Download presentation

Presentation is loading. Please wait.

Published byMarcelo Chaffey Modified over 2 years ago

1
Heat Capacity II Sound Velocity Method CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009 Instructor: Dr. Aleksey I. Filin

2
Sound Velocity v S can be expressed as: CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009 Heat Capacity II Sound Velocity Method Where: Heat capacity @ constant pressure Heat capacity @ constant volume R – universal gas constant T – temperature M – molecular weight Instructor: Dr. Aleksey I. Filin If we know the speed of sound, we can calculate

3
Heat Capacity II Sound Velocity Method Vibrations transmitted through an elastic solid or a liquid or gas, with frequencies in the approximate range of 20 Hz to 20 kHz What the sound is? Definition of sound: Characteristics of sound: f – frequency [number of vibrations per unit of time, Hz, or s -1 ] – wavelength [m] Speed of sound can be found as Instructor: Dr. Aleksey I. Filin So, our goal is to measure the wavelength and the frequency of the sound By definition CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009

4
Heat Capacity II Sound Velocity Method Generator Scope Y1 t Speaker Microphone Instructor: Dr. Aleksey I. Filin Y2 Experimental Setup If the Generator output is connected to Y1 input of the Scope and the Microphone output is connected to Y2 input, the Scope shows 2 similar sin waveforms. CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009

5
Heat Capacity II Sound Velocity Method Generator Scope Y Speaker Microphone X Instructor: Dr. Aleksey I. Filin If the Generator output is connected to Y1 input of the Scope and the Microphone output is connected to X input, the Scope shows so called Lissajous pattern. A Lissajous pattern is a graph of one frequency plotted on the y axis combined with a second frequency plotted on the x axis. In our case both f are the same Experimental Setup CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009

6
Heat Capacity II Sound Velocity Method Mathematical description of the simplest Lissajous pattern Scope shows y as a function of x We know the time dependence both y and x Where is the phase difference between y(t) and x(t) x(t) y(t) Instructor: Dr. Aleksey I. Filin CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009

7
Heat Capacity II Sound Velocity Method Assuming X 0 =Y 0 If = 0 y(x)=x y(x) x If = (a quarter of period) y 2 (t)+x 2 (t)=1 y(x) x Instructor: Dr. Aleksey I. Filin Mathematical description of the simplest Lissajous pattern CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009

8
Heat Capacity II Sound Velocity Method Assuming X 0 =Y 0 y(x)=-x y(x) x If = 3 (three quarters of period) y 2 (t)+x 2 (t)=1 y(x) x If = (a half of period) Instructor: Dr. Aleksey I. Filin Mathematical description of the simplest Lissajous pattern CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009

9
Heat Capacity II Sound Velocity Method Assuming X 0 =Y 0 y(x)=x y(x) x If = 2 (full period) Phase difference 0a quarter of period, or /4 a half of period, or /2 three quarters of period, or 3 /4 full period, or Figure on the scope Note: if X 0 = Y 0 the angle will be not 45 0 and the circles become the ellipses Full period corresponds to Instructor: Dr. Aleksey I. Filin Mathematical description of the simplest Lissajous pattern CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009

10
Heat Capacity II Sound Velocity Method Generator Scope Y Frequency is given by generator Speaker X How to measure the wavelength? Microphone Instructor: Dr. Aleksey I. Filin CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009

11
Heat Capacity II Sound Velocity Method Generator Scope Y Frequency is given by generator Speaker X How to measure the wavelength? Microphone Instructor: Dr. Aleksey I. Filin CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009

12
Heat Capacity II Sound Velocity Method Generator Scope Y Frequency is given by generator Speaker X How to measure the wavelength? Microphone Instructor: Dr. Aleksey I. Filin CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009

13
Heat Capacity II Sound Velocity Method Generator Scope Y Frequency is given by generator Speaker X How to measure the wavelength? Microphone Instructor: Dr. Aleksey I. Filin CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009

14
Heat Capacity II Sound Velocity Method Generator Scope Y Frequency is given by generator Speaker X How to measure the wavelength? Microphone Instructor: Dr. Aleksey I. Filin CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009

15
Heat Capacity II Sound Velocity Method Instructor: Dr. Aleksey I. Filin Summary We measure the shape of Lissajous pattern as a function of microphone position Distance between two positions corresponding to straight 45 0 line is equal to the wavelength of the sound in given gas at given sound frequency We calculate the speed of sound in given gas at given frequency using formula CHEM 4396 (W237) Physical Chemistry Laboratory Fall 2009

Similar presentations

OK

By Aimee Chavez. Wave: a disturbance that transfers energy from place to place. The material through which a wave travels through is called a medium.

By Aimee Chavez. Wave: a disturbance that transfers energy from place to place. The material through which a wave travels through is called a medium.

© 2018 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Ppt on communication skills training Ppt on event driven programming disadvantages Download ppt on c language Ppt on forward rate agreement ppt Jit ppt on manufacturing companies Ppt on diode as rectifier regulator Ppt on active magnetic bearing Ppt on metabolism of drugs Ppt on any topic in hindi Ppt on led driver