3Introduction:the anesthesia delivery system has evolved from a simple pneumatic device to a complex multisystem workstationFor anesthesia care providers understanding of its operation is essential.Caplan and coworkers found that although claims related to the medical gas delivery system were rare, when they did occur, they were usually severe and death or permanent brain injury frequently resulted.
4In this Lecture: we examine the anesthesia workstation piece by piece. We will describe the normal operation, function, and integration of major anesthesia workstation subsystems.we illustrate potential problems and hazards associated with the various components of the anesthesia delivery systemWe overview appropriate preoperative checks that may help detect and prevent such problems.
6Anesthesia Workstation Standards and Pre-use Procedures Standards for Anesthesia Machines and Workstations1979: American National Standards Institute (ANSI) Z 1988: American Society for Testing and Materials (ASTM) F 1994: ASTM F  (reapproved in 1994 and discontinued in 2000)2000: ASTM F 
72000 ASTM F standards:Newly manufactured workstations must have monitors that measure the following parameters: continuous breathing system pressure, exhaled tidal volume, ventilatory carbon dioxide concentration, anesthetic vapor concentration, FiO2, oxygen supply pressure, SpO2, BP, and EKG.The anesthesia workstation must have a prioritized alarm system that groups the alarms into three categories: high, medium, and low priority
8Checking Your Anesthesia Workstation A complete anesthesia apparatus checkout procedure must be performed each day before the anesthesia workstation is first used.An abbreviated version should be performed before each subsequent case.The user must always refer to the original equipment manufacturer's operator's manual for special procedures or precautions related to particular workstations.
9Responsibility:Every anesthesia care provider must be aware that the ultimate responsibility for proper machine pre-use safety checks rests on the provider using the machine to deliver anesthetic care.
10Testing Specific Components of the Anesthesia Delivery System The three most important preoperative checks are:(1) calibration of the oxygen analyzer(2) the low-pressure circuit leak test(3) the circle system tests
11Calibration of the Oxygen Analyzer It is the only machine safety device that evaluates the integrity of the low-pressure circuit in an ongoing fashionThe only machine monitor that detects problems downstream from the flow control valves is the oxygen analyzersensing element must be exposed to room air for calibration to 21%
12Low-Pressure Circuit Leak Test checks the integrity of the anesthesia machine from the flow control valves to the common gas outletLeaks in the low-pressure circuit can cause hypoxia and/or patient awarenessLoose filler caps on vaporizers are a common source of leaksSeveral different methods have been used:the oxygen flush test,the common gas outlet occlusion test,the traditional positive-pressure leak test,the North American Dräger positive-pressure leak test,the Ohmeda 8000 internal positive-pressure leak test,the Ohmeda negative-pressure leak test,the 1993 FDA universal negative-pressure leak test,and others.
13Oxygen Flush Positive-Pressure Leak Test Inappropriate use of the oxygen flush valve to check the low-pressure circuit
141993 FDA Negative-Pressure Leak Test Left, A negative-pressure leak-testing device is attached directly to the machine outlet. Squeezing the bulb creates a vacuum in the low-pressure circuit and opens the check valve.Right, When a leak is present in the low-pressure circuit, room air is entrained through the leak and the suction bulb inflates
151993 FDA Negative-Pressure Leak Test if the bulb re inflates in less than 10 seconds, a leak is present somewhere in the low-pressure circuit.The “universal” negative-pressure leak test is the most sensitive of all contemporary leak tests because it is not dependent on volumeIt can detect leaks as small as 30 mL/min
16Circle System TestsThe leak test is performed by closing the pop-off valve, occluding the Y-piece, and pressurizing the circuit to 30 cm H2O with the oxygen flush valve. The value on the pressure gauge will not decline if the circle system is leak freeThe flow test checks the integrity of the unidirectional valves: The operator should be able to inhale but not exhale through the inspiratory limb and able to exhale but not inhale through the expiratory limb.
17Workstation Self-Tests self-diagnostic tests varies from one model and manufacturer to anotherto detect internal vaporizer leaks on this type of system, the “leak test” portion of the self-diagnostic must be repeated separately with each individual vaporizer turned to the “on” position
18Anatomy of an Anesthesia Workstation Diagram of a generic two-gas anesthesia machine
20Pipeline Supply Source The pipeline supply: primary gas sourcemust supply: correct gases at appropriate pressureIn a survey: 31% reported difficulties with pipeline systems (The most common problem was inadequate oxygen pressure)The most devastating reported hazard: accidental crossing of oxygen and nitrous oxide pipelines, (which has led to many deaths)
21In the event that a pipeline crossover is ever suspected immediately: First, turned on the backup oxygen cylinderSecond, disconnect pipeline supplyThe pipeline inlet fittings are gas-specific Diameter Index Safety System (DISS)
25Cylinder Supply Source Medical gases attached to the anesthesia machine via the hanger yoke assembly.Each hanger yoke is equipped with the Pin Index Safety System (PISS).The PISS is a safeguard introduced to eliminate cylinder interchanging
26Pin Index Safety System (PISS) Pin-Indexed Yoke AssembliesandCylinder Valve Connections
28A check valve is located downstream from each cylinder and serves several functions. First, it minimizes transfer of gas from a cylinder at high pressure to one with lower pressure.Second, it allows an empty cylinder to be exchanged for a full one while gas flow continues from the other cylinder into the machineThird, it minimizes leakage from an open cylinder to the atmosphere if one cylinder is absent.Each cylinder supply source has a pressure-reducing valve known as the cylinder pressure regulatorThe oxygen cylinder pressure regulator reduces the oxygen cylinder pressure from a high of 2200 psig to approximately 45 psig.The nitrous oxide cylinder pressure regulator receives pressure of up to 745 psig and reduces it to approximately 45 psig.
30The gas supply cylinder valves should be turned off when not in use, except during the preoperative machine checkout period.the volume of gas remaining in the cylinder is proportional to cylinder pressure.use of a pneumatically driven mechanical ventilator will dramatically increase oxygen utilization rates,Hand ventilation at low fresh gas flow rates may consume less than 5% of the amount of oxygen consumed by intermediate flow meter settings coupled with the use of pneumatically powered mechanical ventilationNitrous oxide cylinder ??
31Oxygen Supply Pressure Failure Safety Devices abrupt or insidious oxygen pressure failure had the potential to lead to the delivery of a hypoxic mixture.The 2000 ASTM F standard states that “The anesthesia gas supply device shall be designed so that whenever oxygen supply pressure is reduced to below the manufacturer specified minimum, the delivered oxygen concentration shall not decrease below 19% at the common gas outlet.”
32Fail-safe valveA fail-safe valve is present in the gas line supplying each of the flow meters except oxygen. Controlled by oxygen supply pressurethe valve shuts off or proportionally decreases the supply pressure of all other gases as the oxygen supply pressure decreases.
33Pressure sensor shut-off valve An oxygen failure protection device that responds proportionally
34Pneumatic and Electronic Alarm Devices (oxygen supply failure alarm) Many older anesthesia machines have a pneumatic alarm device that sounds a warning when the oxygen supply pressure decreases to a predetermined threshold value, such as 30 psigElectronic alarm devices are now used to meet this guideline.
35Second-Stage Oxygen Pressure Regulator Most contemporary Datex-Ohmeda workstations have a second-stage oxygen pressure regulator set at a specific value ranging from 12 to 19 psig
37precisely controls and measures gas flow to the common gas outlet: Flow Meter Assembliesprecisely controls and measures gas flow to the common gas outlet:Traditional glass flow meterElectronic flow sensorsnumericalgraphiccombination