Presentation on theme: "Introduction to Analytical Chemistry"— Presentation transcript:
1Introduction to Analytical Chemistry CHAPTER 13 POTENTIOMETRY: MEASURING CONCENTRATIONS OF IONS AND OLECULES
213A General PrinciplesThe reference electrode in this figure is a half-cell with an accurately known electrode potential, Eref , that is independent of the concentration of the analyte or any other ions in the solution under study.
313A General PrinciplesPotassium chloride is a nearly ideal electrolyte for the salt bridge because the mobilities of the K⁺ ion and the Cl¯ ion are nearly equal. The net potential across the salt bridge Ej is thereby reduced to a few millivolts or less.(13-1)
413B-1 Calomel Reference Electrodes A calomel electrodeThe electrode potential of the saturated calomel electrode is V at 25°C
5Figure 13-2Figure 13-2 Diagram of a typical commercial saturated calomel electrode.
613B-2 Silver/Silver Chloride Reference Electrodes The potential of this electrode is V at 25°C.
713C Dealing With Liquid-junction Potentials A liquid-junction potential develops across the boundary between two electrolyte solutions that have different compositions.Thus, hydrogen ions diffuse more rapidly than chloride ions, and, as shown in Figure 13-5, a separation of charge results.
8Figure 13-5Figure 13-5 Schematic representation of a liquid junction, showing the source of the junction potential, Ej . The length of the arrows corresponds to the relative mobilities of the ions.
913C Dealing With Liquid-junction Potentials The magnitude of the liquid-junction potential can be minimized by placing a salt bridge between the two solutions. The salt bridge is most effective if the mobilities of the negative and positive ions in the bridge are nearly equal and if their concentrations are large. A saturated solution of potassium chloride is good from both standpoints.
1013D-1 Describing Metallic Indicator Electrodes Electrodes of the First KindAn electrode of the first kind is a pure metal that is in direct equilibrium with its cation in the solution.Electrode systems of the first kind are not widely used for potentiometric determinations for several reasons. For one, metallic indicator electrodes are not very selective and respond not only to their own cations but also other more easily reduced cations.
1113D-1 Describing Metallic Indicator Electrodes Electrodes of the First KindMany metal electrodes, such as zinc and cadmium, can only be used in neutral or basic solutionsOther metals are so easily oxidized that their use is restricted to solutions that have been deaerated.Certain harder metals, such as iron, chromium, cobalt, and nickel, do not provide reproducible potentials.
1213D-1 Electrodes of the First Kind Electrodes of the Second KindEquation 13-4 shows that the potential of a silver electrode is proportional to pCl.Mercury serves as an indicator electrode of the second kind for the EDTA anion Y⁴¯.
1313D-1 Electrodes of the First Kind Inert Metallic Electrodes for Redox SystemsPlatinum, gold, palladium, and carbon can be used to monitor redox systems.
1413D-2 Characterizing Membrane Electrodes Membrane electrodes are sometimes called p-ion electrodes because the data obtained from them are usually presented as p-functions, such as pH, pCa, or pNO₃ .
1513D-3 The Glass Electrode for Measuring pH The cell consists of a glass indicator electrode and a saturated calomel reference electrode immersed in the solution of unknown pH. The indicator electrode consists of a thin, pH-sensitive glass membrane sealed onto one end of a heavy-walled glass or plastic tube.
16Figure 13-9Figure 13-9 Typical electrode system for measuring pH.
1713D-3 The Glass Electrode for Measuring pH The Composition and Structure of Glass MembranesAs shown in Figure 13-11, a silicate glass used for membranes consists of an infinite three-dimensional network of groups in which each silicon is bonded to four oxygens and each oxygen is shared by two silicons. Within the interstices of this structure are sufficient cations to balance the negative charge of the silicate groups.
18Figure 13-10Figure Diagram of a glass/calomel cell for the measurement of pH.
19Figure 13-11Figure (a) Cross-sectional view of a silicate glass structure. In addition to the three bonds shown, each silicon is bonded to an additional oxygen atom, either above or below the plane of the paper. (Adapted with permission from G. A. Perley, Anal. Chem., 1949, 21, 395. Copyright 1949 American Chemical Society.) (b) model showing three-dimensional structure of amorphous silica with Na ion incorporated.
2013D-3 The Glass Electrode for Measuring pH Membrane PotentialsThe lower part of Figure shows four potentials that develop in a cell when pH is being determined with a glass electrode.EAg,AgCl and ESCE, are reference electrode potentials that are constant. A third is the net potential across the salt bridge that separates the calomel electrode from the analyte solution.
2113D-3 The Glass Electrode for Measuring pH Membrane PotentialsJunction potential, EjThe fourth, is the boundary potential, Eb ,which varies with the pH of the analyte solution.
2213D-3 The Glass Electrode for Measuring pH The Boundary PotentialThe boundary potential is then a measure of the hydrogen ion activity of the external solution.(13-9)(13-10)
2313D-3 The Glass Electrode for Measuring pH The Asymmetry PotentialWhen identical solutions and reference electrodes are placed on the two sides of a glass membrane, the boundary potential should in principle be zero. In fact, however, a small asymmetry potential that changes gradually with time is frequently encountered.
