We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byRaegan Wooding
Modified over 2 years ago
Set Data Directory so that GRASP knows where to look for data -File - Set Data Directory - Then browse to data folder
Use -Foreground: & Data Load: - Only use this tag for loading data - Type number on Numor(s) line
First part of SANS data treatment – Correct for Detector Efficiency Using a scattering substance that scatters uniformally across the detector For this use H 2 O
Zoom in around beam transmission Press this icon and drag around beam
Press Centre Calc Defines Beam Centre Coordinates
Press Te Calc Corrects for absorbance
Press Ts Calc Corrects for absorbance
Check box to subtract empty cell Check box to subtract cadmium
Mask edge of the detector and region behind the beam stop
Check box to correct the data
Check box to correct for solid angle
Check box to correct for instrument setup
Calibrates data to water scattering intensity
Check box to correct detector efficiency
Converts data absolute units
Open window for radial regrouping
Second part of SANS data treatment – Subtract buffers and correct for absorbance and detector efficiency
Calculate beam centre for sample data Transmission of buffer Transmission of sample
Check box to subtract Buffer Check box to subtract cadmium
Check box to correct the data
What data need to be measured? Scattering from: specimen in its container empty specimen container standard or calibration specimen.
1. a. Al alloys containing about 5 vol.% of Al 3 ZrTi precipitates (density = 2.92 gm/cc) were prepared by melt spinning (pouring a thin stream of the.
1. HOW TO START THE APPLICATION ? 1. Double click on the Application icon 2. The main menu will be loaded 3. Select the size of the album page 4. Click.
02/08/2015Regional Writing Centre2 02/08/2015Regional Writing Centre3.
COMPENSATION By: Ronald Mathieu. Compensation Why do we need compensation? –1) Because of long emission spectrum of dyes like FITC and PE.
Picasa 3 Tags. Types People Places General People Tags Select the “Un-named” folder Find a GOOD photo of someone you recognise Click on the “Add a name”
Creating First Class Web Pages Log into your account.
Jian Gui WANG Bragg Institute Meeting Java Algorithm Library Dec Java DRA Algorithm Library For Opal Neutron Scattering Data Analysis Team Jian.
All GPS ch 8 Edited 06/14/05 1 Exporting GPS Data Once you have finished correcting your data, you can export it using Pathfinder Office into a form that.
How to use the Hand-In Folder. Click on the Folder icon at the bottom of the screen.
OU NanoLab/NSF NUE/Bumm & Johnson Spectrophotometry Key Concepts Lambert’s Law of Absorption Beer’s Law Beer-Lambert Law Absorption Cross-Sections Photometric.
Welcome To Excel Day 1: Modules 1 & 2 Instructor: Mary Magellan Class will start at approximately 8:05 AM.
How to Use a Screen Recorder CS 292 Step 1 Download the software from (Mac users will need something different –
Project Deployment IT [211 CAP] How to convert your project to a full application.
Loading Files from SFTP Server. Browsing Remote File System 1) Use “Files of Type” combobox to select file format 2) File Browser shows only files of.
If you don’t have Google Earth downloaded already, you can go to to get it.http://earth.google.com/download-earth.html.
© ABB Group September 1, 2014 | Slide Turbidity Systems Accurate, reliable turbidity measurement.
Lesson 1: Learning Worksheet Fundamentals Microsoft Office Excel 2003 Lesson 1 Learning Worksheet Fundamentals.
Printing Set-up Drawing by switching to Layout 1 Insert a title block.
Vocabulary in VB So Far. Assignment: Used to change the value of an object at run time Used to change the value of an object at run time.
Potku – A New Analysis Software for Heavy Ion Elastic Recoil Detection Analysis K. Arstila 1, J. Julin 1, M.I. Laitinen 1, J. Aalto 2, T. Konu 2, S. Ka.
2001-Aug-15 Matt Newville Consortium for Advanced Radiation Sources University of Chicago / Advanced Photon Source XAFS: X-ray Absorption Fine-Structure.
Everything you need to know , 5.5, 5.6 Review.
Stoneware SB Portal Use your district username and password. District (OWA) District Servers including SAN Folder.
06/2006I.Larin PrimEx Collaboration meeting 0 analysis.
The parts of a Computer Window. Parts of a Computer Window Close Button.
A. Checks to run to make sure the instrument is in good working condition: Absorbance optics: ● run an intensity wavelength scan on an empty hole at 5.9.
Are You FRETting? Find Out for Sure With FLIM Frequency Domain FLIM for Your Scope Intelligent Imaging Innovations.
HiPAP Calibration Note if in doubt click the “Help” button.
Oakwood Technology College Task 1 - Problem-solving using a table and paper-based resource In pairs look at the worksheet 7.4P1a Zoo activity.doc Complete.
Analysing X-ray data using GudrunX. Outline Planning an experiment – Absorption – Fluorescence – Beam size Data required Outline of analysis process Step.
IV.Creating & Editing a Video/Movie. Editing Techniques To open a project On the File menu, click Open Project. In the File name box, locate the saved.
This is what we want to produce Import sketch dimensions Label angles Set the scale Add linear dimensions.
Mr. Walter’s Notes on How to Use the Calculator to Find the Equation of a Line when you Know Coordinate Points.
WORKING SMART Crystal M. Thomas Henrico County DSS (804) POSSESS Central Region Member October 31, 2007.
Exporting. Export Step 1 – In Outlook select Contacts.
Morphometric or 3D reconstruction Chinmaya Sadangi.
1 Getting Started with C++. 2 Objective You will be able to create, compile, and run a very simple C++ program on Windows, using Visual Studio 2008.
Sample Maps of Tanzania. Map #1: All ICAP-supported sites in Tanzania 1)Select Tanzania (or any other specific ICAP country) 2)Check the box indicating.
Using Microsoft Visual Studio 2005 Original by Suma Rao Revised by John G. McMahon ( 9/6/2008 )
Glow Experiment Outline Brandon Hanold. Outline 1.Glow Description 2.Experiment Design 3.Acquisition Loop Variables and Keywords 4.Reduction Results 5.Reduction.
VLSI Layout using Microwind Kishore C Acharya. 2 Getting Microwind Go to the website Download the freeware version of the microwind.
Transmission test for waveguide Hideyuki Sakamoto Tracker Phone Meeting, 25 th May,2006 Contents Method Measured luminosity Results and next step.
Jump measuring systems. Jumps measuring systems Now we use three systems for measuring jumps 1.Boettcher system 2.Corson system 3.SplashEye.
1 After completing this lesson, you will be able to: View and open folders. Open, edit, and save files. Print files. Sort files. (continued)
Academy Algebra II 4.2: Building Linear Functions From Data HW: p (3-8 all,18 – by hand,20 – calc) Bring your graphing calculator to class on Monday.
Adobe Photoshop CS5 Chapter 3 Working with Layers.
…….CT Physics - Continued V.G.WimalasenaPrincipal School of radiography.
Scisat Test Readiness Review MAESTRO Evaluation and Verification Tests University of Toronto 29 January 2003.
Presented by: Andrew Bedard English 393. Introduction FTP or file transfer protocol is a way to get files from across a network from one computer to another.
© 2017 SlidePlayer.com Inc. All rights reserved.