We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byThomas Cassell
Modified over 4 years ago
Gilian Air Sampling Pump Field CalibrationA Step-By-Step Guide
Items required Sampling Pump Collection Media Gilibrator Calibrator
A Field Calibration Consists of Two PartsThe Pre-Calibration (Pre-Cal) is done before the sample is taken The Post-Calibration (Post Cal) is done after the sample is taken
The Pre-Cal The Pre-Cal is used to set the desired sample flow rate, and to verify it on an air flow calibrator.
Conducting the Pre-Calibration
The Pre-Cal Set the Flow Rate Observe the Flow RateRecord the Flow Rate An average of 6 to 10 Gilibrator readings is recommended
Run The Sample Pump is placed on the beltSample media is placed on the lapel (Breathing Zone) and pointed downward Record the Start Time Record the End Time
The Post-Cal The Post-Cal is used to obtain the final flow rate and to verify that the flow rate stayed within +/- 5% of the set flow.
The Post-Cal Observe the Final Flow Rate Record the Final Flow RateAn average of 6 to 10 Gilibrator readings is recommended
Average Flow Rate The Pre-Cal flow rate and the Post-Cal flow rate may be averaged to obtain an average flow rate for the sample.
Pre-Cal Flow Rate + Post Cal Flow Rate = Avg Flow Rate 2Determining the Average Flow Rate
Cyclone Calibration The Gilian 10mm nylon cyclone has only an exit fitting and must be placed inside a calibration jar to accept the Gilibrator.
Low Flow Modules Calibrating with low flow modules
Constant Flow Modules Set pump flow to 1500cc/minTurn module switch on Connect sample tubing to pump Adjust desired flow rate at module Read flow rate at Gilibrator
Multi-Flow Modules Set pump to 1500 cc/min Turn module switch onConnect sample tubing to pump Adjust flow rate at holder Adjust each tube twice Read the flow rates
Further Information For detailed information on any of the Gilian products, please consult the product operation manuals. Manuals may be accessed through our web site at
Sensidyne, LP. Thank You Clearwater, FL, USA www.sensidyne.comFax Phone &
By: Edith Leticia Cerda
AREP GAW Section 5 How Are Forecasts Used? Protect Public Health Operate Emissions Reduction Programs.
SPOTLIGHT ON THE SKC AIRCHEK® XR5000 PUMP
An innovative tool for the review of health and safety work practices and the implementation of effective controls of particulate exposures.
ESTIMATING - A CONTRACTORS PERSPECTIVE. Company History/Background Oman Construction was founded in 1876 First bridge contract in 1881 First Highway Contract.
Test plans. Test Plans A test plan states: What the items to be tested are At what level they will be tested What sequence they are to be tested in How.
E2 Management Suite Employee Scheduling & Sales Forecasting
Cyclones A CENTRIFUGAL FORCE IN AIR SAMPLING
WHO - PSM Water –part 3 Workshop on GMP and Quality Assurance of HIV products Shanghai, China 28 Feb - 4 March 2005 Maija Hietava M.Sci.Pharm Quality Assurance.
WORKING DETAILS OF C.F.L. FLARE MAKING MACHINE.
Child Protection Rapid Assessment Tool
SPOTLIGHT ON THE SKC AIRCHEK ® XR5000 PUMP. THE SUPERCHARGED SKC AIRCHEK XR5000 The FIRST personal sampling pump with a lithium-ion battery option. Experience.
inFire HTC speed Furnace Webinar
Operating Nitrox Systems 11 Presented by Nuvair. Operating Nitrox Systems2 Membrane systems produce nitrox efficiently Compressed air flows through the.
A1 Acoustic Rhinometer The following presentation gives an overview of the main functions of the A1Acoustic Rhinometer, it’s accessories and normal operational.
6 th Annual Focus Users’ Conference 6 th Annual Focus Users’ Conference Rollover Presented by: C.J. Obartuck Software Trainer & Miranda Robinson Project.
Alan Gascoyne - Director, Certification Steven Hoy - Engineer, Certification Judi Farrell - Policy & Legislation Team Simon Lilley - RAWS Team RAWS INFORMATION.
QuadMode SM Simultaneous Sampling. Reliable Performance & Innovative Features for Dependable Results High performance – cc/min flow range.
OSHA Personal Air Sampling Presented By: Joseph Cantone Industrial Hygienist/Project Manager.
TSI Nephelometer Zero Filter Assembly Zero filter valve Zero filter Zero filter motor Metal flange.
© 2018 SlidePlayer.com Inc. All rights reserved.