Presentation on theme: "Trey Bates, MD Special Thanks to Judson Mehl The Anesthesia Machine and Breathing Systems."— Presentation transcript:
Trey Bates, MD Special Thanks to Judson Mehl The Anesthesia Machine and Breathing Systems
All those hoses: Oxygen Air Nitrous Vacuum WAGD Waste Anesthetic Gas Disposal A quick word on medical gas - 2000 PSI (FULL) H (7000L) E (700L)
H-Cylinder 2000 psi Manifold55 psi Hospital line55 psi Pressure Reduction Pathway O2 - 2000 psi, 625 L N2O - 750 psi, 1590 L Co2 - 838 piso, 1590 L Air - 1800 psi, 625 L
2000 psi !!! 45 psi Oxygen Failure Protection Device Flow of nitrous-oxide is dependent on oxygen pressure. If oxygen pressure is lost then the other gases cannot flow past their regulators
Liquid oxygen must be stored below its critical temperature of -119 C In oxygen tanks, the pressure falls in proportion to the remaining volume of oxygen If a full E-cylinder at 2000 psi contains 700 L O2, then a half full tank at 1000 psi contains ? What about an H-cylinder at 1000 psi? What about an E-cylinder at 500 psi? Key points for ITE:
ICU transport with an E-cylinder with 700 psi. Need to run a NRB at 10 lpm. Tulane elevator breaks down. How much sh*t are we in? More math... for fun
Nitrous is NOT an ideal gas. Thus it has several unique properties: Transition between liquid and gas states does not lead to huge increases in pressure It is easy to compress, so the cylinders hold a lot moregas. Its critical temp is 36.5 C, so it doesn’t need refrigeration Nitrous
N2O is vaporized at the same rate it is utilized The pressure in the tank never changes You don’t know what you’ve got til its gone (400 L/1600L = 25% remaining)) The only way to tell how much N2O is left, is to measure the tank. Consult the tare weight on the bottle I have never been asked to determine how much N2O was left based on weight on an ITE. More on N2O
This stuff matters because: Oxygen is pretty important Agent delivery is pretty important Getting rid of CO2 is pretty important And these ALWAYS show up on the ITE. ALWAYS !! Breathing Systems
Is a... Cardiac Depressant And it... Increases CBF Increases bleeding Causes acidosis which... Shifts the Carboxy-Hgb curve Shifts Ca2+ and K+ out of the cell CO2
Anesthesia at Tulane
Gas delivery under a drape Serious CO2 accumulation without high gas flow Lets be honest, everybody in the room is breathing this stuff Insufflation anesthesia
Open Drop Anestheisa Schimmelbusch mask
Hose serves as an open-ended reservoir Addition of oxygen possible 1 lpm 30-40% 4 lpm 60-80% Simple Portable No scavenging Draw-over anesthesia
Ingredients: Breathing tube Fresh gas inlet APL valve Reservoir bag The Mapleson Circuit The only real difference is the order in which the ingredients occur
Important points There are NO one-way valves There is no CO2 absorber Some rebreathing is prevented by venting through the APL before the next inspiration Mapleson Circuits
Basic Mapleson A During spontAneous ventilation, the Mapleson A is most efficient. That long breathing tube full of fresh gas is a great reservoir for the patients next breath.
Giving positive pressure is going to require me to partially close that APL valve. When I ventilate, half of my FGF is going to exit the partially open APL valve. During exhalation, all that exhaled air is going to fill the breathing tube because the APL is now closed and the only way it is going to vent is if the gas flows are really high. Why?
The FGF is happening right at the patient’s face. Now, watch this.. Mapleson D
When I positive pressure ventilate, I close the APL and use the old air in the reservoir to generate the force to blow the fresh air into the patient. Anything I lose out of the APL will be old air. Between ventilations the new air is pushing the old air out of the APL and away from the patient.
The Bain circuit deposits the FGF in the same place as Mapleson D, but it traveled through the warm, exhaled air on the way in, so the FGF is warmed. Bain
If not, and you probably wont on the day of the ITE, then check out this aswesome memory aid. Its pretty complex: Ventilation is most efficient in a Mapleson A during spont A neous ventilation There is no D in spontaneous Mapleson D during controle D ventilation There is no A in controled Get it now?
