We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byJordan Reese
Modified over 2 years ago
© 2011 SRS Medical Systems, Inc. All rights reserved. 1 The In-Flow Intraurethral Valve-Pump 29 March 2011
© 2011 SRS Medical Systems, Inc. All rights reserved. 2 How do you get urine out of the bladder without causing infection? 100,000,000 urinary catheters are used globally every year, but they cause infections that kill tens of thousands and cost billions in medical care The In-Flow has been shown to have the lowest infection rate by far of any indwelling device for bladder drainage Active stent-like device replaces passive urinary catheters The In-Flow is also the most discrete and dignified solution for female bladder drainage Significantly improves quality of life Advanced Technology for More Natural Voiding
© 2011 SRS Medical Systems, Inc. All rights reserved. 3 In-Flow Indication Replaceable urinary prosthesis intended to normalize toileting in women with atonic bladder or other voiding dysfunction resulting in an inability to fully empty the bladder A prosthetic device compensates for a specific physiologic deficiency The In-Flow compensates for the lack of bladder pressure in women with atonic bladder by providing forceful, virtually complete evacuation of urine on demand
© 2011 SRS Medical Systems, Inc. All rights reserved. Design and Materials Miniature valve-pump mechanism entirely housed in medical-grade silicone Supplied sterile, 29 day use Sized to length of patients urethra (9 lengths) Flexible bladder fixation - simple and safe removal 4 Female-only Intraurethral Device Magnetic remote-control (Activator) Spins internal magnet at 10,000 RPM, creating turbine pump Engages valve automatically when pump is not in use to maintain continence
© 2011 SRS Medical Systems, Inc. All rights reserved. 5 Principles of Operation - Insertion Device is inserted using disposable introducer Flexible silicone fins hold it in place
© 2011 SRS Medical Systems, Inc. All rights reserved. Principles of Operation - Voiding 6 Patient normally sits on a toilet Patient / caregiver presses the button Pump is activated and provides normal flow, quickly draining the bladder Releasing the button automatically engages the valve, stopping further flow
© 2011 SRS Medical Systems, Inc. All rights reserved. Target Patient Population 7 Women with atonic bladder Lack bladder pressure - cannot spontaneously urinate Typically due to neurologic injury or disease (MS, stroke, SCI, diabetic neuropathies, etc.) The inability to void is one of lifes most discouraging circumstances Incurable condition with very limited treatment options Typically requires life-long use of urinary catheters
© 2011 SRS Medical Systems, Inc. All rights reserved. 8 Present Clinical Alternatives Urinary catheters may be the simplest and most commonly used medical devices, but they cause serious problems, notably: Low quality of life, and High rates of urinary tract infection (UTI) Problems amplified with atonic bladder - must use catheters for life New treatments exist for neurogenic OAB (overactive bladder), but not for atonic (underactive) bladder Two types of urinary catheters are most often used for women Intermittent catheters Indwelling (Foley) catheters
© 2011 SRS Medical Systems, Inc. All rights reserved. 9 Clean Intermittent Catheterization (CIC) Current standard of care for long term bladder drainage Known to have substantially lower clinical UTI rate than indwelling (Foley) catheters However, many cannot reliably self-catheterize Lack necessary manual, visual or cognitive ability Particularly true of many women with atonic bladder due to serious nature of their primary medical condition Others simply will not perform this procedure Find repeated touching of genital area distressing Elderly women in particular often opt for Foleys despite well understood problems Patient inserts a tube into her urethra 4-8 times per day for voiding
© 2011 SRS Medical Systems, Inc. All rights reserved. 10 Indwelling (Foley) Catheters Held in place via a water-filled balloon in the bladder Medically demanding 5% risk of UTI per day (virtually 100% in 30 days) 50% encrustation, routine skin/tissue problems Typically connected to a urine collection bag
