Analytical Spectral Devices, Inc. 5335 Stearling Drive, Suite A Boulder, CO 80301 (303) 444-6522 fax:(303) 444-6825 www.asdi.com Asd_support@asdi.com
Device Specifications Range: 325-1075 nanometers 512 channel silicon photodiode array overlaid with an order separation filter Each channel will receive light in a 1.6 nanometer nominal band Spectral resolution of approximately three nanometers at 700 nanometers
Optics Light is collected through fiber optic head called a “bare head” The bare head has a field of view (FOV) of 25 degrees. Foreoptics may be added to restrict this FOV. 10 and 1 degree FOV restricting foreoptics are included.
Review Don’t over tighten anything 1.) connect cables and FOV limiting foreoptics 2.) add battery and turn on by holding down power 3.) now you are ready to attach to the computer!!
IMPORTANT Do not plug the GPS into the serial port until you have turned on the computer. If the computer freezes you may need to turn off the ballpoint mouse in the device manager Don’t let the spectroradiometer get too hot! If the spectra looks odd and it is hot out, first do a DC. If this will not help you will need to let it cool down.