We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byThomas Cassell
Modified over 2 years ago
Gilian Air Sampling Pump Field Calibration A Step-By-Step Guide
Items required Sampling Pump Collection Media Gilibrator Calibrator
A Field Calibration Consists of Two Parts The Pre-Calibration (Pre-Cal) is done before the sample is taken The Post-Calibration (Post Cal) is done after the sample is taken
The Pre-Cal The Pre-Cal is used to set the desired sample flow rate, and to verify it on an air flow calibrator.
Conducting the Pre-Calibration
The Pre-Cal Set the Flow Rate Observe the Flow Rate Record the Flow Rate An average of 6 to 10 Gilibrator readings is recommended
Run The Sample Pump is placed on the belt Sample media is placed on the lapel (Breathing Zone) and pointed downward Record the Start Time Record the End Time
The Post-Cal The Post-Cal is used to obtain the final flow rate and to verify that the flow rate stayed within +/- 5% of the set flow.
The Post-Cal Observe the Final Flow Rate Record the Final Flow Rate An average of 6 to 10 Gilibrator readings is recommended
Average Flow Rate The Pre-Cal flow rate and the Post-Cal flow rate may be averaged to obtain an average flow rate for the sample.
Pre-Cal Flow Rate + Post Cal Flow Rate = Avg Flow Rate 2 Determining the Average Flow Rate
Cyclone Calibration The Gilian 10mm nylon cyclone has only an exit fitting and must be placed inside a calibration jar to accept the Gilibrator.
Low Flow Modules Calibrating with low flow modules
Constant Flow Modules Set pump flow to 1500cc/min Turn module switch on Connect sample tubing to pump Adjust desired flow rate at module Read flow rate at Gilibrator
Multi-Flow Modules Set pump to 1500 cc/min Turn module switch on Connect sample tubing to pump Adjust flow rate at holder Adjust each tube twice Read the flow rates
Further Information For detailed information on any of the Gilian products, please consult the product operation manuals. Manuals may be accessed through our web site at www.sensidyne.com
Thank You Sensidyne, LP. Clearwater, FL, USA www.sensidyne.com Fax 727-539-0550 Phone 727-530-3602 & 1-800-451-9444
QuadMode SM Simultaneous Sampling. Reliable Performance & Innovative Features for Dependable Results High performance – cc/min flow range.
The World’s Most Reliable Air Sampling Pumps. Introduction to Gilian Gilian air sampling pumps are manufactured by Sensidyne in Clearwater, Florida. Product.
Personal Air Sampling for Crystalline Silica Exposure Ron Roberson Sensidyne.
Water Quality Supplies Waste bottle 1413 solution pH 10 solution pH 4 solution Solution cups Water sample cup Hanna Meter.
Electric Drives FEEDBACK LINEARIZED CONTROL Vector control was invented to produce separate flux and torque control as it is implicitely possible.
Cyclones A CENTRIFUGAL FORCE IN AIR SAMPLING A Presentation by SKC Inc.
OSHA Personal Air Sampling Presented By: Joseph Cantone Industrial Hygienist/Project Manager.
OSHA A GUIDE TO THE NEGATIVE EXPOSURE ASSESSMENT.
TSI Nephelometer Zero Filter Assembly Zero filter valve Zero filter Zero filter motor Metal flange.
Collaborative Action Research Option (2013) Staff Development.
SPOTLIGHT ON THE SKC AIRCHEK ® XR5000 PUMP. THE SUPERCHARGED SKC AIRCHEK XR5000 The FIRST personal sampling pump with a lithium-ion battery option. Experience.
Mass Flow Controller Calibration The following slides show how to calibrate mass flow controllers using a flow reference cell and a PID controller. This.
By: Edith Leticia Cerda
Maricopa County Air Quality Department Performance Test Specifics Rob Arpino Environmental Engineering Specialist.
6 th Annual Focus Users’ Conference 6 th Annual Focus Users’ Conference Rollover Presented by: C.J. Obartuck Software Trainer & Miranda Robinson Project.
Operating Nitrox Systems 11 Presented by Nuvair. Operating Nitrox Systems2 Membrane systems produce nitrox efficiently Compressed air flows through the.
Please let me know of any changes as soon as you are aware: Personnel including names, phone or . Prime Contacts Start or Anticipated Completion.
1600 Series Operation (See Installation and Operation Manual for detailed instructions)
Gilian Air Sampling Pumps The World’s Most Reliable Air Sampling Equipment.
