informative brochures that go into the technical details of EDS hardware (and software):For example: Oxford Instrumentshas a nice technical publication explaining EDS using the SDD as the detector. Well worth downloading and reading.url updated 2/4/14