Download presentation

Presentation is loading. Please wait.

Published byConnor Sawyer Modified over 2 years ago

1
1 Noise In Photodiode Applications By Art Kay and Bryan Zhao ( ) June 1, 2011

2
2 Contents Noise Review Photodiode Review Photodiode Noise Theory Bandwidth and Stability OPA827 Hand Calculation OPA827 Tina Spice Analysis OPA827 Measurement Example

3
3

4
4 Intrinsic Noise Error Source Generated by circuit itself (not pickup) Calculate, Simulate, and Measure

5
5 Time Domain – White noise normal distribution

6
6 What is Spectral Density? (Noise per unit frequency)

7
7 Convert Spectral Density to RMS Convert RMS to Peak-to-Peak

8
8 Why Photodiode Noise? Noise is a key parameter in photodiode design –Wide bandwidth (integrate more noise) –Low signal levels (noise more critical) Photodiode amplifier noise is more complex –Parasitic capacitance and sensor capacitance –Poles and zeros –Gain peaking

9
9 What can we do with the photodiode knowledge? TI Our new friends from National. The competition is in trouble!

10
10

11
11 Photodiode Basics Introduction Photodiodes convert light into current or voltage. Photodiode type PN photodiode – more wavelength selective PIN photodiode – wide spectral range (less selective) APD (Avalanche photodiode) – sensitive to low light, fast

12
12 Basic Photodiode Physics

13
13 Photodiode Basics Figure1.3 Photodiode Equivalent Circuit Figure1.4 Current VS. Voltage Characteristics

14
14 Photodiode Basic Use left equivalent circuit, the output current is given as : The open circuit voltage V oc is the output voltage when I o equals 0. Thus Voc becomes: Figure1.4 Photodiode Equivalent Circuit Figure1.5 Current VS. Voltage Characteristics

15
15 Photodiode and Control Source TINA model Light exciting source: 1) Use VG1 and VG2 voltage sources to simulate light power wave. 2) Use R1 and C1 shape light signal 3) The Voltage Control Current Source (VCCS1) simulates photodiode sensitivity. Photodiode Equivalent Circuit: 1), Current Source Id simulates Dark current 2), Diode is a ideal diode 3), Cd and Rd simulate photodiode's junction capacitor and dark Resistance. 4), Rs is series resistor, which is far smaller than Rd.

16
16

17
17 Photo-Diode Amp Noise Model

18
18 Photodiode noise

19
19 Noise Gain Gain seen by the noise voltage source. Simplify the model to compute Noise Gain

20
20 Noise Gain

21
21 Noise Gain

22
22 Noise Gain

23
23 Simulating Noise gain and noise bandwidth Break the loop to measure Aol, 1/B, and I to V Gain

24
24 Voltage Noise eni, eno and Eno Region 1Region 2Region 3Region 4Region 5 E no1 E no2 E no3 E no4 E no5

25
25 Voltage Noise e ni, e no and E no Region 1 noise: Region 2 noise: Region 3 noise: Region 4 noise: Region 5 noise: Total voltage noise:

26
26 Voltage Noise e ni, e no and E no e no Log scale e no Linear scale E no^2 Linear scale R.1 R.2R.3 R.4 R.5

27
27 Resistor Noise and Current Noise PolesKn Current noise and resistor noise are limited by the transimpedance (I-V gain) bandwidth

28
28

29
29 Parasitic Capacitance Limits the Bandwidth Max bandwidth with Min Cf Low Cf may be unstable Wide BW increases noise As shown Cf=Cs (stray cap)

30
30 Feedback Capacitance Required for Stability Noise Gain is key to stability Also called 1/Beta (in stability analysis) ROC = Rate of Closure ROC = (Aol slope) – (1/Beta slope) Unstable when ROC > 20dB/decade

31
31 Feedback Capacitance Required for Stability Applying a Step Input shows instability at output

32
32 Choosing a Minimum Cf for Stability

33
33

34
34 Noise Model for Simple Transimpedance Amp

35
35 Example Photodiode: PDB-C158 Unfortunately Cj is not specified at Vr=0V. We called the manufacturer for this info Cj=70pF for Vr=0V

36
36 Calculate Diode Current Noise

37
37 OPA827 Noise Hand Calculation (key numbers)

