We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byAaron Hamilton
Modified over 3 years ago
Therapeutic UltrasoundChapter 7
Description Acoustical energy (sound waves) above the range of human hearing Therapeutic range: 0.75 to 3.3 MHz Effects: Thermal Nonthermal (mechanical) © 2005 – FA Davis
Uses Calcific bursitis Inflammatory conditions Joint contractures PainMuscle spasm Acute orthopedic injuries (low pulses, low intensity) © 2005 – FA Davis
Production of UltrasoundAn alternating current is passed through a crystal The current causes the crystal to vibrate Electropiezo effect Vibrating crystal produce high-frequency sound waves © 2005 – FA Davis
Effective Radiating Area (ERA)Area of the crystal that actually produces sound waves Produces more than 5% of the energy at 5 mm from the transducer face ERA is always smaller than the transducer face Energy is concentrated near the center ERA © 2005 – FA Davis
Beam Profile Multiple waves emerge from the headEnergy diverges as it moves away from the source Energy is uniform close to the head Near zone (Fresnel zone) Becomes less consistent farther away from the head Spatial peak intensity Spatial Peak Intensity © 2005 – FA Davis
Beam Nonuniformity Ratio (BNR)Spatial Peak Intensity Describes the amount of variation in the beam Ratio between: Spatial peak intensity (SPI) Average intensity (SAI - metered output) BNR = SPI/SAI Perfect beam would have a BNR of 1:1 Minimally acceptable BNR is 8:1 The actual peak output is equal to the SAI * the BNR 10 W total output * 6:1 BNR 60 watt SPI Spatial Average Intensity (Displayed on the unit) © 2005 – FA Davis
Modes of Application ContinuousUltrasonic energy is constantly produced Can produce thermal effects based on: Output intensity Treatment duration Pulsed Ultrasonic output is regularly interrupted Produces nonthermal effects © 2005 – FA Davis
Continuous Output (100% Duty Cycle) Pulsed Output (67% Duty Cycle)Ultrasonic output is cycled “On” and “Off” On = Pulse length Off = Pulse interval Expressed as a Duty Cycle ON/(ON+OFF) * 100 20mSec/(20mSec+10mSec) * 100 20/30 * 100 67% Continuous Output (100% Duty Cycle) Pulsed Output (67% Duty Cycle) © 2005 – FA Davis
Output Frequency Measured in megahertz (MHz)1 MHz = 1,000,000 waves per second Determines the depth of effects 1 MHz Output Penetrates 5 to 7 cm Thermal effects last longer More divergent beam 3 MHz Output Penetrates 2 to 3 cm Heats 3 times faster than 1 MHz output More collimated beam © 2005 – FA Davis
Power and Intensity Spatial Average IntensitySpatial Average Temporal Peak Intensity Spatial Average Temporal Average Intensity
Spatial Average Intensity (SAI)Describes the energy per unit of area Total output (watts)/area Watts/effective radiating area (cm2) W/cm2 15 watts being applied with a 10 cm2 ERA 15 Watts / 10 cm2 1.5 W/cm2 © 2005 – FA Davis
Power Measures – Pulsed OutputSpatial Average Temporal Peak Intensity (SATP) The average energy delivered during the “On” time of the duty cycle. Spatial Average Temporal Average Intensity (SATA) Energy delivered over time Spatial Average Intensity * Duty Cycle Meaningful only during pulsed output © 2005 – FA Davis
Biophysical Effects Thermal
Thermal Effects Increased sensory nerve conduction velocityIncreased motor nerve conduction velocity Increased extensibility of collagen-rich Increased vascular permeability structures Increased collagen deposition Increased blood flow Reduction of muscle spasm Increased macrophage activity Enhanced adhesion of leukocytes to damaged endothelial cells © 2005 – FA Davis
Heating ClassificationsIncrease Used For Mild 1°C Mild inflammation Accelerate metabolism Moderate 2° – 3°C Decreasing muscle spasm Decreasing pain Increasing blood flow Chronic inflammation Vigorous 3° – 4°C Tissue elongation Scar tissue reduction © 2005 – FA Davis
Heating Rate Heating rate and magnitude is based on: Duty cycleOutput frequency Intensity Target tissues Size of the treatment area © 2005 – FA Davis
Thermal Effects Same as other heat modalitiesSmaller volume of tissue Shorter duration of effects Preheat the skin with a moist heat pack Decreases the time to reach vigorous heating Poorly vascularized, collagen-rich tissues are preferentially heated Fascia, tendon, scar tissue Tissues containing an increased proportion of fluid do not heat as well Adipose tissue, articular fluid © 2005 – FA Davis
Biophysical Effects Nonthermal
Nonthermal Effects Granulation tissue production Synthesis of proteinIncreased cell membrane permeability Altered rates of diffusion across the cell membrane Increased vascular permeability Secretion of cytokines Increased blood flow Increased fibroblastic activity Stimulation of phagocytosis Granulation tissue production Synthesis of protein Synthesis of collagen Reduction of edema Diffusion of ions Tissue regeneration Formation of stronger deformable connective tissue © 2005 – FA Davis
