Download presentation

Presentation is loading. Please wait.

Published byOscar Berisford Modified over 3 years ago

2
BASIC SOLAR CELL TESTING

3
Basic Structure of a Solar Cell

4
Basic Photovoltaic Cell Model This model consists of Built-in voltage Current due to optical generation Series resistance Shunt Resistance

5
Key Parameters Open Circuit Voltage, V oc (V) Short Circuit Current Isc (A) Max Power Voltage, V mp Max Power Current, I mp Max Power Current Density, J mp (I mp / area in cm 2 Max photo current Fill Factor, FF Efficiency, Diode Ideality Factor Shunt Resistance Series Resistance Reverse Saturation Current or Leakage Current – indicates a poor diode

6
A Solar cell is a diode A solar cell is a diode and hence an IV curve of a solar cell under dark conditions will look similar to that of a diode. When illuminated, the photons interact with the material to generate electron hole pairs, which are then driven in opposite directions by the built-in potential.

7
Standard Test Conditions AM 1.5G Temperature = 25C Important device characteristics can be obtained from the I-V measurements.

8
Sunlight Simulator in Clean Room Air Mass 1.5 filter installed Shutter control UV Lamp Housing Cooling fan must be on UV Lamp Power Supply Not shown UV intensity meter and calibration solar cell

9
Procedure for Sunlight Simulator Verify the stage is connected to GND Verify top contact probe R1 is connected to SMU1

10
Procedure for Sunlight Simulator Make sure all fans are working Turn ON the lamp and wait for a few minutes for it to stabilize. Open shutter (Remember to wear safety goggles) Using the calibration cell and the sun meter, adjust the power supply at about 970W or 1 Sun on the sun meter. Replace the calibration cell with the test sample Make top and bottom connections to the Keithley 4200 SCS.

11
Keithley 4200 SCS tutorial Start KITE. Select vfd in diode section. (1) Set Anode to SMU1 and Cathode to GNDU. (2) Click on Force/Measure button to change sweep parameter. (3) Click Run Test/Plan button to start sweep. (4) Click Append button to do another sweep and append the data to the previous sweep. (5)

12
Keithley 4200 SCS tutorial

13
To save measured data, select the data tab (6) and click save as button on the right (7). To view the graph, select the graph tab (8). To save the graph, right click on the graph and select save as.

14
Keithley 4200 SCS tutorial

15
Open Circuit Voltage, V oc (V) In an ideal solar cell, V oc is independent of the illumination intensity. The open circuit voltage (V oc ) occurs when there is no current passing through the cell. V (at I=0) = V oc To read the open circuit voltage from the graph, locate the point on the voltage axis where the current is zero.

16
Short ircuit Current Density, I sc The short circuit current I sc corresponds to the short circuit condition when the impedance is low and is calculated when the voltage equals 0. I (at V=0) = I sc To read the short circuit current from the graph, locate the point on the current axis where the voltage is zero. To find the current density J sc, divide this current by the area of the solar cell under test, to obtain the current density, J sc (mA/cm 2 )

17
PTPT P MAX I SC I MP V MP V OC Load Line Max Power Point Draw a rectangle with the origin, V OC and ISC as the 3 corners. The 4 th corner will give the maximum theoretical power, P T. From the origin, draw a line passing through the maximum theoretical power, P T. This is the load line The point where the load line crosses the I-V curve is the maximum power point, P MAX for the solar cell, for a given load, with maximum current and maximum voltage.

18
Max Power Point The voltage at the maximum power point of the cell is the maximum voltage, V MP. The current at the maximum power point of the cell is the maximum current, I MP From the maximum power point, P MAX, draw a line perpendicular to and meet the voltage axis. The maximum power voltage, V MP is given by the value on the voltage axis. The maximum power current, I MP is given by the value on the current axis.