38
38 OPA827 Noise Hand Calculation

39
39 Poles and Zeros in Noise Gain Curve

40
40 1/f (flicker) Noise Corner

41
41 Output Noise from OPA Noise Voltage

42
42 Thermal (Resistor) Noise at Output

43
43 Current Noise to Voltage Noise at Output

44
44 The Final Total Noise

45
45 Reducing Noise (Higher Cf = Lower BW & Noise)

46
46 Flux Capacitor Advantage –Reduces noise to zero –Available on E-Bay Disadvantage –Requires 1.21 gigawatts –Size ADS1118 Quarter

47
47

48
48 OPA827 – test the model The noise performance is the same as datasheet. Low Noise : 4nV/Hz at 1kHz Low Offset Voltage: 150µV max JFET Input: I B = 15pA typ Wide Bandwidth: 22MHz

49
49 Simulated Spectral Density and Total Noise Output Noise Density Total Output Noise Calculated (rms) Simulated (rms) 116.7uV109.1uV

50
50

51
51 Validating Test Equipment Capability

52
52 Tektronix DPO 4034 Oscilloscope STDEV: 48uV (same as RMS) P-P: 6.6*STDEV=319uV 40s P-P: 320uV 1)Set DC couple, 20MHz bandwidth limit 2)Short input channel to measure noise floor

53
53 Agilent 4395A Spectrum Analyzer 1.Frequency Range: 10Hz~500MHz 2.Noise floor: 10nV/rtHz 3.Input Impedance: 50Ω Reference voltage Noise Floor: 10nV/rtHz

54
54 The Noise Floors are Not Good Enough

55
55 Solution: Use A Post Amp What amp do we Choose?

56
Use OPA847 as post amplifier Wideband, Ultra-Low Noise, Voltage Feedback, Operational Amplifier At the gain of 150, the bandwidth is 26MHz 0.85nV/ Hz Input Voltage Noise 2.5pA/ Hz Input Current Noise ±100uV Input Offset Voltage (Typical)) 575uVrms post amplifier noise in 20MHz bandwidth.

57
57 Post Amplifier Relatively small error! Vn827 = 109.4uV rms Vnpost = 16.03mV rms Vnpost/Gain = 16.03mV/150 = 106.8uV rms Gain=150

58
58 Test the Noise Floor

59
59 Test The Noise Floor – Post Amp Noise Scope Simulated (rms) Measured (rms) 575uV518uV STDEV: 518uV P-P: 6.6*STDEV=3.4mV 40s P-P: 3.88mV

60
60 Hardware Connections 1.PDB-C158-ND photodiode 2.70pF junction capacitance at Vr=0 V 3.100dB I-V gain 4.4pF compensation capacitor 5.±5V power supply.

61
61 Shield the Circuit if Possible Power Supply Vcc,Vss,GND Input & Output BNC connectors Shield

62
62 Divide by Gain for OPA827 Output Noise

63
63 Measured Total Output Noise (DPO 4034 Scope) OPA847 Measured at Scope: STDEV: 21.7mV P-P: 6.6*STDEV=143mV 40s P-P: 170mV At OPA827 (DUT) Output: Divide by gain Vn827 = 21.7mV/150 = 144.6uV Calculated (rms) Simulated (rms) Measured (rms) 116.7uV109.1uV144.6uV

64
64 Measured Spectral Density 4395A Spectrum Analyzer Linear Scale 1.Agilent 4395A Spectrum Analyzer test 1Hz~20MHz span, 3uV/div, REF=24uV. 2.The tested noise density curve shape is the same as simulation. Measurement 4395A Linear Scale Tina Spice Linear Scale Tina Spice Log Scale

65
65 OPA827-Noise Density 4395A Spectrum Analyzer

66
66 Thanks Bryan Zhao ( ) Matt Hann Collin Wells, Peter Semig, Curtis Mayberry References Jerald Graeme Art Kay HAMAMATSU Tim Green Before Questions, I must ask….

67
67 You will move to a new topic my young apprentice More noise presentations have we must Do I move over to the dark side?

68
68 Contact Info: (phone) (cell)

Similar presentations

© 2016 SlidePlayer.com Inc.

All rights reserved.

Ads by Google