1 On Rate for Fuel Oil Equipment. 2 Purpose To insure that the equipment is delivering the correct heating value Oil fired equipment must fire within.
An innovative tool for the review of health and safety work practices and the implementation of effective controls of particulate exposures.
Autoanalyzer Prof. Moustafa M. Mohamed. Autoanalyzer system The autoanalyzer sequentially measures blood displays this on a graphical readout. This accomplished.
CrackingSiebel.com Utility Siebel Repository Extract (SRE) Tool.
1 Zero Point Correction. 2 Basic Steps - I For a given site on a given day: 1.Create a magnetic flat field from the east & west calibration images. 2.Using.
Mrs. Ward Ridge Road Middle School. We need to breathe air to stay alive. Wow, the respiratory system sure has an important job! What are some factors.
Mark Heggli Innovative Hydrology, Inc. Consultant to the World Bank Expert Real-time Hydrology Information Systems Module 12: Assuming Management of an.
1 Copyright © 2007 General Motors Corporation. All Rights Reserved.
NVLAP Overview and Accreditation Process March 2006.
DHP for Houses with Electric FAF Research Plan: Revisions Adam Hadley, Ben Hannas, Bob Davis, My Ton R&E Subcommittee February 25, 2015.
ESTIMATING - A CONTRACTORS PERSPECTIVE. Company History/Background Oman Construction was founded in 1876 First bridge contract in 1881 First Highway Contract.
Flushing. Annual flushing is enough for many systems, but some water and emitter combinations require almost daily flushing to control clogging. If frequent.
Recent Results from Dragonfire Armor Simulation Experiments Farrokh Najmabadi, Lane Carlson, John Pulsifer UC San Diego HAPL Meeting, Naval Research Laboratory.
Gas Detector Tubes Low-cost Detection of Hazardous Gases & Vapors.
¾-1/2” Red. Union 3/8”OD ACSM Inlet Configuration 0.1 lpm to ACSM with PM 1 passing thru aerodynamic lens 2.9 lpm to external Pump* Sheath Flow 6 lpm dry.
GLOBE PROTOCOLS By: Angela Gonyer, Noah Hughes, Tyy Williams, Eli Lopez, and Ariana Viray.
WORKING DETAILS OF C.F.L. FLARE MAKING MACHINE.
Handheld Flow Calibrator for Aerosol Observing System Jonathan Ma, Dung (Yung) Hoang Nguyen, Mario Zuniga Department of Mechanical and Aerospace Engineering,
1 NORTHWEST ENERGY EFFICIENCY ALLIANCE Northwest Ductless Heat Pump Pilot Project Impact & Process Evaluation: Billing Analysis Ecotope, Inc. February.
LSME: L. Flora 1 Detailed Activities Key Activities Role Description To identify expendable and consumable supply needs and requisition, receive, and issue.
Multi- Q Introd uction Welcome to … MME Double Jeopardy.
Can you think of a way that we could measure the amount of air our lungs can hold?
A1 Acoustic Rhinometer The following presentation gives an overview of the main functions of the A1Acoustic Rhinometer, it’s accessories and normal operational.
Registration Process 1. Change Zone team will be glad to help you answering any question about the programs and to give you further explanation. Please.
WHO - PSM Water –part 3 Workshop on GMP and Quality Assurance of HIV products Shanghai, China 28 Feb - 4 March 2005 Maija Hietava M.Sci.Pharm Quality Assurance.
By Saleh A. Al-Hassoun Associate Professor Department of Civil Engineering College of Engineering King Saud University Riyadh, Saudi Arabia
May 2002[NUMUG] Listening to the Wind1 of 17 LISTENING TO THE WIND TVA Experience with Installation and Operation of Ultrasonic Wind Sensors Kenneth G.
SPOTLIGHT ON THE SKC AIRCHEK ® XR5000 PUMP. THE SUPERCHARGED SKC AIRCHEK XR5000 The FIRST personal sampling pump with supercharged lithium-ion battery.
Cleaning Critical Orifice and Focusing Nozzle. Step 1. Remove optics block from column Remove the 2 Allen head screws holding the optics block to the.
Child Protection Rapid Assessment Tool Inter-agency Child Protection Working Group & Save the Children Picture by: Hani Mansourian Training material developed.
Problem Solving Process Questions Research Development Plan Test Data Collection Refine Solution Deployment Plan.
© 2017 SlidePlayer.com Inc. All rights reserved.