Lose all the heat and humidity High FGF to prevent rebreathing All that agent is ventilated out to the room The downfalls of the Mapleson
And then we added: CO2 absorbers Unidirectional valves Scavenging And voila, we have the CIRCLE SYSTEM !! So, science happened
How they work: So why is it bad that the CO2 absorber “dries out?” Well, here is why: CO2 + H2O → H2CO3 (this is carbonic acid) Then the hydroxide salts in the CO2 absorber do this: H2CO3 + 2NaOH → Na2CO3 + 2H2O + heat ( this is why they get warm ) Then all that Na2CO3 (sodium hydroxide) produced in the first reaction does this: Na2CO3 + Ca(OH)2 → CaCo3 + 2NaOH ( We just regenerated our starting reagent ) CO2 absorbers
As the absorbent is used up, it becomes more acidic. That purple color change is a pH indicator When 50-70% has changed color, its time to change the absorber. CO2 absorbers Granule size is a trade off: Larger granules minimize resistance to airflow Smaller granules maximize surface area for more absorption
Inspiratory Expiratory Valve incompetence is usually due to unseated or warped disc Note what is in the reservoir bag And what about these unidirectional valves?
In a closed scavenging system, what happens to the reservoir bag during expiration and inspiration? What does it mean with the opposite happens?
The reservoir bag expands during expiration and deflates during inspiration. During inspiration in MV, the ventilator pressure relief valve closes, directing ventilator bellows into patient breathing circuit. If the PRV is incompetent, there will be a direct communication between breathing circuit and scavenging circuitand the reservoir bag would inflate during inspiration.
A size E compressed-gas cylinder completely filled with N2O contains how many litres? A. 1160 B. 1470 C.1590 D. 1640 E. 1750 A few questions
C Size E compressed gas cylinders completely filled contain 1590 L gas Answer:
The pressure gauge on a size E compressed-gas cylinder containing O2 reads 1600 psi. How long could O2 be delivered from this cylinder at 2 LPM? A. 90 min B. 140 min C. 280 min D. 320 min E. Cannot be calculated Question
If the anesthesia machine is discovered Monday morning having run with 5L/min of O2 all weekend, the most reasonable course of action to take before administering the next anesthetic would be: A. Turn the machine off for 30 min B. Place a humidifier in the expiratory limb C. Avoid the use of Sevoflurane D. Change the CO2 absorbent E. Use N2O for the first hour of the case Question
D – of course, but why change it if its not purple? Answer
A mechanically ventilated patient is transported from the OR to the ICU using a portable ventilator that consumes 2L/min of O2 to run the ventilator itself. The patient gets 100% O2 and tidal volumes of 500 ml at a rate of 10/min. You have an E-cylinder with 2000 psi. The vent will shut off below 200 psi. How long do you have? A. 10 min B. 30 min C. 60 min D. 90 min E. 100 min One last painful question
The DISSThe PISS
Lets take a quick look at these so- called “OFPDs”
Lets look again
One thing I want you to note. We have discussed the HIGH-PRESSURE CIRCUIT to this point. Gas lines proximal to the flow valves (knobs) are considered the high-pressure circuit Distal to the knobs (eg. In the Thorpe tubes and onward) you are in the low-pressure circuit. To be continued... Lets stop for today
Oxygen is universally on the right The knob is larger and fluted Why? Flowmeter sequence:
It is a safety feature, but not 100% fool proof. You can still make a hypoxic gas mixture. The less circuit AFTER the O2 joins, the less chance of a leak in the post-O2 part of the circuit.
These are called “constant- pressure variable- orifice” flowmeters. Conductive coating to reduce effect of static electricity Calibrated to be gas- specific ** This is a Thorpe Tube ** Flow rate across a constriction depends on the gas’s viscosity at low laminar flows and its density at high turbulent flows.
Draeger utilizes this little gem: Oxygen/Nitrous Oxide ratio controllers
But, Datex- Ohmeda got it right
A couple key points on vaporization Anesthetics have a vapor pressure, which is the propensity to come out of solution and form a... Vapor. Vapor pressure is temp- dependent. Higher temp = vapor pressure On to vaporizers The energy required for vaporization is manifested as loss of heat from the anesthetic solution As the anesthetic vaporizes, the solution becomes colder... And when the temp drops, so does the vapor pressure !!!
