Download presentation

Presentation is loading. Please wait.

Published byAmara Creekmore Modified over 2 years ago

1
TPOL Test Beam Telescope Chris Collins-Tooth (ZEUS, IC-London)

2
Outline zTest Beam setup of the Telescope and TPOL zWhat is the Telescope and what does it do? zWhat data was gathered? zAnalysis of the data yMultiple Coulomb Scattering, Beam spread, and Telescope resolution yRelative rotations xbetween parts of the Telescope xbetween the Telescope and the TPOL silicon zWhat can be done about these factors? zWhat does this tell us about the TPOL silicon? yTPOL silicon resolution yTPOL silicon efficiency zSummary

3
Test Beam setup z6 GeV e - beam enters from left ze - beam passes through Telescope then moves into the TPOL zTelescope mounted as close to TPOL as possible on movable table zTelescope has 3 position sensitive detectors T1,T2 and T3 (Td was ‘dead’ material being used for a second experiment) Telescope TPOL

4
The Telescope zPreviously, degree of polarisation estimated using energy asymmetry in calorimeter (Calorimeter resolution ~1000 m) zNow measure polarisation using 80 m pitch Si zSomething more accurate needed to probe TPOL silicon resolution - the Telescope. z3 planes of 50 m pitch Si, with horizontal and vertical strips. zT1,T2,T3 detectors roughly 3cm × 3cm (TPOL silicon ~1 cm 2 ) 50 40 30 20 10 0 -10 60 P(%) t(min)

5
The data zT1,T2,T3 used to predict Si strip to fire. zAs expected, fitted line has slope =1 0.01 zOffset simply due to T1,2,3 being physically larger than TPOL Silicon zWidth of data about fitted line gives indication of TPOL resolution

6
The width zWidth =132.15 ± 2.93 m zTPOL Silicon strip pitch =80 m zIntrinsic resolution ~80/ 12 m 0 0400-400800-800

7
Analysis of the data zObserved width does not relate directly to the TPOL resolution zMultiple Coulomb Scattering (MCS) of e - beam at T1,Td,T2,T3 and TPOL Aluminium Box zFinite resolution of the Telescope ze - beam not 100% collimated zMisalignments of the Telescope detectors T1,T2,T3 zMisalignments of the Telescope (as a whole) and the TPOL

8
Telescope internal misalignment zT1,T2 and T3 could all be misaligned with respect to each other. zRotations would produce systematic shifts of ‘predicted minus actual’ strip firing from left-to-right zMost important are rotations about beam-axis (pictured). A 0.08 o rotation would cause a shift of 1 strip across breadth of detector

9
‘Predicted minus Actual’ shifts for T3,T2 and T1 from L R zUsing T1,T2 to predict T3 (left) we observe a shift from left to right of approximately 50 microns T3T2T1

10
Correction of misalignment zIterative process invoked zRotating T3 by 0.27 o flattened off all the plots (to within errors)

11
TPOL misalignment zThe TPOL silicon had no vertical strips zTrack through T1,T2,T3 used to predict vertical strip to fire to give indication of horizontal position of impact zNo discernable shift observed before or after T3 rotation applied zCorrection for rotations caused no discernable reduction in observed ‘width’

12
MCS, Telescope resolution and beam collimation zSimple Monte-Carlo simulation using PDG formula for MCS with gaussian width: s =(13.6MeV/ cp) (x/X o ) (1+0.038 n [x/ X o ]) zTelescope resolution and beam collimation are small factors in comparison to MCS zTogether, all these factors contribute ~102 m to the width zSubtracting in quadrature, the TPOL resolution obtained is (132 2 -102 2 ) 83 m zBut - MCS is not actually gaussian zAttempting to use GEANT to improve estimate

13
Error propagation zAlternative approach to Monte-Carlo zUse errors introduced by MCS etc., and propagate them to the TPOL silicon zT 4 =T 3 +Z 4 m 3 +Z 4 3 +(Z 4 -Z Al ) Al y Var(T 4 )=Var(T 3 )+Z 4 2 Var(m 3 )+Z 4 2 Var( 3 )+(Z 4 -Z Al ) 2 Var( Al )+Z 4 Cov(T3,m 3 ) zT 3 =T 1 -Z 1 m 3 -Z d d -Z 2 2 y Var(m 3 )=(1/Z 1 2 )(T 1 -T 3 -Z d d -Z 2 2 ) zVariances are calculated, (e.g. Var(T1,T2,T3)= 2 T1,T2,T3 =208 m) z T4 = [Var(T 4 )]=120 m= Error on TPOL Si due to uncertainties in Telescope zResolution of TPOL Si = [132 2 -120 2 ]=55 m T1T1 T2T2 TdTd T3T3 AlT 4 (TPOL) +z m3m3 m3m3 m3+3m3+3

14
TPOL Resolution zFrom Monte-Carlo Simulation we obtain R TPOL 83 m zFrom Error Propagation we obtain R TPOL 55 m

15
TPOL Silicon Efficiency zTelescope used to predict TPOL events zEfficiency is the ratio of: events with TPOL Silicon signal events predicted by Telescope zTPOL Silicon edges found by looking at ratio of hits not registering in the TPOL as a function of position zLog plot reveals edges where Telescope predicts TPOL hits but TPOL does not register zHorizontal edges at 13000 & 20000 m zVertical edges at 11000 & 18000 m zConsistent with active area of Silicon Horizontal Position ( m) Vertical Position ( m) Ratio of hits NOT registering in TPOL

16
Efficiency cuts zUsing the edges from previous slide, we must remove predicted events from efficiency calculations where they miss the boundaries of the TPOL Si zFigures show yevents predicted by the Telescope and registered by the TPOL (black) yFor effect, events in red are added. They are predicted events which had no TPOL response, but the event was inside the opposite direction boundary, and so should have registered. zClearly, the boundaries look correct.

17
Final Efficiency zWith the positional cuts made, the efficiency of the TPOL silicon can be calculated zThis is the ratio of : events with a TPOL silicon response all predicted events inside the boundary zThe efficiency is uniform across the detector, at ~97.8%

18
Summary zMisalignments, Coulomb Scattering and other factors contribute significantly to observed width of 132 m. zTPOL Silicon resolution TBA but preliminary studies suggest 55 and 83 m zTPOL Silicon efficiency uniform at 97.8%

Similar presentations

OK

Lepton Polarisation at HERA A brief overview of lepton polarisation - its use and measurement at HERA, including testbeam analysis of the performance of.

Lepton Polarisation at HERA A brief overview of lepton polarisation - its use and measurement at HERA, including testbeam analysis of the performance of.

© 2017 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Ppt on south african culture and food Ppt on cross site scripting xss Ppt on role of it in indian railways Ppt online download for youtube Ppt on steve jobs download Field emission display ppt online Ppt on cross site scripting internet Ppt on astronomy and astrophysics salary Ppt on windows 2000 operating system Ppt on why technology is important in our daily life