© 2011 SRS Medical Systems, Inc. All rights reserved. 11 Psychologically challenging Literally tied to a bag of your own urine Many patients consider this an end-stage development Correctly or not
© 2011 SRS Medical Systems, Inc. All rights reserved. 12 Comparison to Urinary Catheters In-FlowFoleysCIC Low UTI rate Easy to use Convenient Hygienic Positive body image Allows use of toilet Based on clinical evidence to date, In-Flow compares favorably to both CIC and indwelling (Foley) catheters Compared to CIC, improves quality of life with similar safety and effectiveness Compared to Foley catheters, improves quality of life with vastly superior safety
© 2011 SRS Medical Systems, Inc. All rights reserved. 13 In-Flow Evidence Six clinical studies (total n =510), including two long-term studies Extensive laboratory testing, including independently conducted encrustation study 12 years of patient use in Germany, Canada and Australia
© 2011 SRS Medical Systems, Inc. All rights reserved. 14 IDE Study Design 18-site study ( n =273) to compare safety, effectiveness and patient satisfaction of the In-Flow device versus clean intermittent catheterization (CIC) CIC is current standard of care for long-term bladder drainage Relevant clinical endpoints were selected Primary endpoint of PVR objectively indicates how effectively each device drains bladder Required excluding subjects using indwelling (Foley) catheters Secondary endpoint of QoL indicates effects of using these highly personal devices Wagner I-QOL most appropriate instrument available at time of study All aspects of study independently monitored Study limited to women with atonic bladder who were already using CIC Single-arm crossover design in which each subject acted as her own control
© 2011 SRS Medical Systems, Inc. All rights reserved. 15 IDE Study Results All clinical endpoints were met 1.Primary Endpoint: Post-void residuals (PVRs) - CIC and the In-Flow were equivalent in their ability to fully empty the bladder (98% same) 2.Secondary Endpoint: Quality of life per Wagner I-QOL - Scores were 50% higher for In-Flow than for CIC (clinically & statistically significant superiority) 3.Comparative safety: UTI rates were equivalent, although In-Flow is indwelling and CIC is intermittent (breakthrough technology)
© 2011 SRS Medical Systems, Inc. All rights reserved. 16 Clinical Role The In-Flow distinguished itself in two areas of its IDE study Quality of life scores were significantly higher than for CIC 50% better in the IDE and 80% better in another study In-Flows UTI rate was equivalent to CIC – unprecedented for an indwelling device Showing a UTI rate comparable to that for intermittent catheters is a unique and powerful effect Even so, the In-Flow is probably better positioned on basis of its ability to improve quality of life Can be quickly driven by patient demand vs. requiring large-scale studies Exception is that present data re UTI rate are probably adequate to drive adoption by indwelling (Foley) users
© 2011 SRS Medical Systems, Inc. All rights reserved. 17 Independent Studies Five independent (non-company sponsored) clinical studies (total n =239) report similar findings to the IDE study Device performed as expected in all instances No serious or lasting adverse events were reported in any study Specific findings from the two long-term studies include: In a one-year study of 20 acontractile (atonic) bladder patients, Lynch et al reported 80% improvement in quality of life (QoL) and no negative tissue changes: This study shows that the Inflow device provides an effective method of bladder drainage, with few side-effects and significant improvement in QoL. In a pre-IDE study of 40 patients with voiding dysfunction in which 21 patients were followed for more than a year with a mean follow-up time of 24.6 months, Madjar et al concluded that, although dropout was a problem (due largely to discomfort): Women who continue treatment for a prolonged time are satisfied with the device use. In fact, All patients were satisfied with the device and preferred it to previous treatment modalities used.