2413D-3 The Glass Electrode for Measuring pH The Glass Electrode Potential
2513D-3 The Glass Electrode for Measuring pH The Alkaline ErrorIn basic solutions, glass electrodes respond to the concentration of both hydrogen ion and alkali metal ions.
26Figure 13-12Figure Acid and alkaline error for selected glass electrodes at 25°C. [From R. G. Bates, Determination of pH, 2nd ed. (New York: Wiley, 1973), p With permission.]
2713D-3 The Glass Electrode for Measuring pH The Alkaline ErrorThe alkaline error can be satisfactorily explained by assuming an exchange equilibrium between the hydrogen ions on the glass surface and the cations in solution.(13-12)
2813D-3 The Glass Electrode for Measuring pH For the glasses used for pH electrodes, Kex is so small that the activity ratio b'1 / a'1 is ordinarily minuscule. The situation differs in strongly alkaline media.The activity of the sodium ions relative to that of the hydrogen ions becomes so large that the electrode responds to both species.
2913D-3 The Glass Electrode for Measuring pH The Acid ErrorAll the causes of the acid error are not well understood, but one source is a saturation effect that occurs when all the surface sites on the glass are occupied with H⁺ ions.
3013D-4 Glass Electrodes for Other Cations Glass compositions that permit the determination of cations other than hydrogen. Incorporation of Al₂O₃ or B₂O₃ in the glass has the desired effect.Glass electrodes that permit the direct potentiometric measurement of such singly charged species as Na⁺, K⁺, NH4+,Rb⁺, Cs⁺, Li⁺, and Ag⁺ have been developed.
3113D-5 Liquid-Membrane Electrodes Figure is a schematic of a liquid-membrane electrode for calcium. It consists of a conducting membrane that selectively binds calcium ions, an internal solution containing a fixed concentration of calcium chloride, and a silver electrode that is coated with silver chloride to form an internal reference electrode.
32Figure 13-13Figure Diagram of a liquid-membrane electrode for Ca2.
3313D-6 Crystalline-Membrane Electrodes A membrane with cationic sites might be expected to respond selectively toward anions.An electrode based on a polycrystalline Ag₂S membrane is offered by one manufacturer for the determination of sulfide ion.Mixtures of PbS, CdS, and CuS with Ag₂S provide membranes that are selective for Pb²⁺,Cd²⁺,and Cu²⁺.
3413D-6 Crystalline-Membrane Electrodes A crystalline electrode for fluoride ion is available from commercial sources.The membrane consists of a slice of a single crystal of lanthanum fluoride that has been doped with europium(II) fluoride to improve its conductivity.
3513D-7 Ion-Sensitive Field Effect Transistors (ISFETs) The field effect transistor, or the metal oxide field effect transistor (MOSFET), is a tiny solid state semiconductor device.ISFETs offer a number of significant advantages over membrane electrodes, including ruggedness, small size, inertness toward harsh environments, rapid response, and low electrical impedance.ISFETs do not require hydration before use and can be stored indefinitely in the dry states.
3613D-8 Gas-Sensing ProbesFigure consists of a tube containing a reference electrode, a specific ion electrode, and an electrolyte solution. A thin, replaceable, gas-permeable membrane attached to one end of the tube serves as a barrier between the internal and analyte solutions.13-36
37Figure 13-16Figure Comparison of a liquid-membrane calcium ion electrode with a glass pH electrode. (Courtesy of Orion Research, Boston, MA.)
3813D-8 Gas-Sensing ProbesThe Mechanism of Response
4013F-1 Equations Governing Direct Potentiometry indicator electrode is always treated as the right-hand electrode and the reference electrode as the left-hand electrode(13-17)(13-18)
4113F-1 Equations Governing Direct Potentiometry The constant terms in parentheses can be combined to give a new constant K.(13-19)(13-20)
4213F-2 Applying the Electrode-Calibration Method Electrode Response: Activity or Concentration?Electrode response is related to analyte activity rather than analyte concentration.The difference between activity and concentration is illustrated in FigureActivity coefficients for singly charged species are less affected by changes in ionic strength than are the coefficients for ions with multiple charges.
43Figure 13-17Figure Response of a liquid-membrane electrode to variations in the concentration and activity of calcium ion. (Courtesy of Orion Research, Boston, MA.)
4413F-4 Making pH Measurements with the Glass Electrode¹⁰ Giving an Operational Definition for pHThe operational definition of pH endorsed by the National Institute of Standards and Technology (NIST)When these electrodes are immersed in a standard bufferWhen the electrodes are immersed in a solution of unknown pH,
4513F-4 Making pH Measurements with the Glass Electrode¹⁰ That is, the operational definition of pH does not yield the exact pH as defined by the equation(13-24)
4613G Following Titrations By Potentiometry A potentiometric titration involves measurement of the potential of a suitable indicator electrode as a function of titrant volume.Potentiometric titrations offer advantages over direct potentiometry. Because the measurement is based on the titrant volume that causes a rapid change in potential near the equivalence point, potentiometric titrations are not dependent on measuring absolute values of Ecell .
4713G Following Titrations By Potentiometry The results are obtained in terms of the analyte concentration even though the electrode responds to activity. Hence, ionic strength effects are unimportant in the titration procedure.
48Figure 13-19Figure Titration of mmol of chloride ion with M silver nitrate. (a) Titration curve. (b) First-derivative curve. (c) Second-derivative curve.