Copper kettles Copper has a high specific heat Copper has high thermal conductivity Resistant to the temperature drop from vaporization Best material to maintain a constant temperature
Measured-flow vaporizer Separate flowmeter for the gas flowing through the kettle Gas passing through the kettle becomes fully saturated Then you dilute it out to the proper percentage with the other flowmeter Copper Kettle The math gets a little funny here, and they will throw you a copper kettle equation on the ITE, so watch this...
Vapor pressure of Halothane (and ISO) is about 243 mmHg at 20 C So 243/760 = 32% The math of a copper kettle
At atmospheric pressure, if we put 100 ml O2 through the kettle, we will get 150 ml of FGF on the other side. 50 mL of that will be volatile If we keep the flow at 5LPM total we can do this 100 ml to the kettle → 150 ml, 50 of which is halothane 4850 ml to the dilution limb 50 ml halothane / 5000 ml total FGF = 1% halothane So here is what we know
For the ITE, in a copper kettle there should always be a total of 5L FGF Add 50 ml vapor to every 100 ml you put through the kettle Keeping the totals at 5L/min, every extra 100 ml through the kettle increases the agent concentration 1% Eg. 100 → kettle → 150 (50ml agent) + 4850 = 1% agent 200 → kettle → 300 (100ml agent) + 4700 = 2% agent 300 → kettle → 450 (150ml agent) + 4550 = 3% agent For the ITE Remember, Iso and Halothane have similar vapor pressures, so this applies to Iso too
These are modern vaporizers Tec 4,5,6 all have similar mechanisms Aladin is very different. Well, that sucked. Lets move on. Tec 4 Tec 5 Tec 6 Aladin Vaporizers
Recall saturated vapor pressures: So, at atmospheric pressure and 20 C, if I let all the FGF flow through the vaporizer, it would saturate and produce: Halo243243/760 = 32% Iso 238238/760 = 32% Sevo 157157/760 = 20% Des672672/760 = 88% These are “slightly” above clinically relevant concentrations Lets jump back to physics for one slide
Well, then we have a variable bypass vaporizer. But if we split the FGF between the vaporizer and a bypass channel...
Note the bi- metalic strip This serves to compensate for temperature changes Vapor pressure is temperature dependent. If its warmer, the vapor pressure is higher, we need to slow the gas flow through the chamber. The Datex-Ohmeda version
Tilting old vaporizers could flood the bypass area, in which case you would deliver the full vapor pressure of the agent (Halothane 32%) Tilting hazard !!
The FGF does not actually flow through the sump. Instead fixed concentration Des vapor is added in proportion to the FGF The Des Vaporizer -
Well, the vapor pressure of DES at room temp is 672. Problem #1 – The heat loss from that much vaporization would rapidly cool the vaporizer and end up dropping the vapor pressure dramatically Problem #2 – 672/760 = 88% It would take tremendous FGF through the bypass chamber to dilute that down to a useable level. Just one more point on DES which is covered on the ITE – the old vaporizers for ISO, SEVO automatically compensate for changes in altitude. But, high altitude DECREASES the partial pressure for DES, so you will have to manually increase the concentration of DES at high altitudes. Why vaporize DES this way?
Because it is partial pressure that really matters Forget Volume% for a min Partial pressure is measured in mmHg 2% ISO at 760mmHg = 15.2mmHg (partial pressure) So, of the gas coming out of the machine at 760mmHg, 2% of it (or 15.2 of those mmHg) are Isoflurane. Why?
We know 2% @ 760 is 15.2 mmHg – but what if we climb?? At higher altitude, the decreased Patm will allow more ISO to come out of solution. So, even if the dial is set at 2%, the actual concentration coming out is higher – again because the lower Patm lets more ISO come out of solution. But if we went to altitude with half the Patm, we would double the concentration coming out of the vaporizer, but the partial pressure remains the same 4% @ 380 is 15.2 mmHg – This is how these vaporizers self-equilibrate % increases but Patm decreases so overall is same result The thing is, DES cant do this, because no matter how high you go in altitude, the % is exactly what you set on the dial... So you have to deliver higher concentrations to compensate for altitude
This diagram sucks What I want you to take from this is that there is no bypass channel in the cassette itself As such, the cassette is not a tipping hazard. I bet you can stump the faculty with this one The Aladin Cassette vaporizer
Ye olde breathing circuit A couple things to notice here. This is the FG input. Note it is on the proximal side of the inspiratory valve. This valve would be closed during exhalation. This is the old spirometer. It is just distal to the expiratory valve. So when it is taking a measurement at expiration, it is getting only exhaled tidal volume, and no FGF which would throw the measurement off.