Nonthermal ApplicationPulsed output 20 to 25% duty cycle Nonthermal output intensity Continuous output 100% duty cycle Output intensity of less than 0.3 W/cm2 © 2005 – FA Davis
Acoustical Streaming Ultrasound causes interstitial fluids to flowFluids strike cell membranes Produce eddy currents Eddy currents displace ions and molecules Alter: Cell membrane permeability Cellular function © 2005 – FA Davis
Effect on Injury Response
Cellular Response Acoustical streaming: Thermal effects:Increases cell membrane permeability Alters cell membrane diffusion rate Increased histamine release Mast cell degranulation Increased rate of protein synthesis Thermal effects: Increased cell metabolism Increased rate of inflammation © 2005 – FA Davis
Inflammation May lead to an earlier onset of proliferationIncreased fibroblast proliferation Release of growth factors and platelets Increased macrophage activity Leukocytes bind to damaged endothelial cells Cell division is increased © 2005 – FA Davis
Inflammation Frequency Specificity 1 MHz Output 3 MHz OutputRelease of preformed fibroblasts 3 MHz Output Increased synthesis and secretion of fibroblast precursors Increased in areas of high collagen concentration © 2005 – FA Davis
Blood and Fluid DynamicsMay increase blood flow for 45 minutes Thermal effects Decreased vascular tone Histamine release Causes vasodilation Moist heat application prior to treatment decreases net increase in blood flow © 2005 – FA Davis
Pain Control Direct Pain ReductionIncreased nerve cell sodium permeability Alters nerve function Increases pain threshold Indirect Pain Reduction Increased blood flow Increased capillary permeability Increased oxygen delivery Decreased muscle spasm © 2005 – FA Davis
Muscle Spasm Reduced secondary to: Decreased painAltered nerve conduction velocity Increased temperature (counterirritant effect) Muscle relaxation © 2005 – FA Davis
Tissue Elasticity Ultrasound preferentially heats collagen-rich tissues (tendon, fascia, scar tissue) Temperature must be increased 7.2°F Stretching window lasts approximately 3 minutes following the treatment Place tissues on stretch during application Perform stretching/mobilization immediately following the treatment Multiple treatments are required to gain length © 2005 – FA Davis
Wound Healing Tendon Healing Continuous US application may:Increase tensile strength Increase collagen deposition Skin Ulcers 3 MHz, low-intensity pulsed output may assist the healing process Cover the wound with an occlusive dressing © 2005 – FA Davis
Fracture Healing Low-intensity pulsed outputAccelerates rate of fracture healing for: Acute fractures Nonunion fractures Stress fractures Requires specialized unit Biophysical Effects: Mechanical (sound) energy strikes bone Microvibration of bone triggers growth (osteogenesis) PARAMETERS Frequency 1.5 MHz ERA 3.88 cm2 Intensity 30 mW/cm2 Treatment Duration 20 minutes Daily © 2005 – FA Davis
Contraindications Acute conditions (thermal mode) Ischemic areasAreas of impaired circulation Over areas of deep vein thrombosis Anesthetic areas Over cancerous tumors Sites of active infection or sepsis Over the spinal cord or large nerve plexus in high doses Exposed penetrating metal (eg, external fixation devices) Around the eyes, heart, skull, or genitals Over the thorax in the presence of an implanted pacemaker Pregnancy when used over the pelvic or lumbar areas Over a fracture site before healing is complete Stress fracture sites or sites of osteoporosis Over the pelvic or lumbar area in menstruating female patients © 2005 – FA Davis
ESAT 3640 Therapeutic Modalities
Shortwave Diathermy Chapter 9.
Ch. 12 Therapeutic Modalities
ULTRASOUND Chapter 7.
Ultrasound. History: Available in 19th century. Available in 19th century. Was for sonar (SONAR Sound Navigation and Ranging) Was for sonar (SONAR
By Ass.Prof.Dr. Gehan Mosad
Ultrasound 10cm 2 Ultrasound (US) Therapeutic ultrasound –Uses acoustic energy, delivered at high frequencies for therapeutic purposes Electrical current.
Supriya Babu. Topics Physical basis of heat and cold Measurement of temperature: Thermometry Mapping of body’s surface temperature: Thermography Heat.
Heat in Therapy Class 3 Supriya Babu.
Injury Treatment Ms. Bowman. Injury Treatment 2 areas o Therapeutic Modalities o Therapeutic Exercise.
Therapeutic Ultrasound (Part two)
Therapeutic Ultrasound Sports Medicine 6/11/
Physiological Properties of Thermal Modalities (2)
Infrared Radiation Prof.Dr. Gehan Mosaad.
© 2017 SlidePlayer.com Inc. All rights reserved.