19
Fill Factor Fill Factor is the measure of the quality of the solar cell. It is the ratio of the maximum power, P max to the theoretical power, P T. FF = P MAX /P T FF = I MP. V MP / I sc. V oc Fill Factor is a number between 0.0 and 1.0. The higher the number, the better the solar cell

20
Efficiency Efficiency is the ratio of the electrical power output P OUT, compared to the solar power input, P IN, into the PV cell η = P OUT /P IN P OUT = P MAX (W/m 2 ) = (I mp )(V mp )/area = (J mp )(V mp ) Where J mp = I mp /area For Standard Test Conditions P IN = 1000 (W/m 2 )= 100(mW/cm 2 )

21
Reverse Saturation Current The saturation current I 0, is the current that flows in the reverse direction when the diode is reverse biased. It is also called as the leakage current. Saturation current, I o Specified voltage point for leakage current measurement

22
Shunt Resistance Shunt resistance is the change in the voltage for change in the unit current and is ideally equal to infinity.

23
Series Resistance Series resistance is due to – Resistance of the metal contacts – Ohmic losses in the front surface of the cell – Impurity concentrations – Junction depth Series resistance reduces both short circuit current and maximum power output of the cell

24
Series Resistance For the measurement of internal series resistance, 2 I-V curves of different irradiance but of the same spectrum and at the same temperature are necessary. The series resistance is calculated as: R S = (V 2 -V 1 )/(I SC1 – I SC2 ) V1V1 V2V2 I SC1 I SC2

25
Diode Ideality Factor The diode ideality factor n, is an indicator of the behavioral proximity of the device under test, to an ideal diode. n is between 1 and 2, ideally equal to 1.

26
Alternate view of I/V Plot Dark current plot (no sunlight) Current plot with 1 sun illuminating the solar cell Zero voltage, zero current point

27
Max photocurrent Max photocurrent about 160mA

28
V oc and I sc V oc – where I = 0 About 0.5v Isc – where V = 0 About 150mA J sc = (I sc )/area of cell in cm 2

29
For P max point 2) Draw a vertical line from V oc 1) Draw a horizontal line from I sc 3) Draw a line from origin to where they intersect

30
For P max point P max - Max Power point I max = 75mA V max =.35v J mp = I mp / area of cell

31
Efficiency Assume for previous example, area of cell = 6 cm 2 η = P OUT /P IN P out = P max (W/m 2 ) = (I mp )(V mp )/area P max = (75mA)(.35V) = 26mW for 6cm 2 = 4.3mW/cm 2 Or Jmp = 75mA/6cm 2 =12.5mA/cm 2 P max = (J mp )(V mp ) = (12.5mA/cm 2 )(.35V) = 4.3 mW/cm 2 For Standard Test Conditions P IN = 1000 (W/m 2 )= 100(mW/cm 2 ) n = (4.3 mW/cm 2 )/(100mW/cm 2 )x 100% n = 4.3% Efficiency is an area dependent parameter – you must factor in the area. Use J ms and V mp or I mp /area and V mp for output power calculations

32
Fill Factor Fill Factor is a number between 0.0 and 1.0, the higher the number, the better the solar cell Fill Factor is the ratio of the maximum power, P max to the theoretical power, P T. FF = P MAX /P T FF = I MP. V MP / I sc. V oc For the previous sample: FF = (.075A)(.35V)/(.150A)(.50V) FF =.02625 VI/.0750 VI FF =.35 (unit less)

33
Assignment on I/V plots due next class There will be a test on solar cell I/V plot analysis

Similar presentations

OK

Recall-Lecture 5 DC Analysis Representation of diode into three models

Recall-Lecture 5 DC Analysis Representation of diode into three models

© 2018 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Ppt on tropical deciduous forest in india Ppt on industrial revolution in europe Ppt on wireless network technologies Download ppt on coordinate geometry for class 9th result Ppt on trial and error approach Ppt on articles of association for social clubs Waters view ppt online Create ppt on ipad 2 Ppt on contract labour act 1970 Ppt on bionics