As proven at lakeside, these spiromed spirometers really stand the test of time Designing spiromed spirometer... In dog costume.
This is the vane aneometer. vane, like weather vane. It spins in the wind. It was placed in line proximal to the exp. valve and the TV was calculated based on the spin. Last few slides on spirometers, I promise
This is a Pilot tube. It is used in aviation. Here is how it works : Fixed orifice flowmeter Pitot tubes are tube shaped and contain 2 holes. One hole faces the direction of movement and, measures the stagnation pressure of oncoming air. The other hole is on the side and measures static pressure. The difference between these two pressure types allows for the measurement of dynamic pressure, which is then used to calculate the aircraft's airspeed
We can use those same principles to integrate flow over time to determine tidal volumes. If you know the airway pressures you can integrate that info as well and produce the flow-volume loops which can tell you lots about airway and lung mechanics. So how does that help us?
Older methods of external ventilation relied on generating negative pressure around the chest wall. Ventilators
4 phases are identified Inspiration Transition from inspiration to expiration Expiration Transition from expiration to inspitation We classify a vent based on its inspiration and transition from inspiration to expiration characteristics Phases of ventilation
Constant pressure generator Constant flow generator Non-constant generator Note they are based on the way they handle inspiration
You can terminate inspiration based on one of three parameters Time Volume Pressure I expect most of us understand the volume and pressure modes. Time mode: You set a time allotment for inspiration and you vary the gas inflow rate during that time allotment until you reach a tidal volume you are happy with. You can really play with this mode in patients with terrible lung compliance and adjust it to try and limit your Ppeak. Transition from inspiration to expiration
The expiratory phase is simple. Return the lungs to atmospheric pressure, unless you have set some PEEP A completely passive process Expiratory phase
It is really this phase on which we base our nomenclature for vent modes Volume-Control → vent adjusts gas flow rate to deliver set tidal volume based on set vent rate and I:E ratio Pressure-Control → vent adjusts gas flow rate to deliver a constant pressure based on set vent rate and I:E ratio You already knew that. Transition from Exp to Insp
2 main types of Ventilator Circuit Design: Double-Circuit Piston-Driven Now its time for some meat and potatoes
This is a double circuit ventilator
Remember this guy? He puts out O2 at hospital line pressure (?psi)
This is him too... sneaky So it makes sense that it is delivering 50 psi of O2
1. In a double-circuit system the tidal volume is delivered from a bellow inside a plastic bucket Older vents used hanging bellows that were weighted. But when the circuit disconnected the weight would pull the bellow down and you may think you were still ventilating the patient. This is what they looked like: Remember him, and lets make a few points Obviously this was not too safe.
Because for some reason, we tend to notice a lot faster when the bellows are just laying flat So, we switched to ascending bellows Before we flip to the next slide, I need a CA-1 to tell me where the air inside the bellows is coming from…
It’s the patient’s last exhaled breath Just like the breathing bag, the bellows is simply pushing air around the circle system. So, the vent is really just an automatic breathing bag This is now the vent.
This is how they really work: The inside of the bellows is continuous with the circle system The 50 psi O2 from the oxygen power outlet enters here and pressurizes the bucket. This squeezes the bellow
When you are running the vent, the APL is excluded from the circuit But, luckily, the vent has its own APL – the “spill valve” Exhaled air beyond the capacity of the bellows and circle system opens the spill valve just like a pop-off and gets shunted out the WAGD Spill valve
Again, the drive gas is completely isolated from the circle system Drive mechanism is electrically powered Piston Ventilators
Don’t require much drive gas More accurate tidal volumes Better for patients with poor lung compliance Better for pediatrics s and small patients But, pistons have one big downfall: During that downward stroke of the expiratory phase they actually generate negative pressure in the circuit. Look again: Why pistons can be better:
So we modified the circle system: Ever wonder why the breathing bag in room 5 at children’s keeps moving while you are on the vent? Its not to amuse your during the 5 th circumcision of the day...
