for internal use only First Clinical Experience with a new accelerometer-based Sensor during different daily activities, Candinas R, Erne P, Schönbeck M, European Trilogy DR+ Investigators, EUR.J.C.P.E, 1996:6.177:A706 26/5 Omnisense (literature)
for internal use only Sensor Pacemakers Maximum Rate (Minimum Rate = Basic Rate) Threshold Slope (Rate responsive factor) Reaction time Recovery time Parameters
for internal use only Activity Level and Sensor Signal Z Z Z ZZ Z Z Z
for internal use only Mean Value Max Rate Min Rate Rate Activity Z Z Z ZZ Z Z Z Threshold AUTO -0.5 Sensor Threshold 60 70 80 90 100 110 120 50 40 Too low threshold: Sensor driven rate at rest! Too high threshold: Basic rate at low level exercise!
for internal use only Max Rate Min Rate Rate Activity Z Z Z ZZ Z Z Z Sensor Slope 60 70 80 90 100 110 120 50 40 Slope 8 Slope 8: This activity level gives a rate increase of 25 min -1
for internal use only Max Rate Min Rate Rate Activity Z Z Z ZZ Z Z Z Sensor Slope 60 70 80 90 100 110 120 50 40 Slope 12 Slope 12: The same activity level gives a rate increase of 40 min -1
for internal use only Max Rate Min Rate Rate Activity Z Z Z ZZ Z Z Z Sensor Slope, Basic Rate and Max. Sensor Rate 60 70 80 90 100 110 120 50 40 Slope 6 A low slope, a low basic rate or a high max. sensor rate may exclude ability to reach max. sensor rate.
for internal use only Slope and Target Rate Target Rate, slope 8 Target Rate, slope 12
for internal use only Reaction- & Recovery Time Reaction TimeRecovery Time
Your consent to our cookies if you continue to use this website.