© 2011 SRS Medical Systems, Inc. All rights reserved. 18 Laboratory Testing A series of pre-clinical bench studies were conducted to characterize the performance of the In-Flow system. The results of these tests demonstrated that the In-Flow device and Activator meet their performance specifications and, where applicable, conform with the requirements of relevant voluntary standards. The tests performed were as follows: Device Pull-out Force Testing (as per "Inflated Balloon Response to Traction" test in ASTM F Standard Performance Specification for Foley Catheters) Device Flow Rate Testing (as per "Flow Rate through Drainage Lumen" test in ASTM F623-89) High Pressure Test (seal maintained under 200 cm H20 bladder pressure) Device Pump and Valve Endurance Test (1,140 voiding cycles = 6 months use) Activator Endurance Testing (11,552 operation cycles=5 years) Activator Drop Testing (50 cm onto hard surface) Activator Battery Endurance Testing (2 months) Catheter's DC Magnetic Field Levels (Alpen Committee standards) Activator DC Magnetic Field Testing (Alpen Committee standards) Activator AC Magnetic Field Testing (IEEE Standard for Safety Levels with Respect to Human Exposure to Radio Frequency Electromagnetic Fields 3 kHz to 300 GHz )
© 2011 SRS Medical Systems, Inc. All rights reserved. 19 Biocompatibility Testing A series of biocompatibility tests was conducted to characterize the biocompatibility of the materials used in the In-Flow device, inserter, Activator, and Sizing Device. These tests demonstrated that these components do not cause cytotoxic response, irritation, or sensitization. The device was also shown to be non- mutagenic. The tests were conducted by NAmSA on final, gamma radiation-sterilized devices, in accordance with GLP regulations. The tests conducted are listed below, according to component. In-Flow Device: Cytotoxicity (MEM elution as per ISO ) Vaginal irritation in rabbits Sensitization (guinea pig maximization) Genotoxicity (Ames mutagenicity test) Implantation testing (as per ISO ) Urinary bladder irritation study in rabbit (surgical implantation) Urinary bladder irritation study in rabbit (saline extract) In-Flow Device Magnet Assembly: Cytotoxicity testing of nickel and parylene coated magnets (MEM elution as per ISO ) Acute systemic toxicity of nickel and parylene coated magnets (U.S.P. methodology) Acute intracutaneous reactivity of nickel and parylene coated magnets (ISO Tests for Irritation and Sensitization) Corrosion potential of nickel and parylene coated magnets Individual In-Flow Device Materials: Silicone elastomerElastosil (Certificate of Compliance with USP Class VI Biological Tests) PolycarbonateLexan HF 1140 (USP Class VI) Parylene Coating (USP Class VI) AdhesiveLoctite 3311 (USP Class VI) In-Flow Inserter (Nylon 6,6; transient contact with labia): Cytotoxicity (MEM elution as per ISO ) Acute intracutaneous reactivity (ISO Tests for Irritation and Sensitization) Sensitization (guinea pig maximization test) In-Flow Activator (Nylon 6,6: brief contact with hands and abdomen): Cytotoxicity (MEM elution as per ISO )
© 2011 SRS Medical Systems, Inc. All rights reserved. Encrustation Study Stickler et al 1 – in vitro study – showed In-Flow encrustation resistance to be at least 8.4x superior to an all-silicone Foley: Under conditions that simulated a heavy infection of P. mirabilis, where a conventional Foley catheter blocked with crystalline biofilm after 25.7 hours, the In-Flow device drained the bladder for at least 9 days… (and its) central lumen appeared to be essentially clear Microbiology Research Group at Cardiff UK (internationally recognized experts on CAUTI)
© 2011 SRS Medical Systems, Inc. All rights reserved. 21 Transformational Technology The In-Flow allows use of a toilet Psychologically significant as that is the normal way to void Promotes hygiene, personal dignity Eliminates need to self-cath 4-8 times daily Eliminates tubes, drainage bags It can simply, yet absolutely transform the quality of individuals lives. - Brother of IDE patient …the device has been an unqualified success. It is difficult to put into words the effect that (In-Flow) has had on (my daughters) life. - Father of IDE patient
© 2011 SRS Medical Systems, Inc. All rights reserved. 22 Thank You
© 2011 SRS Medical Systems, Inc. All rights reserved The In-Flow device conforms to Council Directive 93/42/EEC and bears the CE mark. Authorized European Representative: CAUTION: Not cleared by the FDA for sale in the USA. Investigational device. Limited by US law to investigational use. U.S. Patent Nos. 5,762,599 and 6,417,750 and related patents worldwide. Additional patents pending. In-Flow is a trademark of SRS Medical Systems, Inc. © SRS Medical Systems, Inc. All rights reserved.
© 2011 SRS Medical Systems, Inc. All rights reserved. 24 New: Sizing Device In order to improve the accuracy of In-Flow sizing, both the Sizing Device and measurement protocol have been revised A graduated version of the longest In-Flow device (7cm) is now used to measure urethral length Bladder neck contact is now consistent, minimizing sizing errors The measurement protocol now includes several postures and not just the lithotomy position in order to account for possible prolapse
Internal Quality Assurance seminar Common challenges among QAA ENQA Review Committee perspective Guy Aelterman Den Haag,15-16 June 2009.
