Download presentation

Presentation is loading. Please wait.

Published byNathaniel Hatchell Modified over 2 years ago

1
DESIGN REQUIREMENTS FOR SENSORS BASED ON AMORPHOUS WIRES RESULTING FROM MEASUREMENTS AND SIMULATION

2
Introduction Amorphous wires are attractive components for sensor and micro- sensor realization. The characteristics of the sensing element can be carried-out starting from the measurement results. A model of a magneto-impedance sensing element was obtained and was used for a PSPICE simulation step. The paper presents the results obtained at radio-frequency, at once with the mathematical modeling of the sensitive part. In addition, there are also the requirements which must be fulfilled in the design of this kind of sensors.

3
Current sensor based on magnetic amorphous wire

4
The fitting law from previous results

5
Simulation Step I/H I Z LS H ex I HF Z(H ex, I HF, T a ) H NF Z(H ex, I HF ) HQHQ TaTa I 1 = 0.2 A Figure 3. Modeling of a current sensor with amorphous wire by two transducers; H Q - biasing magnetic field, H NF - negative feedback field Figure 4. Collpits oscillator for amorphous magnetic wire based on sensor

6
PSPICE simulation circuit Figure 5. Electrical diagram for PSPICE simulation

7
Simulation of GMI

8
Simulated Results Time 0s10us20us I(F1) 0A 400mA 800mA I 1 = 0 I 1 = 0.1 A I 1 = 0.2 A I 1 = 0.3 A I 1 = 0.4 A Figure 6. PSPICE results of the feedback current for various input currents. I_I 1 0200mA 400mA (Min(I(F1))+Max(I(F1)))/2 (Min(I(H2))+Max(I(H2)))/2 0A 100mA 200mA 300mA 400mA Figure 7. Transfer characteristic of the feedback current I F1 vs. I 1.

9
Simulated Results I_I1 0 200mA 400mA 2*Max(V(E1:OUT+) - V(E1:OUT-))/(Max(I(H1)) - Min(I(H1))) 55 10 15 20 Figure 8. Amorphous wire resistance variation vs. I 1. I_I1 0200mA400mA Period(I(H1)) 31.2ns 31.4ns 31.6ns 31.8ns Figure 9. Period of oscillation of the test current I H1 vs. I 1.

10
Simulated Results Frequency 50MHz100MHz30MHz130MHz I(H1) 1.0 A 100 A 10mA Figure 10. Power spectral density of the test current I H1 ; parameter - I 1. I_I1 0200mA400mA Max(I(H1))/1.41 6.0mA 7.0mA 8.0mA 9.0mA Figure 11. The effective value of the current I H1 through the amorphous wire vs. I 1.

11
Conclusions The characteristics of the amorphous wires can be theoretically modeled, but vectorial measurements are important for obtaining the fitting laws in the real conditions. Starting from the experimental data [1], the mathematical, informational and energetic models are developed as a first step in PSPICE modeling for a current sensor based on magnetic amorphous wires. In the simulation phase, magnetic amorphous wires based sensors nearly complete behaviors were obtained. The resulted fast balancing loop, obtained by applying an incomplete compensation method, has good behaviors proved by many quality parameters. Moreover, not easy to do in current measurements are obtained as function of the dc input quantity. A compact non-contact current sensor and its electronic processing unit, designed on Collpits oscillator structure, can be developed starting from the presented simulation diagram. The aim for the future is to extend the study of this current sensor in ac and to find the requirements for signal processing in this case.

12
References [1] E. Vremera, C. Fosalau, L. Brunetti, Instrumentation Set-Up for Characterization of the Sensors Based on Amorphous Wires, IMEKO TC4 Symposium, pp. 458-461, Gdinya (Poland), 2005 [2] L. Brunetti, et al., “Magneto-impedance measurements in amorphous Co- based magnetic wires at high frequency”, Journal of Magnetism and Magnetic Materials, no. 249, pp. 310–314, 2002. [3] K. Bushida, et al., Sensitive and Quick Response Micro Magnetic Sensor Using Amorphous Wire MI Element Collpits Oscillator, IEEE Transaction on Magnetics, vol. 31, no. 6, pp. 3134-3136, 1995 [4] A.F.P. van Putten, Electronic Measurement Systems, Institute of Physics Publishing, 1996. [5] F. Monssen, MicroSim PSpice with circuit analysis - 2nd ed. - Prentice Hall, 1998.

Similar presentations

OK

Investigation of different types radio sources by IPS method at 111MHz S.A.Tyul’bashev Pushchino Radio Astronomy Observatory, Astro Space Center of P.N.Lebedev.

Investigation of different types radio sources by IPS method at 111MHz S.A.Tyul’bashev Pushchino Radio Astronomy Observatory, Astro Space Center of P.N.Lebedev.

© 2018 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Ppt on mammals and egg laying animals Free ppt on brain machine interface technology Ppt on service design and development Ppt on centring concrete Training ppt on email etiquettes Ppt on google chrome web browser Ppt on water resources for class 4 Export pdf to ppt online converter Ppt on political parties and electoral process in the philippines Ppt on social contract theory example