When the piston generates that negative pressure This valve closes to protect the patients lungs This valve opens which allows... Air to be pulled from the breathing bag So in a piston ventilator the bag is not excluded from the circle when on the vent.
In the older machines, there was no mechanism to compensate for FGF during vent inspiration. So, tidal volumes were higher than set values based upon the fresh gas flows. Another frequently tested ITE topic
Look at this: Vent While the vent is delivering your 500 ml set tidal volume... There is FGF still coming in through the inlet. Any FGF coming in during inspiration on the vent will be added to the tidal volume.
They will give you a: Rate (10 breaths/min) TV (1000 ml) I:E ratio (1:2) FGF rate (6 lpm) With that info you will calculate this: 10 breaths / min = 6 second breaths I:E of 1:2 means 2 of those seconds spent in inhalation, 10 times per minute. A total of 20 seconds inhalation per min 20 seconds x (6000 ml FGF/ 60 seconds) = 2000 ml added TV over a minute At 10 breaths/ min that is 200 ml added to the TV of each breath This is how you work through this problem on the test:
Peak pressures → highest pressure during insp. cycle Reflects dynamic compliance Plateau pressure → pressure during insp. pause Reflects static compliance IN NORMAL HEALTHY LUNGS THESE TWO NUMBERS SHOULD BE VERY CLOSE. Airway pressures:
A SEVO vaporizer will deliver an accurate concentration of an unknown volatile anesthetic if the latter shares which property with sevoflurane? A. Molecular weight B. Viscosity C. Vapor Pressure D. Blood gas partition coefficient E. Oil gas partition coefficient Question time:
Which of the following valves prevents transfilling between compressed gas cylinders? A. Fail safe valve B. Pop off valve C. Pressure sensor shutoff valve D. APL valve E. Check valve
When the pressure gauge on a size E N20 cylinder begins to fall from 750 psi, how many liters of gas remain in the cylinder? A. 200 B. 400 C. 600 D. 800 E. Cannot be calculated
For any given concentration of volatile anesthetic, the splitting ratio is dependent on which of the following characteristics of the anesthetic? A. Vapor pressure B. Barometric pressure C. Molecular weight D. Specific heat E. MAC at 1 atmosphere
Answer : A
After induction and intubation with confirmation of tracheal placement the O2 sat begins to fall. The analyzer as well as the mass spectrometer show inspired O2 concentration of 4%. O2 line pressure is 55 psi and there is a E cylinder with 1600 psi. What do you do? A. Exchange the tank B. Swith the O2 and N2O lines out C. Disconnect the hospital O2 line D. Extubate and mask ventilate E. Replace the pulse ox probe
Micrognathia Glossoptosis (retraction and posterior displaced tongue) Upper airway obstruction Frequent cleft palate Pierre Robin
Absent cheekbones Other craniofacial deformities Malformed or absent ears Conductive hearing loss Micrognathia Treacher Collins
Anomalous development of first and second branchial arches Hemifacial microsomia Ear, nose soft palate, vertebral abnormalities, often unilateral Goldenhar
Myopathy similar to central core disease Kids also have Kyphosis/Lordosis Micrognathia Cryptorchidism Pectus Carinatum Other facial features THESE KIDS ALMOST ALWAYS HAVE MH !!! King-Denborough
Hypotonia Mild developmental delay Skeletal abnormalities (scoliosis) Absent mitochondria on light microscopy VERY HIGH RISK OF MH Can be autosomal dominant, recessive, but frequently De Novo mutation of RYR1 Central Core disease
Trisomy 21 Hypothyroid Small trachea (reduce tube size) OA instability Congenital heart Dz NO INCREASED RISK OF MH Downs
Common peroneal nerve injury Foot drop Loss of eversion Loss of toe extension May also cause stretch on sciatic, causing sciatica Lithotomy
Which causes Meralgia paraesthetica Lateral Femoral Cutaneous Nerve A SENSORY ONLY NEUROPATHY Do not confuse with excessive hip flexion
Excessive retraction in lower abdominal surgery Femoral nerve Quads weakness (decreased knee extension) Numbness anterior thigh, medial leg
Obturator nerve Decreased leg adduction Numbness at the medial thigh Difficult Forceps Delivery
Thank you all for a great year ! AND GOOD LUCK SATURDAY