Epic2: National evidence-based guidelines for preventing healthcare-associated infections in NHS hospitals in England Funded and endorsed by the Department.
1 © Medtronic, Inc Medtronic Pain Therapies Referral Strategies for Managing Chronic Pain.
Briefing for Interagency Review June 4, 2012 Draft Proposal for the Particulate Matter (PM) National Ambient Air Quality Standards (NAAQS) Washington,
George Allen PhD, CIC, CNOR. No Disclosures Identify mandates, clinical & regulatory for monitoring and preventing CAUTI Review the surveillance.
CHANGES IN THE MEDICAL GAS AND VACUUM SYSTEM REQUIREMENTS P RESENTED B Y : J ONATHAN C. W ILLARD, CHC, PMP, CMGV C ERTIFIED M EDICAL G AS S ERVICES.
Training Manual on Risk Assessment of Living Modified Organisms in the context of the Cartagena Protocol on Biosafety (aligned with the Roadmap for Risk.
1Elsevier items and derived items © 2007 by Saunders, an imprint of Elsevier, Inc. Chapter 4 The Leadership Role of the Licensed Practical Nurse.
1 by L Goel Professor & Head of Division of Power Engineering School of Electrical & Electronic Engineering Nanyang Technological University,
OR-OSHA 103 Job Hazard Analysis This material for training purposes only 1 Welcome! Welcome to OR-OSHA Course 103: Conducting a Job Hazard Analysis (JHA).
FDA/Industry Workshop September 14-16, 2005 Critical Path Initiative: What it means for pharmaceutical industry statisticians Walter Offen, Lilly Brenda.
Final Report – November 3, 2003 Organization of American States Management Study of the Operations of the General Secretariat Part I – Executive Summary.
Washington Laboratories (301) web: Lindbergh Dr. Gaithersburg, MD ROHS and WEEE Directives -After the Deadline- September.
Schunk Kohlenstofftechnik SKT / GB4 Page: 1 Frequently asked questions Frequently asked questions - FAQ.
ISO INTRODUCTION In the present day, environmental matter is not limited only in one country or specific area. The environmental impact effects.
Presentation : IMS – Tech Managers ConferenceAuthor : IMS StaffCreation date : 08 March 2012Classification : D3Conservation :Page : # 04 - Power and Control.
MFG Assessment Application: Assessment Criteria and Metrics 1 Performance assessment criteria and metrics may be used as the basis for determining the.
PLANNING THE AUDIT Individual audits must be properly planned to ensure: Appropriate and sufficient evidence is obtained to support the auditors opinion;
The External Auditors Perspective and use of Internal Audit Brent Currey Live Seminar 9:00am – 4:30pm October 12, 2011 Relationships backed by performance.
Aseptic Technique, Sterile Compounding: Intravenous and Admixture Lecture #1 IV PREPARATION COURSE.
1 Pacemaker Patient Follow-up Module 8. 2 Objectives List the steps for performing a pacemaker follow-up Locate threshold, sensing and impedance data.
CREATING A SAFETY PROGRAM for YOUR SMALL BUSINESS HCA.
1.Five years after a new product has been introduced, sales begin to level off because customers are purchasing the competitor's brand. What strategy would.
Developed for the Department of Environmental Protection Bureau of Radiation Protection Refresher Training for X-Ray Equipment Operators Presented by David.
1 ointAviationAuthorities JAR 145 Course Part 2. 2 ointAviationAuthorities JAR General (a) No aircraft when used for Commercial Air Transport may.
1 Welcome The purpose of this presentation is to introduce you to the principles and methods of writing and presenting effective recommendations to improve.
Step To Excellence … Getting to Zero Adverse Errors 1.
Start This CD provides information for disabled employees, employers and other employees about employment of disabled people.
Moshe Kam* and Michael Lightner** * Robert Quinn Professor and Head; Electrical and Computer Engineering, Drexel University **Professor and Chair; Electrical,
© 2016 SlidePlayer.com Inc. All rights reserved.